Cui, Junning; He, Zhangqiang; Jiu, Yuanwei; Tan, Jiubin; Sun, Tao
2016-09-01
The demand for minimal cyclic nonlinearity error in laser interferometry is increasing as a result of advanced scientific research projects. Research shows that the quadrature phase error is the main effect that introduces cyclic nonlinearity error, and polarization-mixing cross talk during beam splitting is the main error source that causes the quadrature phase error. In this paper, a new homodyne quadrature laser interferometer configuration based on nonpolarization beam splitting and balanced interference between two circularly polarized laser beams is proposed. Theoretical modeling indicates that the polarization-mixing cross talk is elaborately avoided through nonpolarizing and Wollaston beam splitting, with a minimum number of quadrature phase error sources involved. Experimental results show that the cyclic nonlinearity error of the interferometer is up to 0.6 nm (peak-to-valley value) without any correction and can be further suppressed to 0.2 nm with a simple gain and offset correction method. PMID:27607285
A Homodyne Quadrature Laser Interferometer for Micro-Asperity Deformation Analysis
PogaČnik, Aljaž; Požar, Tomaž; Kalin, Mitjan; Možina, Janez
2013-01-01
We report on the successful realization of a contactless, non-perturbing, displacement-measuring system for characterizing the surface roughness of polymer materials used in tribological applications. A single, time-dependent, scalar value, dubbed the collective micro-asperity deformation, is extracted from the normal-displacement measurements of normally loaded polymer samples. The displacement measurements with a sub-nanometer resolution are obtained with a homodyne quadrature laser interferometer. The measured collective micro-asperity deformation is critical for a determination of the real contact area and thus for the realistic contact conditions in tribological applications. The designed measuring system senses both the bulk creep as well as the micro-asperity creep occurring at the roughness peaks. The final results of our experimental measurements are three time-dependent values of the collective micro-asperity deformation for the three selected surface roughnesses. These values can be directly compared to theoretical deformation curves, which can be derived using existing real-contact-area models. PMID:23296328
Optimization of quadrature signal processing for laser interferometers for demanding applications
NASA Astrophysics Data System (ADS)
PodŻorny, Tomasz; Budzyń, Grzegorz; Tkaczyk, Jakub
2016-06-01
Presented paper performs an analysis of quadrature signal processing algorithms for high demanding laser interferometry applications. Careful signal processing is required to minimize nonlinearities which come from optical path and components' imperfections, and reduce overall instrumental error. Paper focuses on algebraic fits, because implementation for real time systems was a main requirement. The most demanding applications are stationary measurements where the position slightly fluctuates in the range below one fringe period. Therefore, analysis was performed for samples that were spread along a few milliradians of a full circle.
Quadrature phase interferometer for high resolution force spectroscopy
NASA Astrophysics Data System (ADS)
Paolino, Pierdomenico; Aguilar Sandoval, Felipe A.; Bellon, Ludovic
2013-09-01
In this article, we present a deflection measurement setup for Atomic Force Microscopy (AFM). It is based on a quadrature phase differential interferometer: we measure the optical path difference between a laser beam reflecting above the cantilever tip and a reference beam reflecting on the static base of the sensor. A design with very low environmental susceptibility and another allowing calibrated measurements on a wide spectral range are described. Both enable a very high resolution (down to 2.5 × 10^{-15} m/sqrtHz), illustrated by thermal noise measurements on AFM cantilevers. They present an excellent long-term stability and a constant sensitivity independent of the optical phase of the interferometer. A quick review shows that our precision is equaling or out-performing the best results reported in the literature, but for a much larger deflection range, up to a few μm.
Quadrature phase interferometer for high resolution force spectroscopy
Paolino, Pierdomenico; Aguilar Sandoval, Felipe A.; Bellon, Ludovic
2013-09-15
In this article, we present a deflection measurement setup for Atomic Force Microscopy (AFM). It is based on a quadrature phase differential interferometer: we measure the optical path difference between a laser beam reflecting above the cantilever tip and a reference beam reflecting on the static base of the sensor. A design with very low environmental susceptibility and another allowing calibrated measurements on a wide spectral range are described. Both enable a very high resolution (down to 2.5×10{sup −15} m/√(Hz)), illustrated by thermal noise measurements on AFM cantilevers. They present an excellent long-term stability and a constant sensitivity independent of the optical phase of the interferometer. A quick review shows that our precision is equaling or out-performing the best results reported in the literature, but for a much larger deflection range, up to a few μm.
Quadrature phase interferometer for high resolution force spectroscopy.
Paolino, Pierdomenico; Aguilar Sandoval, Felipe A; Bellon, Ludovic
2013-09-01
In this article, we present a deflection measurement setup for Atomic Force Microscopy (AFM). It is based on a quadrature phase differential interferometer: we measure the optical path difference between a laser beam reflecting above the cantilever tip and a reference beam reflecting on the static base of the sensor. A design with very low environmental susceptibility and another allowing calibrated measurements on a wide spectral range are described. Both enable a very high resolution (down to 2.5×10(-15) m/√Hz), illustrated by thermal noise measurements on AFM cantilevers. They present an excellent long-term stability and a constant sensitivity independent of the optical phase of the interferometer. A quick review shows that our precision is equaling or out-performing the best results reported in the literature, but for a much larger deflection range, up to a few μm. PMID:24089852
Quantitative phase imaging using grating-based quadrature phase interferometer
NASA Astrophysics Data System (ADS)
Wu, Jigang; Yaqoob, Zahid; Heng, Xin; Cui, Xiquan; Yang, Changhuei
2007-02-01
In this paper, we report the use of holographic gratings, which act as the free-space equivalent of the 3x3 fiber-optic coupler, to perform full field phase imaging. By recording two harmonically-related gratings in the same holographic plate, we are able to obtain nontrivial phase shift between different output ports of the gratings-based Mach-Zehnder interferometer. The phase difference can be adjusted by changing the relative phase of the recording beams when recording the hologram. We have built a Mach-Zehnder interferometer using harmonically-related holographic gratings with 600 and 1200 lines/mm spacing. Two CCD cameras at the output ports of the gratings-based Mach-Zehnder interferometer are used to record the full-field quadrature interferograms, which are subsequently processed to reconstruct the phase image. The imaging system has ~12X magnification with ~420μmx315μm field-of-view. To demonstrate the capability of our system, we have successfully performed phase imaging of a pure phase object and a paramecium caudatum.
Laser interferometer micrometer system
NASA Technical Reports Server (NTRS)
Logue, S. H.
1969-01-01
Laser micrometer measures dimensions of precision gyro and gas bearing parts using the principle of measuring light phase changes rather than a direct fringe count. The system uses light beams to eliminate errors due to deformations and surface irregularities, and three interferometers.
Niebauer, T M; Constantino, A; Billson, R; Hankla, A; Nelson, P G
2015-06-20
A corner-cube retroreflector has the property that the optical path length for a reflected laser beam is insensitive to rotations about a mathematical point called its optical center (OC). This property is exploited in ballistic absolute gravity meters in which a proof mass containing a corner-cube retroreflector is dropped in a vacuum, and its position is accurately determined with a laser interferometer. In order to avoid vertical position errors when the proof mass rotates during free fall, it is important to collocate its center of mass (COM) with the OC of the retroreflector. This is commonly done using a mechanical scale-based balancing procedure, which has limited accuracy due to the difficulty in finding the exact position of the COM and the OC. This paper describes a novel way to achieve the collocation by incorporating the proof mass into a pendulum and using a quadrature interferometer to interrogate its apparent translation in its twist mode. The mismatch between the COM and OC generates a signal in a quiet part of the spectrum where no mechanical resonance exists. This allows us to tune the position of the COM relative to the OC to an accuracy of about 1 μm in all three axes. This provides a way to directly demonstrate that a rotation of the proof mass by several degrees causes an apparent translation in the direction of the laser beam of less than 1 nm. This technique allows an order of magnitude improvement over traditional methods of balancing. PMID:26193025
Laser interferometer calibration station
NASA Astrophysics Data System (ADS)
Campolmi, R. W.; Krupski, S. J.
1981-10-01
The laser interferometer is a versatile tool, used for calibration over both long and short distances. It is considered traceable to the National Bureau of Standards. The system developed under this project was to be capable of providing for the calibration of many types of small linear measurement devices. The logistics of the original concept of one location for calibration of all mics, calipers, etc. at a large manufacturing facility proved unworkable. The equipment was instead used for the calibration of the large machines used to manufacture cannon tubes.
NASA Technical Reports Server (NTRS)
Dobson, Chris C.; Jones, Jonathan E.; Chavers, Greg
2003-01-01
A polychromatic microwave quadrature interferometer has been characterized using several laboratory plasmas. Reflections between the transmitter and the receiver have been observed, and the effects of including reflection terms in the data reduction equation have been examined. An error analysis which includes the reflections, modulation of the scene beam amplitude by the plasma, and simultaneous measurements at two frequencies has been applied to the empirical database, and the results are summarized. For reflection amplitudes around 1096, the reflection terms were found to reduce the calculated error bars for electron density measurements by about a factor of 2. The impact of amplitude modulation is also quantified. In the complete analysis, the mean error bar for high- density measurements is 7.596, and the mean phase shift error for low-density measurements is 1.2". .
Laser interferometer for absolute distance measurement based on a tunable VCSEL laser
NASA Astrophysics Data System (ADS)
Cip, Ondrej; Mikel, Bretislav; Lazar, Josef
2005-02-01
In the work, we present the absolute distance interferometer with a narrow-linewidth tunable VCSEL laser (Vertical-Cavity Surface-Emitting Laser) working at &lambda ~760 nm. As a detection technique, we use a fast wavelength-scanning interferometry improved by an amplitude division of the interference fringe with using two signals in quadrature. Used VCSEL laser is wide tunable with the mod-hop free tuning range more than 1.2 nm by means of the amplitude modulation of the injection current. We control the stabilization and tuning process of the laser wavelength with using the frequency lock to a Fabry-Perot resonator. We build that resonator as a glass plan-parallel etalon with high-fines. Except the frequency lock, the etalon helps us to measure a wavelength-tuning interval of VCSEL laser during the scanning process. We have stabilized an operating temperature of the VCSEL laser by means of a fast digital temperature controller. The optical set-up of the interferometer begins with a polarizing beam-splitter. It splits the laser beam into the measuring and reference arm of the Michelson interferometer. Two cubic corner cubes reflect beams back to this beam-splitter. It collects reflected beams to the same axis of propagation. Then a detection unit produces the combination of two perpendicularly polarized laser beams with production of two electronic signals that are in the quadrature. A fast analog-to-digital card equipped with the digital signal processor (DSP) samples these signals. DSP also controls the course of the scanning process. After Δλ ~ 1 nm scan of the wavelength of VCSEL laser we obtain a record of passed interference fringes and passed Fabry-Perot resonance modes at the same time. On basis of these measured quantities we are able to calculate with high precision the instantaneous value of the optical path length difference between the measuring and reference arm of the Michelson interferometer. We experimentally compared the developed absolute
Two color far infrared laser interferometer
Kawahata, K.; Akiyama, T.; Pavlichenko, R.; Tanaka, K.; Tokuzawa, T.; Ito, Y.; Okajima, S.; Nakayama, K.; Wood, K.
2006-10-15
Two color interferometer using a short wavelength far infrared laser has been developed for high performance plasmas on large helical device and for future fusion devices such as ITER. High power laser lines simultaneously oscillating at 57.2 and 47.6 {mu}m were achieved in a CO{sub 2}-laser-pumped CH{sub 3}OD laser. By introducing Ge:Ga photoconductive detectors operating at liquid He temperature, we have successfully detected two color beat signals (0.55 and 1.2 MHz) with excellent signal-to-noise ratio ({approx}40 dB). These beat signals were fed into phase comparators for phase measurement after passing through intermediate frequency bandpass filters. Two color far infraned laser interferometer work was successful in the demonstration of mechanical vibration compensation.
Modeling the Laser Interferometer Space Antenna Optics
NASA Technical Reports Server (NTRS)
Waluschka, Eugene; Pedersen, Tracy R.; McNamara, paul
2005-01-01
The Laser Interferometer Space Antenna (LISA), shown below, will detect gravitational waves produced by objects such as binary black holes or objects falling into black holes (extreme mass ratio inspirals) over a frequency range of l0(exp -4) to 0.1 Hz. Within the conceptual frame work of Newtonian physics, a gravitational wave produces a strain, (Delta)l/l, with magnitudes of the order of Earth based gravitational wave detectors, such as the Laser Interferometer Gravitational-Wave Observatory (LIGO) project, use Michelson interferometers with arm lengths l = 4 km to detect these strains. Earth induced seismic noise limits ground-based instruments detecting gravitational waves with frequencies lower than approx. 1 Hz.
CO/sub 2/ laser interferometer array for Big Dee
Fairbanks, E.S.
1985-05-01
A 12 channel interferometer array is planned for obtaining electron density profiles on Big Dee. Each channel consists of coaxial CO/sub 2/ and HeNe laser beams. The reference beam is formed by splitting off half of the laser power at each wavelength by using acousto-optic modulators which introduce a 40-MHz frequency shift in the reference beam. The detected interference signals from room-temperature visible and IR detectors are mixed with the 40-MHz drive for the acousto-optic cells. The quadrature mixer signals result in phase information due to vibration and phase-velocity variation in the plasma. Simultaneous solution of the two equations describing the phase shift at the two wavelengths yields the phase shift due only to the plasma effects. The arithmetic operations will be performed digitally in real time by a dedicated microprocessor in each channel. The execution speed of the microprocessor is expected to be sufficient for measuring line integral density in the presence of vibrations of 1-mm amplitude at 1 Hz or 10-..mu..m amplitude at 100 Hz. Phase resolution for each channel is 2..pi../64 and time between samples is 25 ..mu..s. Density profiles are derived from chordal line integral density data by computer tomography.
CO/sub 2/ laser interferometer array for Big Dee
Fairbanks, E.S.
1984-09-01
A twelve channel interferometer array is planned for obtaining electron density profiles on Big Dee. Each channel consists of coaxial CO/sub 2/ and HeNe laser beams. The reference beam is formed by splitting off half of the laser power at each wavelength by using acousto-optic modulators which introduce a 40 MHz frequency shift in the reference beam. The detected interference signals from room temperature visible and IR detectors are mixed with the 40 MHz drive for the acousto-optic cells. The quadrature mixer signals result in phase information due to vibration and phase velocity variation in the plasma. Simultaneous solution of the two equations describing the phase shift at the two wavelengths yields the phase shift due only to the plasma effects. The arithmetic operations will be performed digitally in real time by a dedicated microprocessor in each channel. The execution speed of the microprocessor is expected to be sufficent for measuring line integral density in the presence of vibrations of 1 mm amplitude at 1 Hz or 10 ..mu..m amplitude at 100 Hz. Phase resolution for each channel is 2..pi../64 and time between samples is 25 ..mu..sec. Density profiles are derived from chordal line integral density data by computer tomography.
Single-shot full-field OCT based on four quadrature phase-stepped interferometer
NASA Astrophysics Data System (ADS)
Hrebesh, Molly Subhash; Watanabe, Yuuki; Dabu, Razvan; Sato, Manabu
2008-02-01
We demonstrate a compact single-shot full-field optical coherence tomography (OCT) system for obtaining real-time high-resolution depth resolved en-face OCT images from weakly scattering specimens. The experimental setup is based on a Linnik type polarization Michelson interferometer and a four-channel compact polarization phase stepper optics. The four-channel phase-stepper optics comprise of a dual channel beam splitter, a Wollaston prism and a pair of wave plate for simultaneously capturing four quadratually phase-stepped images on a single CCD. The interferometer is illuminated using a SLD source with a central wavelength of 842 nm and a bandwidth of 16.2 nm, yielding an axial resolution of 19.8 μm. Using a 10 × (0.25-NA) microscope objective and a CCD camera with 400 × 400 pixels, the system covers an area of 225 μm × 225 μm with a transverse resolution of 4.4 μm. The en-face OCT images of an onion is measured with an exposure time of 7ms and a frame rate of 28 fps.
Measuring Cyclic Error in Laser Heterodyne Interferometers
NASA Technical Reports Server (NTRS)
Ryan, Daniel; Abramovici, Alexander; Zhao, Feng; Dekens, Frank; An, Xin; Azizi, Alireza; Chapsky, Jacob; Halverson, Peter
2010-01-01
An improved method and apparatus have been devised for measuring cyclic errors in the readouts of laser heterodyne interferometers that are configured and operated as displacement gauges. The cyclic errors arise as a consequence of mixing of spurious optical and electrical signals in beam launchers that are subsystems of such interferometers. The conventional approach to measurement of cyclic error involves phase measurements and yields values precise to within about 10 pm over air optical paths at laser wavelengths in the visible and near infrared. The present approach, which involves amplitude measurements instead of phase measurements, yields values precise to about .0.1 microns . about 100 times the precision of the conventional approach. In a displacement gauge of the type of interest here, the laser heterodyne interferometer is used to measure any change in distance along an optical axis between two corner-cube retroreflectors. One of the corner-cube retroreflectors is mounted on a piezoelectric transducer (see figure), which is used to introduce a low-frequency periodic displacement that can be measured by the gauges. The transducer is excited at a frequency of 9 Hz by a triangular waveform to generate a 9-Hz triangular-wave displacement having an amplitude of 25 microns. The displacement gives rise to both amplitude and phase modulation of the heterodyne signals in the gauges. The modulation includes cyclic error components, and the magnitude of the cyclic-error component of the phase modulation is what one needs to measure in order to determine the magnitude of the cyclic displacement error. The precision attainable in the conventional (phase measurement) approach to measuring cyclic error is limited because the phase measurements are af-
Modified Phasemeter for a Heterodyne Laser Interferometer
NASA Technical Reports Server (NTRS)
Loya, Frank M.
2010-01-01
Modifications have been made in the design of instruments of the type described in "Digital Averaging Phasemeter for Heterodyne Interferometry". A phasemeter of this type measures the difference between the phases of the unknown and reference heterodyne signals in a heterodyne laser interferometer. The phasemeter design lacked immunity to drift of the heterodyne frequency, was bandwidth-limited by computer bus architectures then in use, and was resolution-limited by the nature of field-programmable gate arrays (FPGAs) then available. The modifications have overcome these limitations and have afforded additional improvements in accuracy, speed, and modularity. The modifications are summarized.
Modeling the Laser Interferometer Space Antenna Optics
NASA Technical Reports Server (NTRS)
Waluschka, Eugene; Pedersen, Trace R.; McNamara, Paul
2005-01-01
Creating an optical model of the Laser Interferometer Space antenna which can be used to predict optical sensitivities and set tolerances sufficiently well such that picometer level displacements can be reliably seen poses certain challenges. In part, because the distances between key optical elements, the proof masses, are constantly changing, at speeds of meters/second, the separation between them is about 5 million kilometers and a contributing factor to optical jitter is the self-gravity of the spacecraft. A discussion of the current state and future approach(s) to the creation of such an optical model will be presented.
Gravitational Wave Detection with Single-Laser Atom Interferometers
NASA Technical Reports Server (NTRS)
Yu, Nan; Tinto, Massimo
2011-01-01
A new design for a broadband detector of gravitational radiation relies on two atom interferometers separated by a distance L. In this scheme, only one arm and one laser are used for operating the two atom interferometers. The innovation here involves the fact that the atoms in the atom interferometers are not only considered as perfect test masses, but also as highly stable clocks. Atomic coherence is intrinsically stable, and can be many orders of magnitude more stable than a laser.
Pulsed laser interferometry with sub-picometer resolution using quadrature detection.
Shao, Lei; Gorman, Jason J
2016-07-25
Femtosecond pulsed laser interferometry has important applications in measuring picometer-level displacements on sub-nanosecond time scales. In this paper, we experimentally examine its achievable displacement resolution, as well as the relationship between the laser's optical spectrum and the interferometer's effective wavelength. The resulting broadband displacement noise and noise floor of the pulsed laser Michelson interferometer are equivalent to that achieved with a stabilized continuous wave HeNe laser, where values of 1.01 nm RMS and 27.75 fm/√Hz have been demonstrated. It is also shown that a single effective wavelength can accurately describe the fringes of the pulsed laser interferometer but the effective wavelength value can only be determined from the optical spectrum under certain conditions. These results will be used for time-resolved displacement metrology with picosecond temporal resolution in the future. PMID:27464192
Fiber-optic interferometer using frequency-modulated laser diodes
NASA Technical Reports Server (NTRS)
Beheim, G.
1986-01-01
This paper describes an electrically passive fiber-optic interferometer which uses dual frequency-modulated laser diodes. Experimental results show that this type of interferometer can attain a displacement range of 100 micron with subnanometer resolution. This technique can serve as the basis for a number of high-precision fiber-optic sensors.
Interferometer combines laser light source and digital counting system
NASA Technical Reports Server (NTRS)
1965-01-01
Measurement of small linear displacements in digital readouts with extreme accuracy and sensitivity is achieved by an interferometer. The instrument combines a digital electro-optical fringe-counting system and a laser light source.
Submillimeter laser interferometer for high density plasma diagnostic
NASA Astrophysics Data System (ADS)
Kamenev, Yu. E.; Kiselyev, V. K.; Kuleshov, E. M.; Knyaz'kov, B. N.; Kononenko, V. K.; Nesterov, P. K.; Yanovsky, M. S.
1995-06-01
There are presented the results of investigation of the one-channel homodyne laser interferometer λ=119 µm made on the basis of the hollow dielectric beamguide and quasioptical functional devices. The interferometer is designed for determination of the plasma electron density of the TOKAMAK-7. The density response threshold is 0.7% from the expected plasma density and the phase difference measurement total error is 5°
Polarization Considerations for the Laser Interferometer Space Antenna (LISA)
NASA Technical Reports Server (NTRS)
Waluschka, Eugene; Pedersen, Trace R.; McNamara, Paul
2005-01-01
A polarization ray trace model of the Laser Interferometer Space Antenna's (LISA) optical path is being created. The model will be able to assess the effects of various polarizing elements and the optical coatings on the required picometer level interferometry. All of the computational steps are described in detail. This should eliminate any ambiguities associated with polarization ray trace modeling of interferometers and provide a basis for determining its limitations and serve as a clearly defined starting point for future improvements.
Study on avalanche photodiode influence on heterodyne laser interferometer linearity
NASA Astrophysics Data System (ADS)
Budzyn, Grzegorz; Podzorny, Tomasz
2016-06-01
In the paper we analyze factors reducing the possible accuracy of the heterodyne laser interferometers. The analysis is performed for the avalanche-photodiode input stages but is in main points valid also for stages with other type of photodetectors. Instrumental error originating from optical, electronic and digital signal processing factors is taken into consideration. We stress factors which are critical and those which can be neglected at certain accuracy requirements. In the work we prove that it is possible to reduce errors of the laser instrument below 1 nm point for multiaxial APD based interferometers by precise control of incident optical power and the temperature of the photodiode.
Suppressing the mechanical quadrature error of a quartz double-H gyroscope through laser trimming
NASA Astrophysics Data System (ADS)
Zhao, Ke; Feng, Li-Hui; Wang, Qian-Qian; Liu, Ming-Zhi; Wang, Ben-Guo; Cui, Fang; Sun, Yu-Nan
2013-11-01
In this paper, we introduce a z-axis quartz gyroscope using a double-H tuning fork, which has a high sensitivity. However, it also causes a large mechanical quadrature error. The laser trimming method is used to suppress this error at quartz level. The trimming law is obtained through the finite element method (FEM). A femtosecond laser processing system is used to trim the gold balancing masses on the beams, and experimental results are basically consistent with the simulated ones. The mechanical quadrature error is suppressed by 96%, from 26.3° s-1 to 1.1° s-1. Nonlinearity changes from 1.48% to 0.30%, angular random walk (ARW) is reduced from 2.19° h-1/2 to 1.42° h-1/2, and bias instability is improved by a factor of 7.7, from 197.6° h-1 to 25.4° h-1.
Confocal Fabry-Perot interferometer for frequency stabilization of laser
NASA Astrophysics Data System (ADS)
Pan, H.-J.; Ruan, P.; Wang, H.-W.; Li, F.
2011-02-01
The frequency shift of laser source of Doppler lidar is required in the range of a few megahertzs. To satisfy this demand, a confocal Fabry-Perot (F-P) interferometer was manufactured as the frequency standard for frequency stabilization. After analyzing and contrasting the center frequency shift of confocal Fabry-Perot interferometers that are made of three different types of material with the change of temperature, the zerodur material was selected to fabricate the interferometer, and the cavity mirrors were optically contacted onto the end of spacer. The confocal Fabry-Perot interferometer was situated within a double-walled chamber, and the change of temperature in the chamber was less than 0.01 K. The experimental results indicate that the free spectral range is 500 MHz, the full-width at half maximum is 3.33 MHz, and the finesse is 150.
Submillimeter laser interferometer-polarimeter for plasma diagnostics
NASA Astrophysics Data System (ADS)
Kamenev, Yu. E.; Kiselyev, Vladimir K.; Kuleshov, E. M.; Knyaz'kov, B. N.; Kononenko, V. K.; Nesterov, P. K.; Yanovsky, M. S.
1994-08-01
There are presented the results of investigation of the homodyne laser interferometer-polarimeter (lambda) equals 195 micrometers made on the quasioptical element basis and designed for the synchronous determination of the plasma electron density ne and the poloidal magnetic field Bp in 'TOKAMAK' mountings of the thermonuclear fusion.
NASA Astrophysics Data System (ADS)
Mao, Youxin; Flueraru, Costel; Chang, Shoude; Popescu, Dan P.; Sowa, Michael G.
2011-05-01
A performance analysis of signal to noise ratio for an optical coherence tomography system with quadrature detection and a semiconductor optical amplifier in the sample arm is discussed. The results are compared and discussed in relation to a conventional OCT system (without optical amplification). An increase of the signal to noise ratio up to 14 dB at a depth of 0.5 mm is obtained compared to the system without the optical amplifier. Overall, an improvement was demonstrated for signal coming from deeper regions within the samples. Arterial plaque from a myocardial infarction-prone Watanabe heritable hyperlipidemic (WHHLMI) rabbit is visualized and characterized using this system. Improvement of signal to noise ratio increases the penetration depth possible for OCT images, from 1 mm to 2 mm within the vessel wall of an artery. Preliminary results show that vulnerable plaque with fibrous cap, macrophage accumulations and calcification in the arterial tissue are measurable with this OCT system.
Laser Cooled Strontium Source for an Ion Interferometer
NASA Astrophysics Data System (ADS)
Lyon, Mary; Archibald, James; Erickson, Christopher; Durfee, Dallin
2010-10-01
We present a Strontium-87 magneto-optical trap (MOT) in a Low-Velocity-Intense-Source (LVIS) as the source of cooled, collimated atoms for an ion interferometer. Laser cooling and trapping is accomplished with a 461 nm frequency doubled laser and a pair of permanent magnets. A beam of cooled atoms is produced by passing the atoms through a hole drilled in one of the retroreflecting optics. The atoms are then photo-ionized in a two photon process.
Arm Locking for the Laser Interferometer Space Antenna
NASA Technical Reports Server (NTRS)
Maghami, P. G.; Thorpe, J. I.; Livas, J.
2009-01-01
The Laser Interferometer Space Antenna (LISA) mission is a planned gravitational wave detector consisting of three spacecraft in heliocentric orbit. Laser interferometry is used to measure distance fluctuations between test masses aboard each spacecraft to the picometer level over a 5 million kilometer separation. Laser frequency fluctuations must be suppressed in order to meet the measurement requirements. Arm-locking, a technique that uses the constellation of spacecraft as a frequency reference, is a proposed method for stabilizing the laser frequency. We consider the problem of arm-locking using classical optimal control theory and find that our designs satisfy the LISA requirements.
Long-term laser frequency stabilization using fiber interferometers.
Kong, Jia; Lucivero, Vito Giovanni; Jiménez-Martínez, Ricardo; Mitchell, Morgan W
2015-07-01
We report long-term laser frequency stabilization using only the target laser and a pair of 5 m fiber interferometers, one as a frequency reference and the second as a sensitive thermometer to stabilize the frequency reference. When used to stabilize a distributed feedback laser at 795 nm, the frequency Allan deviation at 1000 s drops from 5.6 × 10(-8) to 6.9 × 10(-10). The performance equals that of an offset lock employing a second, atom-stabilized laser in the temperature control. PMID:26233353
Long-term laser frequency stabilization using fiber interferometers
Kong, Jia; Lucivero, Vito Giovanni; Jiménez-Martínez, Ricardo; Mitchell, Morgan W.
2015-07-15
We report long-term laser frequency stabilization using only the target laser and a pair of 5 m fiber interferometers, one as a frequency reference and the second as a sensitive thermometer to stabilize the frequency reference. When used to stabilize a distributed feedback laser at 795 nm, the frequency Allan deviation at 1000 s drops from 5.6 × 10{sup −8} to 6.9 × 10{sup −10}. The performance equals that of an offset lock employing a second, atom-stabilized laser in the temperature control.
Polarization Considerations for the Laser Interferometer Space Antenna
NASA Technical Reports Server (NTRS)
Waluschka, Eugene; Pedersen, Tracy R.; McNamara, Paul
2005-01-01
A polarization ray trace model of the Laser Interferometer Space Antenna s (LISA) optical path is being created. The model will be able to assess the effects of various polarizing elements and the optical coatings on the required, very long path length, picometer level dynamic interferometry. The computational steps are described. This should eliminate any ambiguities associated with polarization ray tracing of interferometers and provide a basis for determining the computer model s limitations and serve as a clearly defined starting point for future work.
Research on beam splitting prism in laser heterodyne interferometer
NASA Astrophysics Data System (ADS)
Fu, Xiu-hua; Xiong, Shi-fu; Kou, Yang; Pan, Yong-gang; Chen, Heng; Li, Zeng-yu; Zhang, Chuan-xin
2014-08-01
With the rapid development of optical testing technology, laser heterodyne interferometer has been used more and more widely. As the testing precision requirements continue to increase, the technical prism is an important component of heterodyne interference. The research utilizing thin film technology to improve optical performance of interferometer has been a new focus. In the article, based on the use requirements of interferometer beam splitting prism, select Ta2O5 and SiO2 as high and low refractive index materials respectively, deposit on substrate K9. With the help of TFCalc design software and Needle method, adopting electron gun evaporation and ion assisted deposition, the beam splitting prism is prepared successfully and the ratio of transmittance and reflectance for this beam splitting prism in 500~850 nm band, incident angle 45 degree is 8:2. After repeated tests, solved the difference problem of film deposition process parameters ,controlled thickness monitoring precision effectively and finally prepared the ideal beam splitting prism which is high adhesion and stable optics properties. The film the laser induced damage threshold and it meet the requirements of heterodyne interferometer for use.
Four-level atom interferometer with trichromatic laser fields
Honda, Kazuhito; Kobayashi, Yoshiyuki; Morinaga, Atsuo
2007-02-15
A four-level atom interferometer comprised of three excited states and one ground state with trichromatic fields coupled between them is investigated using Zeeman sublevels of {sup 3}P{sub 1} and {sup 1}S{sub 0} states of a calcium atom. A theoretical description of the interaction of four-level atoms with trichromatic laser fields is presented and compared with the experimental results of the interference fringes which are generated by the three excited states.
A simple pendulum laser interferometer for determining the gravitational constant
Parks, Harold V.; Faller, James E.
2014-01-01
We present a detailed account of our 2004 experiment to measure the Newtonian constant of gravitation with a suspended laser interferometer. The apparatus consists of two simple pendulums hanging from a common support. Each pendulum has a length of 72 cm and their separation is 34 cm. A mirror is embedded in each pendulum bob, which then in combination form a Fabry–Perot cavity. A laser locked to the cavity measures the change in pendulum separation as the gravitational field is modulated due to the displacement of four 120 kg tungsten masses. PMID:25201994
A compact semiconductor digital interferometer and its applications
NASA Astrophysics Data System (ADS)
Britsky, Oleksander I.; Gorbov, Ivan V.; Petrov, Viacheslav V.; Balagura, Iryna V.
2015-05-01
The possibility of using semiconductor laser interferometers to measure displacements at the nanometer scale was demonstrated. The creation principles of miniature digital Michelson interferometers based on semiconductor lasers were proposed. The advanced processing algorithm for the interferometer quadrature signals was designed. It enabled to reduce restrictions on speed of measured movements. A miniature semiconductor digital Michelson interferometer was developed. Designing of the precision temperature stability system for miniature low-cost semiconductor laser with 0.01ºС accuracy enabled to use it for creation of compact interferometer rather than a helium-neon one. Proper firmware and software was designed for the interferometer signals real-time processing and conversion in to respective shifts. In the result the relative displacement between 0-500 mm was measured with a resolution of better than 1 nm. Advantages and disadvantages of practical use of the compact semiconductor digital interferometer in seismometers for the measurement of shifts were shown.
Fuel droplet size measurements with a laser Doppler interferometer
NASA Astrophysics Data System (ADS)
Taylor, J. R.
1985-07-01
It is pointed out that the injection and atomization of liquid fuel into a gas turbine combustion system plays a major role in many key aspects of combustion system performance. Fuel droplet diameters represent one of the parameters which determine the burning rate. The present investigation is concerned with a commercially available laser Doppler interferometer which has been used by an American manufacturer of aircraft engines to measure fuel droplet size distributions downstream of several different fuel injectors and combinations of fuel injectors and combustor dome swirl cups. The considered instrument has a very small sample volume, which permits measurements of droplet size distributions and droplet Sauter Mean Diameters (SMD) at a large number of discrete points in the spray pattern. The design and the principles of operation of the droplet sizing interferometer (DSI) are discussed along with alignment procedures, test configurations, and test results.
Wang, Lutang; Fang, Nian; Wu, Chunxu; Qin, Haijuan; Huang, Zhaoming
2014-01-01
A novel fiber-optic acoustic sensor using an erbium-doped fiber amplifier (EDFA)-based fiber ring laser and a balanced Sagnac interferometer for acoustic sensing of the partial discharge (PD) in power transformers is proposed and demonstrated. As a technical background, an experimental investigation on how the variations of the fiber birefringence affect the sensor performances was carried out, and the results are discussed. The operation principles are described, and the relevant formulas are derived. The analytical results show that an EDFA-based fiber ring laser operating in chaotic mode can provide a degree of polarization (DOP) tunable light beam for effectively suppressing polarization fading noises. The balanced Sagnac interferometer can eliminate command intensity noises and enhance the signal-to-noise ratio (SNR). Furthermore, it inherently operates at the quadrature point of the response curve without any active stabilizations. Several experiments are conducted for evaluating the performances of the sensor system, as well as for investigating the ability of the detection of high-frequency acoustic emission signals. The experimental results demonstrate that the DOP of the laser beam can be continuously tuned from 0.2% to 100%, and the power fluctuation in the whole DOP tuning range is less than 0.05 dBm. A high-frequency response up to 300 kHz is reached, and the high sensing sensitivity for detections of weak corona discharges, as well as partial discharges also is verified. PMID:24824371
Wang, Lutang; Fang, Nian; Wu, Chunxu; Qin, Haijuan; Huang, Zhaoming
2014-01-01
A novel fiber-optic acoustic sensor using an erbium-doped fiber amplifier (EDFA)-based fiber ring laser and a balanced Sagnac interferometer for acoustic sensing of the partial discharge (PD) in power transformers is proposed and demonstrated. As a technical background, an experimental investigation on how the variations of the fiber birefringence affect the sensor performances was carried out, and the results are discussed. The operation principles are described, and the relevant formulas are derived. The analytical results show that an EDFA-based fiber ring laser operating in chaotic mode can provide a degree of polarization (DOP) tunable light beam for effectively suppressing polarization fading noises. The balanced Sagnac interferometer can eliminate command intensity noises and enhance the signal-to-noise ratio (SNR). Furthermore, it inherently operates at the quadrature point of the response curve without any active stabilizations. Several experiments are conducted for evaluating the performances of the sensor system, as well as for investigating the ability of the detection of high-frequency acoustic emission signals. The experimental results demonstrate that the DOP of the laser beam can be continuously tuned from 0.2% to 100%, and the power fluctuation in the whole DOP tuning range is less than 0.05 dBm. A high-frequency response up to 300 kHz is reached, and the high sensing sensitivity for detections of weak corona discharges, as well as partial discharges also is verified. PMID:24824371
Note: Laser wavelength precision measurement based on a laser synthetic wavelength interferometer.
Yan, Liping; Chen, Benyong; Zhang, Shihua; Liu, Pengpeng; Zhang, Enzheng
2016-08-01
A laser wavelength precision measurement method is presented based on the laser synthetic wavelength interferometer (LSWI). According to the linear relation between the displacements of measurement and reference arms in the interferometer, the synthetic wavelength produced by an unknown wavelength and a reference wavelength can be measured by detecting the phase coincidences of two interference signals. The advantage of the method is that a larger synthetic wavelength resulting from an unknown wavelength very close to the reference wavelength can be easily determined according to the linear relation in the interferometer. Then the unknown wavelength is derived according to the one-to-one corresponding relationship between single wavelength and synthetic wavelength. Wavelengths of an external cavity diode laser and two He-Ne lasers were determined experimentally. The experimental results show that the proposed method is able to realize a relative uncertainty on the order of 10(-8). PMID:27587172
Note: Laser wavelength precision measurement based on a laser synthetic wavelength interferometer
NASA Astrophysics Data System (ADS)
Yan, Liping; Chen, Benyong; Zhang, Shihua; Liu, Pengpeng; Zhang, Enzheng
2016-08-01
A laser wavelength precision measurement method is presented based on the laser synthetic wavelength interferometer (LSWI). According to the linear relation between the displacements of measurement and reference arms in the interferometer, the synthetic wavelength produced by an unknown wavelength and a reference wavelength can be measured by detecting the phase coincidences of two interference signals. The advantage of the method is that a larger synthetic wavelength resulting from an unknown wavelength very close to the reference wavelength can be easily determined according to the linear relation in the interferometer. Then the unknown wavelength is derived according to the one-to-one corresponding relationship between single wavelength and synthetic wavelength. Wavelengths of an external cavity diode laser and two He-Ne lasers were determined experimentally. The experimental results show that the proposed method is able to realize a relative uncertainty on the order of 10-8.
Detection of volcanic infrasound with a ring laser interferometer
NASA Astrophysics Data System (ADS)
Dunn, Robert W.; Hosman, Ashley R.
2014-11-01
Over the last 15 years, large horizontally mounted ring lasers have been used to study numerous geophysical phenomena. This paper provides examples of the sensitivity of large active ring laser interferometers to far field infrasound emissions from explosive volcanic eruptions. Volcanic infrasound is reported from representative eruptions of volcanoes Kelut (Kelud), Klyuchevskaya (Kliuchevskoi), Puyehua, Santa Maria, Sakurajima, and Tungurahua. The detected infrasound frequencies are in basic agreement with the far field air wave frequencies from the 1980 eruption of Mount St. Helens and the 1991 eruption of Mount Pinatubo. Cavities of large horizontally mounted active ring lasers contain two counter-propagating waves that in the presence Earth's rotation become traveling waves of slightly different frequencies. The Sagnac or beat frequency due to the difference in the traveling wave frequencies is modulated by geophysical phenomena, in this case volcanic infrasound. Signatures of the infrasound are found in the frequency modulated side bands.
Explosive component acceptance tester using laser interferometer technology
NASA Technical Reports Server (NTRS)
Wickstrom, Richard D.; Tarbell, William W.
1993-01-01
Acceptance testing of explosive components requires a reliable and simple to use testing method that can discern less than optimal performance. For hot-wire detonators, traditional techniques use dent blocks or photographic diagnostic methods. More complicated approaches are avoided because of their inherent problems with setup and maintenance. A recently developed tester is based on using a laser interferometer to measure the velocity of flying plates accelerated by explosively actuated detonators. Unlike ordinary interferometers that monitor displacement of the test article, this device measures velocity directly and is commonly used with non-spectral surfaces. Most often referred to as the VISAR technique (Velocity Interferometer System for Any Reflecting Surface), it has become the most widely-accepted choice for accurate measurement of velocity in the range greater than 1 mm/micro-s. Traditional VISAR devices require extensive setup and adjustment and therefore are unacceptable in a production-testing environment. This paper describes a new VISAR approach which requires virtually no adjustments, yet provides data with accuracy comparable to the more complicated systems. The device, termed the Fixed-Cavity VISAR, is currently being developed to serve as a product verification tool for hot-wire detonators and slappers. An extensive data acquisition and analysis computer code was also created to automate the manipulation of raw data into final results.
The status of laser interferometer gravitational-wave detectors
NASA Astrophysics Data System (ADS)
Raab, F. J.; Ligo Scientific Collaboration
2006-05-01
There has been a rapid advance in the sensitivity of broadband searches for gravitational waves, using an international network of kilometer-scale laser interferometers. The LIGO detectors in North America, the GEO600 detector in Germany and the TAMA300 detector in Japan have conducted searches for gravitational waves covering a frequency range from below 100 Hz up to many kHz. These detectors and the VIRGO detector in Italy are in a mature state of commissioning and technology development for a generation of more advanced detectors is ongoing.
Design of laser system for absolute gravimeter based on 87Rb atom interferometer
NASA Astrophysics Data System (ADS)
Zhao, Yang; Wang, Shaokai; Zhuang, Wei; Fang, Fang; Li, Tianchu
2015-08-01
We present a laser system design for an absolute gravimeter based on 87Rb atom interferometer. By skillful design, lasers with 9 different frequencies are based on two diode lasers including tapered amplifier. Two electrical feedback systems are used for laser frequency stabilization and the Raman lasers generation respectively. All other lasers are based on two Raman lasers and realized with frequency shift by acoustic optical modulators. This laser system not only has the compact and simple construction, but meets all requirements for laser power and frequency controlling for the atom interferometer. It has the characteristic of reliability and integrity.
Trikshev, A I; Kurkov, Andrei S; Tsvetkov, V B; Pyrkov, Yu N; Paramonov, V N
2011-07-31
A simple scanning interferometer is implemented for measuring the emission linewidth of single-frequency semiconductor lasers. The free dispersion region of the interferometer is 28 MHz, the spectral resolution being 470 kHz. (laser spectroscopy)
The laser interferometer system for the large optics diamond turning machine
Baird, E D; Donaldson, R R; Patterson, S R
1999-06-29
The purpose of this report is to describe the Laser Interferometer System designed for the Large Optics Diamond Turning Machine (LODTM). To better understand the laser interferometer system, it is useful to begin with an overview of the LODTM metrology system.
Path-length measurement performance evaluation of polarizing laser interferometer prototype
NASA Astrophysics Data System (ADS)
Li, Yu-qiong; Luo, Zi-ren; Liu, He-shan; Dong, Yu-hui; Jin, Gang
2015-02-01
The space laser interferometer has been considered the most promising means for detecting gravitational waves and improving the accuracy and spatial resolution of the Earth's gravity model. An on-ground polarizing laser interferometer prototype equipped with one reference interferometer and two measurement interferometers having equal-length arms is presented in the paper. The laser interferometer prototype is designed as the demonstration of a Chinese space laser interferometer antenna in the future, of which the path-length measurement performance evaluation and preliminary noise analysis are investigated here. The results show that the path-length measurement sensitivity is better than 200 pm/Hz½ in the frequency band of 10 mHz-1 Hz, and the sensitivity of measuring the motion of a sinusoidally driven testmass is better than 100 pm within the frequency regime of 1 mHz-1 Hz. In this way, laboratory activities have demonstrated the feasibility of this prototype to measure tiny path-length fluctuations of the simulated testmass. As a next step, adopting an integrated design of optics and optical substrate to enhance the stability of the laser interferometer is being planned, and other key techniques included in the space laser interferometer such as laser pointing modulation and laser phase-locking control are to be implanted into this prototype are under consideration.
Analytic and interferometric techniques for the Laser Interferometer Space Antenna
NASA Astrophysics Data System (ADS)
Pollack, Scott E.
The Laser Interferometer Space Antenna (LISA) is being designed to detect and study in detail gravitational waves from sources throughout the Universe such as massive black holes. The conceptual formulation of the LISA space-borne gravitational wave detector is now well developed. The interferometric measurements between the sciencecraft remain one of the most important technological and scientific design areas for the mission. Our work has concentrated on developing the interferometric technologies to create a LISA-like optical signal and to measure the phase of that signal using commercially available instruments. One of the most important goals of this research is to demonstrate the LISA phase timing and phase reconstruction for a LISA-like fringe signal, in the case of a high fringe rate and a low signal level. To this end we have constructed a table-top interferometer which produces LISA-like fringe signals. Over the past few years questions have been raised concerning the use of laser communications links between sciencecraft to transmit phase information crucial to the reduction of laser frequency noise in the LISA science measurement. The concern is that applying medium frequency phase modulations to the laser carrier could compromise the phase stability of the LISA fringe signal. We have modified our table-top interferometer by applying a phase modulation to the laser beam in order to evaluate the effects of such modulations on the LISA science fringe signal. We have demonstrated that the phase resolution of the science signal is not degraded by the presence of medium frequency phase modulations. Each spacecraft in LISA houses a proof mass which follows a geodesic through space. Disturbances that change the proof mass position, momentum, and acceleration will appear in the LISA data stream as additive quadratic functions. These data disturbances inhibit signal extraction and must be removed. Much of our analytical work has been focused on discussing the
Comb-referenced laser distance interferometer for industrial nanotechnology
Jang, Yoon-Soo; Wang, Guochao; Hyun, Sangwon; Kang, Hyun Jay; Chun, Byung Jae; Kim, Young-Jin; Kim, Seung-Woo
2016-01-01
A prototype laser distance interferometer is demonstrated by incorporating the frequency comb of a femtosecond laser for mass-production of optoelectronic devices such as flat panel displays and solar cell devices. This comb-referenced interferometer uses four different wavelengths simultaneously to enable absolute distance measurement with the capability of comprehensive evaluation of the measurement stability and uncertainty. The measurement result reveals that the stability reaches 3.4 nm for a 3.8 m distance at 1.0 s averaging, which further reduces to 0.57 nm at 100 s averaging with a fractional stability of 1.5 × 10−10. The uncertainty is estimated to be in a 10−8 level when distance is measured in air due to the inevitable ambiguity in estimating the refractive index, but it can be enhanced to a 10−10 level in vacuum. PMID:27558016
Comb-referenced laser distance interferometer for industrial nanotechnology.
Jang, Yoon-Soo; Wang, Guochao; Hyun, Sangwon; Kang, Hyun Jay; Chun, Byung Jae; Kim, Young-Jin; Kim, Seung-Woo
2016-01-01
A prototype laser distance interferometer is demonstrated by incorporating the frequency comb of a femtosecond laser for mass-production of optoelectronic devices such as flat panel displays and solar cell devices. This comb-referenced interferometer uses four different wavelengths simultaneously to enable absolute distance measurement with the capability of comprehensive evaluation of the measurement stability and uncertainty. The measurement result reveals that the stability reaches 3.4 nm for a 3.8 m distance at 1.0 s averaging, which further reduces to 0.57 nm at 100 s averaging with a fractional stability of 1.5 × 10(-10). The uncertainty is estimated to be in a 10(-8) level when distance is measured in air due to the inevitable ambiguity in estimating the refractive index, but it can be enhanced to a 10(-10) level in vacuum. PMID:27558016
Orbit design for the Laser Interferometer Space Antenna (LISA)
NASA Astrophysics Data System (ADS)
Xia, Yan; Li, Guangyu; Heinzel, Gerhard; Rüdiger, Albrecht; Luo, Yongjie
2010-01-01
The Laser Interferometer Space Antenna (LISA) is a joint ESA-NASA mission for detecting low-frequency gravitational waves in the frequency range from 0.1 mHz to 1 Hz, by using accurate laser interferometry between three spacecrafts, which will be launched around 2018 and one year later reach their operational orbits around the Sun. In order to operate successfully, it is crucial for the constellation of the three spacecrafts to have extremely high stability. Based on the study of operational orbits for a 2015 launch, we design the operational orbits of beginning epoch on 2019-03-01, and introduce the method of orbit design and optimization. We design the orbits of the transfer from Earth to the operational orbits, including launch phase and separation phase; furthermore, the relationship between energy requirement and flight time of these two orbit phases is investigated. Finally, an example of the whole orbit design is presented.
Selection of linear-cavity fibre laser radiation using a reflection interferometer
Terentyev, V S; Simonov, V A
2013-08-31
We consider the use of a two-mirror multibeam reflection interferometer as a selector of linear-cavity single-mode fibre laser radiation and present experimental data on continuous wavelength tuning of an erbium-doped fibre laser. Conditions are found for single-longitudinal-mode operation of the fibre laser cavity using a reflection interferometer, with the possibility of broadband wavelength tuning. (control of laser pulse parameters)
Chen Benyong; Zhang Enzheng; Yan Liping; Li Chaorong; Tang Wuhua; Feng Qibo
2009-11-15
Not only the magnitude but also the position of straightness errors are of concern to users. However, current laser interferometers used for measuring straightness seldom give the relative position of the straightness error. To solve this problem, a laser interferometer for measuring straightness and its position based on heterodyne interferometry is proposed. The optical configuration of the interferometer is designed and the measurement principle is analyzed theoretically. Two experiments were carried out. The first experiment verifies the validity and repeatability of the interferometer by measuring a linear stage. Also, the second one for measuring a flexure-hinge stage demonstrates that the interferometer is capable of nanometer measurement accuracy. These results show that this interferometer has advantages of simultaneously measuring straightness error and the relative position with high precision, and a compact structure.
Digital Phase Meter for a Laser Heterodyne Interferometer
NASA Technical Reports Server (NTRS)
Loya, Frank
2008-01-01
The Digital Phase Meter is based on a modified phase-locked loop. When phase alignment between the reference input and the phase-shifted metrological input is achieved, the loop locks and the phase shift of the digital phase shifter equals the phase difference that one seeks to measure. This digital phase meter is being developed for incorporation into a laser heterodyne interferometer in a metrological apparatus, but could also be adapted to other uses. Relative to prior phase meters of similar capability, including digital ones, this digital phase meter is smaller, less complex, and less expensive. The phase meter has been constructed and tested in the form of a field-programmable gate array (FPGA).
The GRACE Follow-On Laser Ranging Interferometer
NASA Astrophysics Data System (ADS)
Müller, Vitali
2016-07-01
The GRACE Follow-On mission consists of a pair of satellites to be launched in 2017 into a low-Earth polar orbit. As the precursor mission GRACE, it will provide monthly global maps of Earth's gravity field to study mass changes within the System Earth, like glacier melting or ground-water depletion. The new mission will be equipped with two ranging instruments: a conventional Microwave Ranging Instrument, as already present in the precursor mission, and with a Laser Ranging Interferometer (LRI). Latter acts as a technical demonstrator, which will show the capability for enhanced sensitivity and additional precise attitude information of this new technology. The satellite and in particular the LRI working principle will be introduced together with observables and major noise and error contributors. Furthermore potential modifications and extensions for future gravimetric missions are addressed as well as applications in space-based gravitational wave detectors (i.e. eLISA).
Thermal Noise in Laser Interferometer Gravitational Wave Detectors
NASA Astrophysics Data System (ADS)
Flaminio, Raffaele
Thermal noise is one of the major limitations to the sensitivity of present and future laser interferometers devoted to gravitational wave detection. According to the fluctuation-dissipation theorem any mechanical oscillator is affected by a motion of thermal origin directly related to its thermodynamic temperature. The mirrors and their suspensions that are used in gravitational wave detectors such as Virgo or LIGO are examples of such mechanical oscillators. As a consequence their position is affected by this thermal vibration and the sensitivity of the gravitational wave detector is thermal noise limited over a wide range of frequencies. After recalling briefly the fluctuation-dissipation theorem and its origins, this chapter describes the main types of thermal noise affecting gravitational wave detectors. In the last part of the chapter a special emphasis is given to the thermal noise due to dissipation in the mirrors optical coatings.
Listening to the Universe with the Laser Interferometer Space Antenna
NASA Astrophysics Data System (ADS)
Cornish, Neil J.; Crowder, J.; Porter, E.
2006-12-01
Gravitational Wave observatories here on Earth are poised to end the "silent era" of Astronomy by measuring the vibrations in spacetime caused by colliding black holes and other energetic events. NASA and ESA are working together to open up the source-rich low frequency portion of the gravitational wave spectrum with the Laser Interferometer Space Antenna (LISA). A major challenge for this mission is how to deal with a data stream that is expected to contain the signals from tens of thousands of sources, including massive black hole binaries, stellar remant captures by galactic black holes, and a foreground of compact binaries in our own galaxy. Here we describe a fast and robust technique for simultaneously resolving tens of thousands of overlapping signals and recovering the source parameters. Our approach has been sucessfully tested on simulated LISA data, including the blind data sets of the Mock LISA Data Challenge.
LIGO - The Laser Interferometer Gravitational-Wave Observatory
NASA Technical Reports Server (NTRS)
Abramovici, Alex; Althouse, William E.; Drever, Ronald W. P.; Gursel, Yekta; Kawamura, Seiji; Raab, Frederick J.; Shoemaker, David; Sievers, Lisa; Spero, Robert E.; Thorne, Kip S.
1992-01-01
The goal of the Laser Interferometer Gravitational-Wave Observatory (LIGO) Project is to detect and study astrophysical gravitational waves and use data from them for research in physics and astronomy. LIGO will support studies concerning the nature and nonlinear dynamics for gravity, the structures of black holes, and the equation of state of nuclear matter. It will also measure the masses, birth rates, collisions, and distributions of black holes and neutron stars in the universe and probe the cores of supernovae and the very early universe. The technology for LIGO has been developed during the past 20 years. Construction will begin in 1992, and under the present schedule, LIGO's gravitational-wave searches will begin in 1998.
Multicomponent wavefield characterization with a novel scanning laser interferometer.
Blum, Thomas E; van Wijk, Kasper; Pouet, Bruno; Wartelle, Alexis
2010-07-01
The in-plane component of the wavefield provides valuable information about media properties from seismology to nondestructive testing. A new compact scanning laser ultrasonic interferometer collects light scattered away from the angle of incidence to provide the absolute ultrasonic displacement for both the out-of-plane and an in-plane components. This new system is tested by measuring the radial and vertical polarization of a Rayleigh wave in an aluminum half-space. The estimated amplitude ratio of the horizontal and vertical displacement agrees well with the theoretical value. The phase difference exhibits a small bias between the two components due to a slightly different frequency response between the two processing channels of the prototype electronic circuitry. PMID:20687699
Multicomponent wavefield characterization with a novel scanning laser interferometer
Blum, Thomas E.; Wijk, Kasper van; Pouet, Bruno; Wartelle, Alexis
2010-07-15
The in-plane component of the wavefield provides valuable information about media properties from seismology to nondestructive testing. A new compact scanning laser ultrasonic interferometer collects light scattered away from the angle of incidence to provide the absolute ultrasonic displacement for both the out-of-plane and an in-plane components. This new system is tested by measuring the radial and vertical polarization of a Rayleigh wave in an aluminum half-space. The estimated amplitude ratio of the horizontal and vertical displacement agrees well with the theoretical value. The phase difference exhibits a small bias between the two components due to a slightly different frequency response between the two processing channels of the prototype electronic circuitry.
Differential interferometer for measurement of displacement of laser resonator mirrors
NASA Astrophysics Data System (ADS)
Macúchová, Karolina; Němcová, Šárka; Hošek, Jan
2015-01-01
This paper covers a description and a technique of a possible optical method of mode locking within a laser resonator. The measurement system is a part of instrumentation of laser-based experiment OSQAR at CERN. The OSQAR experiment aims at search of axions, axion-like particles and measuring of ultra-fine vacuum magnetic birefringence. It uses a laser resonator to enhance the coupling constant of hypothetical photon-to-axion conversion. The developed locking-in technique is based on differential interferometry. Signal obtained from the measurement provide crucial information for adaptive control of the locking-in of the resonator in real time. In this paper we propose several optical setups used for measurement and analysis of mutual position of the resonator mirrors. We have set up a differential interferometer under our laboratory conditions. We have done measurements with hemi-spherical cavity resonator detuned with piezo crystals. The measurement was set up in a single plane. Laser light was directed through half-wave retarder to a polarizing beam splitter and then converted to circular polarization by lambda/4 plates. After reflection at the mirrors, the beam is recombined in a beam splitter, sent to analyser and non-polarizing beam splitter and then inspected by two detectors with mutually perpendicular polarizers. The 90 degrees phase shift between the two arms allows precise analysis of a mutual distance change of the mirrors. Because our setup was sufficiently stable, we were able to measure the piezo constant and piezo hysteresis. The final goal is to adapt the first prototype to 23 m resonator and measure the displacement in two planes.
NASA Astrophysics Data System (ADS)
Zhu, Minhao; Wei, Haoyun; Li, Yan
2014-07-01
Periodic error is a main error source that limits the measurement accuracy in heterodyne laser interferometry. An external cavity diode laser (ECDL) based Fabry-Perot (F-P) interferometer referenced to an optical frequency comb (OFC) is proposed to characterize the periodic error in heterodyne interferometers. The Pound-Drever-Hall locking technique is employed to lock the tracking ECDL frequency to the resonance of a high finesse F-P cavity. The frequency of a reference ECDL is locked to a selected mode of an OFC to generate a stable single optical frequency. The frequency change of the tracking ECDL induced by the cavity displacement is measured by beating with the reference ECDL locked to the OFC. Experiments show that the F-P interferometer system has a displacement resolution of 1.96 pm. We compared the measurement results of our system with a commercial plane mirror heterodyne interferometer. The period if the periodic error is about half wavelength, with an error amplitude of 4.8 nm.
Laser beam collimation using a phase conjugate Twyman-Green interferometer
NASA Technical Reports Server (NTRS)
Shukla, R. P.; Dokhanian, M.; George, M. C.; Venkateswarlu, Putcha
1991-01-01
This paper presents an improved technique for testing laser beam collimation using a phase conjugate Twyman-Green interferometer. The technique is useful for measuring laser beam divergence. It is possible using this technique to detect the defocusing of the order of one micrometer for a well corrected collimating lens. A relation is derived for the defocusing that can be detected by the phase conjugate interferometer.
Laser Interferometer Space Antenna (LISA) Far Field Phase Patterns
NASA Technical Reports Server (NTRS)
Waluschka, Eugene; Obenschain, Arthur F. (Technical Monitor)
2000-01-01
The Laser Interferometer Space Antenna (LISA) consists of three spacecraft in orbit about the sun. The orbits are chosen such that the three spacecraft are always at (roughly) the vertices of a equilateral triangle with 5 million kilometer leg lengths. Even though the distances between the three spacecraft are 5 million kilometers, the expected phase shifts between any two beams, due to a gravitational wave, only correspond to a distance change of about 10 pico meters, which is about 10(exp -5) waves for a laser wavelength of 1064 nm. To obtain the best signal-to-noise ratio, noise sources such as changes in the apparent distances due to pointing jitter must be controlled carefully. This is the main reason for determining the far-field phase patterns of a LISA type telescope. Because of torque on the LISA spacecraft and other disturbances, continuous adjustments to the pointing of the telescopes are required. These pointing adjustments will be a "jitter" source. If the transmitted wave is perfectly spherical then rotations (Jitter) about its geometric center will not produce any effect at the receiving spacecraft. However, if the outgoing wave is not perfectly spherical, then pointing jitter will produce a phase variation at the receiving spacecraft. The following sections describe the "brute force" computational approach used to determine the scalar wave front as a function of exit pupil (Zernike) aberrations and to show the results (mostly graphically) of the computations. This approach is straightforward and produces believable phase variations to sub-pico meter accuracy over distances on the order of 5 million kilometers. As such this analyzes the far field phase sensitivity to exit pupil aberrations.
Ramette, Joshua; Kasprzack, Marie; Brooks, Aidan; Blair, Carl; Wang, Haoyu; Heintze, Matthew
2016-04-01
Advanced laser interferometer gravitational-wave detectors use high laser power to achieve design sensitivity. A small part of this power is absorbed in the interferometer cavity mirrors where it creates thermal lenses, causing aberrations in the main laser beam that must be minimized by the actuation of "ring heaters," which are additional heater elements that are aimed to reduce the temperature gradients in the mirrors. In this article we derive the first, to the best of our knowledge, analytical model of the temperature field generated by an ideal ring heater. We express the resulting optical aberration contribution to the main laser beam in this axisymmetric case. Used in conjunction with wavefront measurements, our model provides a more complete understanding of the thermal state of the cavity mirrors and will allow a more efficient use of the ring heaters in the Advanced Laser Interferometer Gravitational-wave Observatory. PMID:27139664
An extreme ultraviolet Michelson interferometer for experiments at free-electron lasers
Hilbert, Vinzenz; Fuchs, Silvio; Paulus, Gerhard G.; Zastrau, Ulf; Blinne, Alexander; Feigl, Torsten; Kämpfer, Tino; Rödel, Christian; Uschmann, Ingo; Wünsche, Martin; Förster, Eckhart
2013-09-15
We present a Michelson interferometer for 13.5 nm soft x-ray radiation. It is characterized in a proof-of-principle experiment using synchrotron radiation, where the temporal coherence is measured to be 13 fs. The curvature of the thin-film beam splitter membrane is derived from the observed fringe pattern. The applicability of this Michelson interferometer at intense free-electron lasers is investigated, particularly with respect to radiation damage. This study highlights the potential role of such Michelson interferometers in solid density plasma investigations using, for instance, extreme soft x-ray free-electron lasers. A setup using the Michelson interferometer for pseudo-Nomarski-interferometry is proposed.
Detection of low frequency hurricane emissions using a ring laser interferometer
NASA Astrophysics Data System (ADS)
Dunn, Robert W.; Slaton, William V.; Kendall, Lauren M.
2012-10-01
Over the last decade, large horizontally mounted ring laser interferometers have demonstrated the capacity to measure numerous geophysical effects. In this paper, responses from large ring laser interferometers to low frequency hurricane emissions are presented. Hurricanes create a broad spectrum of noise that extends into the millihertz range. In addition to microseisms, hurricanes with established eyewalls were found to create distinct frequency peaks close to 7 mHz as they came ashore or moved over shallow water. Selected emissions from Hurricanes Katrina, Wilma, and Dean are presented. The exact coupling mechanism between the ˜7 mHz hurricane emissions and the ring lasers remains under active investigation.
Development of CO{sub 2} laser dispersion interferometer with photoelastic modulator
Akiyama, T.; Kawahata, K.; Okajima, S.; Nakayama, K.
2010-10-15
A dispersion interferometer is one of the promising methods of the electron density measurement on large and high density fusion devices. This paper describes development of a CO{sub 2} laser dispersion interferometer with a photoelastic modulator for phase modulation. In order to make the dispersion interferometer free from variations of the detected intensity, a new phase extraction method is introduced: The phase shift is evaluated from a ratio of amplitudes of the fundamental and the second harmonics of the phase modulation frequency in the detected interference signal. The proof-of-principle experiments demonstrate the feasibility of this method.
Cancellation of Laser Noise in an Unequal-arm Interferometer Detector of Gravitational Radiation
NASA Technical Reports Server (NTRS)
Tinto, M.; Armstrong, J. W.
1998-01-01
In this paper we present a method for exactly cancelling the laser noise in a one-bounce unequal-arm Michelson interferometer. The method requries separate measurements of the phase difference in each arm, made by interfering the returning laser light in each arm with the outgoing light.
NASA Technical Reports Server (NTRS)
Tinto, M.
1998-01-01
Searches for gravitational radiation can be performed in space with two spacecraft tracking each other with coherent laser light. This experimental technique could be implemented with two spacecraft carrying an appropriate optical payload, or with the proposed broad-band, space-based laser interferometer detectors of gravitational waves operated in this non-interferometric mode.
The Laser Interferometer Gravitational-Wave Observatory: Lasers at the Frontiers of Astrophysics
NASA Astrophysics Data System (ADS)
Reitze, David
2005-04-01
The Laser Interferometric Gravitational-Wave Observatory (LIGO) is poised to open a new window on the universe - the detection of gravitational waves from distant large-scale astrophysical sources. Gravitational waves were predicted by Einstein almost 90 years ago but never been observed directly despite a number of experiments over the last 40 years. While there exists strong indirect evidence for gravitational waves, it is only with the construction of large-scale high precision interferometers that direct detection of gravitational waves is possible. Gravitational waves are miniscule dynamic strains applied to space-time by motion of massive astrophysical objects. A passing gravitational wave will expand and contract the distance between two mirrors (`test masses') in the arms of an interferometer. Direct observation of gravitational waves presents a formidable challenge, because the magnitude of the dynamic strain is expected to be infinitesimal, less than one part in 10-22. The astrophysical motivation for detecting gravitational waves is compelling. Unlike the visible sky, the gravitational wave `sky' is completely unexplored. The LIGO detectors and its partner GEO600 in Europe have the sensitivity to observe gravitational waves not only in our own galaxy, but in neighboring galaxies, thus opening an absolutely unique window into these phenomena. In the first part of the presentation, we will give an overview of gravitational waves - what they are and where they come from -- and describe in general terms the techniques that gravitational wave astrophysicists use to hunt for them. In the second part of the presentation, we describe the LIGO interferometers emphasizing the critical role that lasers and optics play in its operation.
Cancellation of Laser Noise in Space-Based Interferometer Detectors of Gravitational Radiation
NASA Technical Reports Server (NTRS)
Tinto, Massimo
1999-01-01
We presented a time-domain procedure for accurately cancelling laser noise fluctuations in an unequal-arm Michelson interferometer. The method involves separately measuring the phase of the returning light relative to the phase of the transmitted light in each arm. By suitable offsetting and differencing of these two time series, the common laser noise is cancelled exactly. The technique presented in this paper is general, in such that it can be implemented with any (Earth as well as space-based) unequal-arms Michelson interferometers,
Interspacecraft link simulator for the laser ranging interferometer onboard GRACE Follow-On.
Sanjuan, Josep; Gohlke, Martin; Rasch, Stefan; Abich, Klaus; Görth, Alexander; Heinzel, Gerhard; Braxmaier, Claus
2015-08-01
Link acquisition strategies are key aspects for interspacecraft laser interferometers. We present an optical fiber-based setup able to simulate the interspacecraft link for the laser ranging interferometer (LRI) on gravity recovery and climate experiment Follow-On. It allows one to accurately recreate the far-field intensity profile depending on the mispointing between the spacecraft, Doppler shifts, and spacecraft attitude jitter. Furthermore, it can be used in late integration stages of the mission, since no physical contact with the spacecraft is required. The setup can also be easily adapted to other similar missions and different acquisition algorithms. PMID:26368080
Measuring of object vibration using sinusoidal-modulation laser-diode active interferometer
NASA Astrophysics Data System (ADS)
Ai, Yong; Cao, Qinfeng; Lu, Su
1996-09-01
Using the character that the emitting optical frequency of the laser diode is controlled by the injected current, the ability of eliminating environmental disturbance of the sinusoidal modulation laser diode active interferometer will be raised by more than one hundred times through putting the disturbed interference signal produced by the environment into the interferometer. When vibrating frequency of objects is different from that of the sinusoidol modulation, 'beat- frequency' will be produced in the interfere signal, which can be analyzed to get the vibrating frequency of objects. This paper described the operation principle and theoretical delusion of the 'beat-frequency' method.
A laser interferometer for measuring skin friction in three-dimensional flows
NASA Technical Reports Server (NTRS)
Monson, D. J.
1983-01-01
A new, nonintrusive method is described for measuring skin friction in three-dimensional flows with unknown direction. The method uses a laser interferometer to measure the changing slope of a thin oil film applied to a surface experiencing shear stress. The details of the method are described, and skin friction measurements taken in a swirling three-dimensional boundary-layer flow are presented. Comparisons between analytical results and experimental values from the laser interferometer method and from a bidirectional surface-fence gauge are made.
A phase-modulated laser system of ultra-low phase noise for compact atom interferometers
NASA Astrophysics Data System (ADS)
Lee, Ki-Se; Kim, Jaewan; Lee, Sang-Bum; Park, Sang Eon; Kwon, Taek Yong
2015-07-01
A compact and robust laser system is essential for mobile atom interferometers. Phase modulation can provide the two necessary phase-coherent frequencies without sophisticated phase-locking between two different lasers. However, the additional laser frequencies generated can perturb the atom interferometer. In this article, we report on a novel method to produce a single high-power laser beam composed of two phase-coherent sidebands without the perturbing carrier mode. Light from a diode laser is phase-modulated by using a fiber-coupled electro-optic modulator driven at 3.4 GHz and passes through a Fabry-Perot cavity with a 6.8 GHz free spectral range. The cavity filters the carrier mode to leave the two first-order sidebands for the two-photon Raman transition between the two hyperfine ground states of 87Rb. The laser beam is then fed to a single tapered amplifier, and the two sidebands are both amplified without mode competition. The phase noise is lower than that of a state-of-the-art optically phase-locked external-cavity diode laser (-135 dBrad2/Hz at 10 kHz) at frequencies above 10 Hz. This technique can be used in all-fiber-based laser systems for future mobile atom interferometers.
Laser-ranging long-baseline differential atom interferometers for space
NASA Astrophysics Data System (ADS)
Chiow, Sheng-wey; Williams, Jason; Yu, Nan
2015-12-01
High-sensitivity differential atom interferometers (AIs) are promising for precision measurements in science frontiers in space, including gravity-field mapping for Earth science studies and gravitational wave detection. Difficulties associated with implementing long-baseline differential AIs have previously included the need for a high optical power, large differential Doppler shifts, and narrow dynamic range. We propose a configuration of twin AIs connected by a laser-ranging interferometer (LRI-AI) to provide precise information of the displacements between the two AI reference mirrors and also to phase-lock the two independent interferometer lasers over long distances, thereby drastically improving the practical feasibility of long-baseline differential AI measurements. We show that a properly implemented LRI-AI can achieve equivalent functionality to the conventional differential AI measurement configuration.
NASA Astrophysics Data System (ADS)
Yan, Hao; Duan, Hui-Zong; Li, Lin-Tao; Liang, Yu-Rong; Luo, Jun; Yeh, Hsien-Chi
2015-12-01
Picometer laser interferometry is an essential tool for ultra-precision measurements in frontier scientific research and advanced manufacturing. In this paper, we present a dual-heterodyne laser interferometer for simultaneously measuring linear and angular displacements with resolutions of picometer and nanoradian, respectively. The phase measurement method is based on cross-correlation analysis and realized by a PXI-bus data acquisition system. By implementing a dual-heterodyne interferometer with a highly symmetric optical configuration, low frequency noises caused by the environmental fluctuations can be suppressed to very low levels via common-mode noise rejection. Experimental results for the dual-heterodyne interferometer configuration presented demonstrate that the noise levels of the linear and angular displacement measurements are approximately 1 pm/Hz1/2 and 0.5 nrad/Hz1/2 at 1 Hz.
NASA Astrophysics Data System (ADS)
Libbrecht, Kenneth G.; Black, Eric D.
2015-05-01
We describe a basic Michelson laser interferometer experiment for the undergraduate teaching laboratory that achieves picometer sensitivity in a hands-on, table-top instrument. In addition to providing an introduction to interferometer physics and optical hardware, the experiment also focuses on precision measurement techniques including servo control, signal modulation, phase-sensitive detection, and different types of signal averaging. Students examine these techniques in a series of steps that take them from micron-scale sensitivity using direct fringe counting to picometer sensitivity using a modulated signal and phase-sensitive signal averaging. After students assemble, align, and characterize the interferometer, they then use it to measure nanoscale motions of a simple harmonic oscillator system as a substantive example of how laser interferometry can be used as an effective tool in experimental science.
NASA Astrophysics Data System (ADS)
Ristori, Pablo; Otero, Lidia; Jin, Yoshitaka; Sugimoto, Nobuo; Nishizawa, Tomoaki; Quel, Eduardo
2016-06-01
A High Spectral Resolution Lidar (HSRL) using an unseeded laser is designed to separate Mie from total atmospheric backscatter by means of a tunable interferometer. The separation is achieved by tuning the interferometer' s free spectral range (FSR) to longitudinal mode separation of the laser. When this condition is achieved, the interferometer transmits aerosol and molecular backscatter with different efficiencies due to their spectral properties. We estimate those transmissions for the perfect tuning case and introducing accuracy or precision errors on the interferometer alignment.
Šmíd, Radek; Čížek, Martin; Mikel, Břetislav; Číp, Ondřej
2015-01-01
We present a method of noise suppression of laser diodes by an unbalanced Michelson fiber interferometer. The unstabilized laser source is represented by compact planar waveguide external cavity laser module, ORIONTM (Redfern Integrated Optics, Inc.), working at 1540.57 nm with a 1.5-kHz linewidth. We built up the unbalanced Michelson interferometer with a 2.09 km-long arm based on the standard telecommunication single-mode fiber (SMF-28) spool to suppress the frequency noise by the servo-loop control by 20 dB to 40 dB within the Fourier frequency range, remaining the tuning range of the laser frequency. PMID:25587980
Šmíd, Radek; Čížek, Martin; Mikel, Břetislav; Číp, Ondřej
2015-01-01
We present a method of noise suppression of laser diodes by an unbalanced Michelson fiber interferometer. The unstabilized laser source is represented by compact planar waveguide external cavity laser module, ORIONTM (Redfern Integrated Optics, Inc.), working at 1540.57 nm with a 1.5-kHz linewidth. We built up the unbalanced Michelson interferometer with a 2.09 km-long arm based on the standard telecommunication single-mode fiber (SMF-28) spool to suppress the frequency noise by the servo-loop control by 20 dB to 40 dB within the Fourier frequency range, remaining the tuning range of the laser frequency. PMID:25587980
Development of a new two color far infrared laser interferometer for future fusion devices
Kawahata, K.; Tanaka, K.; Tokuzawa, T.; Akiyama, T.; Ito, Y.; Okajima, S.; Nakayama, K.; Wylde, R.J.
2004-10-01
A new two color far infrared (FIR) laser interferometer under development for future fusion devices will be presented. The laser wavelength is optimized from the consideration of the beam refraction effect due to plasma density gradient and signal-to-noise ratio for an expected phase shift due to plasmas. Laser lines of 57.2 and 47.6 {mu}m are found to be suitable for the applications to high performance plasmas of Large Helical Device and future fusion devices such as the International Thermonuclear Experimental Reactor. The output power of 57.2 {mu}m CH{sub 3}OD laser is estimated to be {approx}1.6 W, which is the highest laser power in the FIR wavelength regime. The optical configuration of a new interferometer system using two colors will be proposed. In the system, one detector simultaneously detects the beat signals of the 57.2 and 47.6 {mu}m laser lines, and each interference signal can be separated electronically (1 MHz for 57.2 {mu}m and 0.84 MHz expected for 47.6 {mu}m). Mechanical vibration can be compensated by using the two color interferometer. The present status of the development of the system is also presented.
NASA Astrophysics Data System (ADS)
Belovolov, M. I.; Gur'yanov, A. N.; Gusovskiĭ, D. D.; Dianov, Evgenii M.; Kuznetsov, A. V.; Pencheva, V. Kh; Prokhorov, A. M.
1987-04-01
A fiber Michelson interferometer, which is an analog of a scanning Michelson interferometer, was constructed from single-mode fiber waveguides and several highly directional Y-type couplers operating at the wavelength of 0.85 μ. It was found that this interferometer could be used to investigate the coherence of radiation from single-frequency lasers with a spectral line width in the range from 10 kHz to 1 GHz. The experimental dependences of the degree of coherence on the optical path difference were determined for cw single-frequency AlGaAs double heterostructure lasers with an external dispersive resonator, operating in the regime of self-stabilization of single-frequency emission. It was found that the width of the spectral lines was between 10 kHz and 1 MHz, and that the line profiles were Lorentzian.
Characterization of the high frequency response of LASER interferometer gravitational wave detectors
NASA Astrophysics Data System (ADS)
Butler, William E.
This thesis describes a search for a stochastic background of gravitational waves at high frequency, 37.52 kHz. At this frequency the separation between the available instruments excludes the use of a correlation technique. Instead I rely on the spectral response of the LASER interferometer to isolate a possible signal from the underlying noise. This research was carried out at the LIGO (LASER Interferometer Gravitational Observatory) located in Hanford, WA and within the LIGO Scientific Collaboration (LSC). Chapter 1 serves as a general introduction to the present state of the search for gravitational waves (GW). I discuss the indirect observation of gravitational radiation as well as the expected sources for GW and their characteristics. I also discuss possible future developments, in particular the Advanced LIGO instruments and the LASER Interferometer Space Antenna (LISA). The characteristics of the large LASER interferometers, layout, terminology and necessary formulae are developed in Chapter 2. To carry out the proposed search it is essential that the frequency response of the interferometer be thoroughly understood, including possible noise sources. This was the subject of a series of experimental investigations using sideband injection and mirror excitations to characterize the IFO response in the region of the first free spectral range, which is at 37.52 kHz. The results of these experiments as well as their theoretical model are presented in Chapter 3. Contributions to the spectrum from mechanical noise are investigated in Chapter 4, and compared to the expected contribution thermal excitation. The results of my search are based on data obtained during the third science run of LIGO (S3) and are presented in Chapter 5. I show that a signal such as expected from a stochastic gravitational wave background is manifest in the data and compare it to the expected noise signal. This allows me to postulate a limit on a possible stochastic background. I also
Arm locking with the GRACE follow-on laser ranging interferometer
NASA Astrophysics Data System (ADS)
Thorpe, James Ira; McKenzie, Kirk
2016-02-01
Arm locking is a technique for stabilizing the frequency of a laser in an interspacecraft interferometer by using the spacecraft separation as the frequency reference. A candidate technique for future space-based gravitational wave detectors such as the Laser Interferometer Space Antenna, arm locking has been extensive studied in this context through analytic models, time-domain simulations, and hardware-in-the-loop laboratory demonstrations. In this paper we show the laser ranging interferometer instrument flying aboard the upcoming Gravity Recovery and Climate Experiment follow-on (GRACE-FO) mission provides an appropriate platform for an on-orbit demonstration of the arm-locking technique. We describe an arm-locking controller design for the GRACE-FO system and a series of time-domain simulations that demonstrate its feasibility. We conclude that it is possible to achieve laser frequency noise suppression of roughly 2 orders of magnitude around a Fourier frequency of 1 Hz with conservative margins on the system's stability. We further demonstrate that "pulling" of the master laser frequency due to fluctuating Doppler shifts and lock acquisition transients is less than 100 MHz over several GRACE-FO orbits. These findings motivate further study of the implementation of such a demonstration.
Wolfe, C.R.; Downie, J.D.; Lawson, J.K.
1996-06-27
The power spectral density (PSD) function is being employed to specify the surface finish and transmitted wavefront in the mid- spatial frequency regime for laser beam optics of the National Ignition Facility (NIF). The instrument used to measure the PSD is a phase measuring Fizeau interferometer. The phase map produced by the interferometer is digitally processed to create the PSD. Before one can use the PSD information, it is necessary to evaluate the fidelity of the interferometer spatial frequency response. Specifically, one must measure the overall transfer function of the instrument. To accomplish this, we perform a two-step ``calibration`` process. We first measure a known precision phase object with the interferometer and then compare the measured PSD to an ideal numerical simulation which represents the theoretical PSD. The square root of the ratio of the measured function to the simulation is defined as the transfer function of the instrument. We present experimental results for both reflective and transmissive test objects, including effects such as the test object orientation and longitudinal location in the interferometer cavity. We also evaluate the accuracy levels obtained using different test objects. 11 refs., 5 figs.
NASA Technical Reports Server (NTRS)
Seidel, David J.; Dubovitsky, Serge
2000-01-01
We report on the development, functional performance and space-qualification status of a laser stabilization system supporting a space-based metrology source used to measure changes in optical path lengths in space-based stellar interferometers. The Space Interferometry Mission (SIM) and Deep Space 3 (DS-3) are two missions currently funded by the National Aeronautics and Space Administration (NASA) that are space-based optical interferometers. In order to properly recombine the starlight received at each telescope of the interferometer it is necessary to perform high resolution laser metrology to stabilize the interferometer. A potentially significant error source in performing high resolution metrology length measurements is the potential for fluctuations in the laser gauge itself. If the laser frequency or wavelength is changing over time it will be misinterpreted as a length change in one of the legs of the interferometer. An analysis of the frequency stability requirement for SIM resulted in a fractional frequency stability requirement of square root (S(sub y)(f)) = <2 x 10(exp -12)/square root(Hz) at Fourier frequencies between 10 Hz and 1000 Hz. The DS-3 mission stability requirement is further increased to square root (S(sub y)(f)) = <5 x 10(exp -14)/Square root(Hz) at Fourier frequencies between 0.2 Hz and 10 kHz with a goal of extending the low frequency range to 0.05 Hz. The free running performance of the Lightwave Electronics NPRO lasers, which are the baseline laser for both SIM and DS-3 vary in stability and we have measured them to perform as follows (9 x l0(exp -11)/ f(Hz))(Hz)/square root(Hz)) = <( square root (S(sub y)(f)) = <(1.3 x l0(exp -8)/ f(Hz))/Square root(Hz). In order to improve the frequency stability of the laser we stabilize the laser to a high finesse optical cavity by locking the optical frequency of the laser to one of the transmission modes of the cavity. At JPL we have built a prototype space-qualifiable system meeting the
Fu, Haijin; Hu, Pengcheng; Tan, Jiubin; Fan, Zhigang
2015-07-10
A simple method was proposed by using a tunable attenuator fitted in the reference or measurement arm of a heterodyne laser interferometer to adjust the values of mixing laser beams while the spectrum of the measurement signal is monitored using a signal analyzer. The effectiveness of the proposed method in reducing the first-order optical nonlinearity was verified through experiments. Results indicated that the peak value of the first-order optical nonlinearity could be reduced from 5.15 to 0.24 nm. It was therefore concluded that the proposed method was applicable to ultraprecision laser interferometry. PMID:26193410
NASA Astrophysics Data System (ADS)
Mitryk, Shawn J.
2012-06-01
The existence of gravitational waves was theorized in 1916 by Albert Einstein in accordance with the linearized theory of general relativity. Most experiments and observations to date have supported general relativity, but now, nearly 100 years later, the scientific community has yet devise a method to directly measure gravitational radiation. With the first attempts towards a gravitational wave measurement in the 1960s, many methods have been proposed and tested since then, all failing thus far to provide a positive detection. The most promising gravitational radiation detection method is through the use of a space-based laser interferometer and with the advancement of modern technologies, these space-based gravitational wave measurements will eventually provide important scientific data to physics, astro-physics, and astronomy communities. The Laser Interferometer Space Antenna (LISA) is one such space-based laser interferometer. LISA's proposed design objective is to measure gravitational radiation in the frequency range from 30 microHz to 1 Hz using a modified Michelson interferometer. The interferometer arms are 5 Gm in length measured between each of the 3 spacecraft in the interferometer constellation. The differential arm-length will be measured to an accuracy of 18 pm/ Hz resulting in a baseline strain sensitivity of 3.6 x 10 --21 / Hz . Unfortunately, the dynamics of the spacecraft orbits complicate the differential arm-length measurements. The arms of the interferometer change in length resulting in time-dependent, unequal arm-lengths and laser Doppler shifts. Thus, to cancel the laser noise, laser beatnotes are formed between lasers on separate SC and, using these one-way laser phase measurements, one can reconstruct an equal-arm interferometer in post-processing. This is commonly referred to as time-delay interferometry (TDI) and can be exploited to cancel the laser phase noise and extract the gravitational wave (GW) induced arm-length strain. The
NASA Astrophysics Data System (ADS)
Herz, Markus
2005-09-01
Laser frequency stabilization is notably one of the major challenges on the way to a space-borne gravitational wave observatory. The proposed Laser Interferometer Space Antenna (LISA) is presently under development in an ESA, NASA collaboration. We present a novel method for active laser stabilization and phase noise suppression in such a gravitational wave detector. The proposed approach is a further evolution of the "arm-locking" method, which in essence consists of using an interferometer arm as an optical cavity, exploiting the extreme long-run stability of the cavity size in the frequency band of interest. We extend this method by using the natural interferometer arm length differences and existing interferometer signals as additional information sources for the reconstruction and active suppression of the quasi-periodic laser frequency noise, enhancing the resolution power of space-borne gravitational wave detectors.
NASA Astrophysics Data System (ADS)
Takatsuji, Toshiyuki; Koseki, Yoshihiko; Goto, Mitsuo; Kurosawa, Tomizo; Tanimura, Yoshihisa
1998-07-01
To measure three dimensional coordinate we have been developing a laser tracking interferometer system (LTS). Four laser interferometers chase the movement of a target cat's eye and measure the change in distance between them. The position of the cat's eye is determined from the measured distances based on the principle of trilateration. Taking advantage of measurement redundancy produced by the fourth tracker, the position of the trackers and the initial position of the cat's eye can be estimated by a self-calibration algorithm. A restriction on the arrangement of the laser trackers to perform the self-calibration algorithm is theoretically studied. Finally a preliminary experiment was made to show the measurement error of about 40 micrometers for a 1 m measurement.
Measuring Earth: Current status of the GRACE Follow-On Laser Ranging Interferometer
NASA Astrophysics Data System (ADS)
Schütze, Daniel; LRI team
2016-05-01
The GRACE mission that was launched in 2002 has impressively proven the feasibility of low-orbit satellite-to-satellite tracking for Earth gravity observations. Especially mass transport related to Earth's hydrological system could be well resolved both spatially and temporally. This allows to study processes such as polar ice sheet decline and ground water depletion in great detail. Owing to GRACE's success, NASA and GFZ will launch the successor mission GRACE Follow-On in 2017. In addition to the microwave ranging system, GRACE Follow-On will be the first mission to use a Laser Ranging Interferometer as technology demonstrator to track intersatellite distance changes with unprecedented precision. This new ranging device inherits some of the technologies which have been developed for the future spaceborne gravitational wave detector LISA. I will present the architecture of the Laser Ranging Interferometer, point out similarities and differences to LISA, and conclude with the current status of the flight hardware production.
NASA Astrophysics Data System (ADS)
Wei, Hong; Stout, Kenneth J.
1995-12-01
In this paper, vibration monitoring and measurement carried out in the newly developed nanometer metrology laboratory in Birmingham University, is described with respect to measurement methods and instrument performance. Two types of instrument -- piezoelectric accelerometer (B&K type 8318 with a type 2140 bus-controlled frequency analyzer) and laser interferometer (Renishaw ML10 laser interferometer with the Keithley multi-channel FFT analyzer) were used in vibration measurement on capability verification of a vibration isolation system. Vibration results from a concrete block are presented. From the point of view of the measured vibration results, it is demonstrated that the performance of the above two instruments is not completely the same in the different frequency ranges. The related comparison and discussions are presented in this paper.
Recent Progress of the HL-2A Multi-Channel HCOOH Laser Interferometer/Polarimeter
NASA Astrophysics Data System (ADS)
Li, Yonggao; Zhou, Yan; Deng, Zhongchao; Li, Yuan; Yi, Jiang; Wang, Haoxi
2015-05-01
A multichannel methanoic acid (HCOOH, λ = 432.5 μm) laser interferometer/polarimeter is being developed from the previous eight-channel hydrogen cyanide (HCN, λ = 337 μm) laser interferometer in the HL-2A tokamak. A conventional Michelson-type interometer is used for the electron density measurement, and a Dodel-Kunz-type polarimeter is used for the Faraday rotation effect measurement, respectively. Each HCOOH laser can produce a linearly polarized radiation at a power lever of ˜30 mW, and a power stability <10% in 50 min. A beam waist (diameter d0 ≈12.0 mm, about 200 mm away from the outlet) is finally determined through a chopping modulation technique. The latest optical layout of the interferometer/polarimeter has been finished, and the hardware data processing system based on the fast Fourier transform phase-comparator technique is being explored. In order to demonstrate the feasibility of the diagnostic scheme, two associated bench simulation experiments were carried out in the laboratory, in which the plasma was simulated by a piece of polytetrafluoroethene plate, and the Faraday rotation effect was simulated by a rotating half-wave plate. Simulation results agreed well with the initial experimental conditions. At present, the HCOOH laser interferometer/polarimeter system is being assembled on HL-2A, and is planned to be applied in the 2014-2015 experimental campaign. supported by the National Magnetic Confinement Fusion Science Programs of China (Nos. 2010GB101002 and 2014GB109001), and National Natural Science Foundation of China (Nos. 11075048 and 11275059)
NASA Astrophysics Data System (ADS)
Stadnik, Y. V.; Flambaum, V. V.
2016-06-01
We outline laser interferometer measurements to search for variation of the electromagnetic fine-structure constant α and particle masses (including a nonzero photon mass). We propose a strontium optical lattice clock—silicon single-crystal cavity interferometer as a small-scale platform for these measurements. Our proposed laser interferometer measurements, which may also be performed with large-scale gravitational-wave detectors, such as LIGO, Virgo, GEO600, or TAMA300, may be implemented as an extremely precise tool in the direct detection of scalar dark matter that forms an oscillating classical field or topological defects.
Shu, Hua; Fu, Sizu; Huang, Xiuguang; Wu, Jiang; Xie, Zhiyong; Zhang, Fan; Ye, Junjian; Jia, Guo; Zhou, Huazhen
2014-08-15
In this paper, we systematically study preheating in laser-direct-drive shocks by using a velocity interferometer system for any reflector (VISAR). Using the VISAR, we measured free surface velocity histories of Al samples over time, 10–70 μm thick, driven directly by a laser at different frequencies (2ω, 3ω). Analyzing our experimental results, we concluded that the dominant preheating source was X-ray radiation. We also discussed how preheating affected the material initial density and the measurement of Hugoniot data for high-Z materials (such as Au) using impedance matching. To reduce preheating, we proposed and tested three kinds of targets.
Pulsed picosecond KGW:Nd3+ laser based on the Sagnac interferometer
NASA Astrophysics Data System (ADS)
Grabovski, Vitaly V.; Prokhorenko, Valentin I.; Yatskiv, Dmytro Y.
1995-04-01
Energetic, statistical, and temporal characteristics of a pico-second KGW:Nd laser based on the Sagnac interferometer are studied. All experiments are provided in comparison with the traditional linear cavity. The effect of stimulated Raman scattering suppression in such a specific cavity was found. Stimulated Raman scattering in KGW dumped all output parameters in the laser with a linear cavity. A new cavity scheme can solve such problems and provide for good output handling. It is shown that output pulse duration is larger in the proposed scheme compared with a linear cavity but other characteristics are much better.
Status of advanced ground-based laser interferometers for gravitational-wave detection
NASA Astrophysics Data System (ADS)
Dooley, K. L.; Akutsu, T.; Dwyer, S.; Puppo, P.
2015-05-01
Ground-based laser interferometers for gravitational-wave (GW) detection were first constructed starting 20 years ago and as of 2010 collection of several years’ worth of science data at initial design sensitivities was completed. Upgrades to the initial detectors together with construction of brand new detectors are ongoing and feature advanced technologies to improve the sensitivity to GWs. This conference proceeding provides an overview of the common design features of ground-based laser interferometric GW detectors and establishes the context for the status updates of each of the four gravitational-wave detectors around the world: Advanced LIGO, Advanced Virgo, GEO 600 and KAGRA.
Fast wavelength-scanning interferometry technique with derivative detection of quadrature signals
NASA Astrophysics Data System (ADS)
Číp, O.; Mikel, B.; Lazar, J.
2006-04-01
We present a laser interferometer where a narrow-line width tuneable VCSEL laser (Vertical-Cavity Surface-Emitting Laser) working at 760 nm is used. For the detection of an absolute distance, we have used a fast wavelength-scanning interferometry technique. In the first part of the work we introduce the absolute laser interferometer as a demonstrator for research of a digital detection of quadrature signals (X-cos and Y-sin). This interferometer uses polarized beams and magnitude division of interference fringes. The wavelength of VCSEL laser is swept with the mode-hop free tuning range more than 1.2 nm, by means of the amplitude modulation of the injection current. At the same time, the operating temperature of the VCSEL is stabilized with a fast digital temperature controller. We control the wavelength value and whole tuning process of the laser with the frequency lock to selected modes of an external Fabry-Perot etalon. Except the frequency lock, the Fabry-Perot mode spectrum identifies wavelength-tuning interval of VCSEL during each sweep. A digital signal processor (DSP) is heart of the control and detection system. It samples intensity signal from Fabry- Perot etalon and X-Y quadrature signals from the detection unit of the interferometer. After 1 nm sweep of the VCSEL wavelength, we obtain a number of passed interference fringes and the number of passed Fabry-Perot resonance modes, at the same time. On basis of these measured quantities we are able to calculate the instantaneous value of the optical path length difference between the measuring and reference arm of the demonstrational interferometer. The other part of the work is oriented to research and experimental testing of the digital detection of quadrature signals (X-cos and Y-sin) processed only on basis of one intensity signal (X-axis) that is produced by a simple photo-detector. On basis of traditional inversion function arctan(Y/X) we are able to determine instantaneous phase between interference
NASA Astrophysics Data System (ADS)
Rousseau, Guy; Blouin, Alain; Monchalin, Jean-Pierre
2009-02-01
Ultrasound-modulated optical imaging is an emerging biodiagnostic technique which provides the optical spectroscopic signature and the spatial localization of an optically absorbing object embedded in a strongly scattering medium. The transverse resolution of the technique is determined by the lateral extent of ultrasound beam focal zone while the axial resolution is obtained by using short ultrasound pulses. The practical application of this technique is presently limited by its poor sensitivity. Moreover, any method to enhance the signal-to-noise ratio must satisfy the biomedical safety limits. In this paper, we propose to use a pulsed single-frequency laser source to raise the optical peak power applied to the scattering medium and to collect more ultrasonically tagged photons. Such a laser source allows illuminating the tissues mainly during the transit time of the ultrasonic wave. A single-frequency Nd:YAG laser emitting 500-μs pulses with a peak power superior to 100 W was used. Tagged photons were detected with a GaAs photorefractive interferometer characterized by a large optical etendue. When pumped by high intensity laser pulses, such an interferometer provides the fast response time essential to obtain an apparatus insensitive to the speckle decorrelation encountered in biomedical applications. Consequently, the combination of a large-etendue photorefractive interferometer with a high-power pulsed laser could allow obtaining both the sensitivity and the fast response time necessary for biomedical applications. Measurements performed in 30- and 60-mm thick optical phantoms made of titanium dioxide particles dispersed in sunflower oil are presented. Results obtained in 30- and 60-mm thick chicken breast samples are also reported.
Interferometer for the measurement of plasma density
Jacobson, Abram R.
1980-01-01
An interferometer which combines the advantages of a coupled cavity interferometer requiring alignment of only one light beam, and a quadrature interferometer which has the ability to track multi-fringe phase excursions unambiguously. The device utilizes a Bragg cell for generating a signal which is electronically analyzed to unambiguously determine phase modulation which is proportional to the path integral of the plasma density.
A nonintrusive laser interferometer method for measurement of skin friction
NASA Technical Reports Server (NTRS)
Monson, D. J.
1982-01-01
A method is described for monitoring the changing thickness of a thin oil film subject to an aerodynamic shear stress using two focused laser beams. The measurement is then simply analyzed in terms of the surface skin friction of the flow. The analysis includes the effects of arbitrarily large pressure and skin friction gradients, gravity, and time varying oil temperature. It may also be applied to three dimensional flows with unknown direction. Applications are presented for a variety of flows including two dimensional flows, three dimensional swirling flows, separated flow, supersonic high Reynolds number flows, and delta wing vortical flows.
Alekseev, A E; Potapov, V T
2013-10-31
Spectral characteristics of the noise intensity fluctuations at the output of a scattered-light interferometer, caused by phase fluctuations of semiconductor laser radiation are considered. This kind of noise is one of the main factors limiting sensitivity of interferometric sensors. For the first time, to our knowledge, the expression is obtained for the average noise power spectral density at the interferometer output versus the degree of a light source coherence and length of the scattering segment. Also, the approximate expressions are considered which determine the power spectral density in the low-frequency range (up to 200 kHz) and in the limiting case of extended scattering segments. The expression obtained for the noise power spectral density agrees with experimental normalised power spectra with a high accuracy. (interferometry of radiation)
Large-Optics white light interferometer for laser wavefront test: apparatus and application
NASA Astrophysics Data System (ADS)
Luan, Zhu; Liu, Liren; Wang, Lijuan; Liu, De'an
2008-08-01
There is transmitting optics of 250mm aperture with about 8 microradians in SILEX system. This is often large aperture and diffraction-limited laser beam in the laser communications. Large-Optics white light interferometer using double-shearing structure has been submitted to analysis the laser wavefront before. Six optical plates of 490 millimeters apertures are manufactured now one of which is also aperture-divided so that the precision of measured wave front is higher than the full aperture design. It is suitable for measurement of minimum diffraction-limited laser wave front and any wavelength. The interference is happened between equal optical path of the reflection and the other. The plates are the basic structures which are precisely parallel or perpendicular needed for either two plates. There are several tools equipped with the interferometer including white light test source and collimators and so on to confirm the precision of several seconds angle. The apparatus and application is explained in detail in this paper. The adjustment is important for the realization of white light test.
NASA Technical Reports Server (NTRS)
Byer, R. L.
1990-01-01
Two laser pumped Nd:YAG lasers were frequency stabilized to a commercial 6.327 GHz free spectral range Fabry-Perot interferometer yielding a best case beatnote linewidth of 330 MHz. In addition, a Fabry-Perot interferometer with a free spectral range of 680 MHz, a linewidth of 25 kHz, and a finesse of 27,500 was built, and when it was substituted in place of the commercial interferometer, it produced a robust and easily repeatable beatnote linewidth of 700 MHz.
Compressive imaging and dual moire laser interferometer as metrology tools
NASA Astrophysics Data System (ADS)
Abolbashari, Mehrdad
Metrology is the science of measurement and deals with measuring different physical aspects of objects. In this research the focus has been on two basic problems that metrologists encounter. The first problem is the trade-off between the range of measurement and the corresponding resolution; measurement of physical parameters of a large object or scene accompanies by losing detailed information about small regions of the object. Indeed, instruments and techniques that perform coarse measurements are different from those that make fine measurements. This problem persists in the field of surface metrology, which deals with accurate measurement and detailed analysis of surfaces. For example, laser interferometry is used for fine measurement (in nanometer scale) while to measure the form of in object, which lies in the field of coarse measurement, a different technique like moire technique is used. We introduced a new technique to combine measurement from instruments with better resolution and smaller measurement range with those with coarser resolution and larger measurement range. We first measure the form of the object with coarse measurement techniques and then make some fine measurement for features in regions of interest. The second problem is the measurement conditions that lead to difficulties in measurement. These conditions include low light condition, large range of intensity variation, hyperspectral measurement, etc. Under low light condition there is not enough light for detector to detect light from object, which results in poor measurements. Large range of intensity variation results in a measurement with some saturated regions on the camera as well as some dark regions. We use compressive sampling based imaging systems to address these problems. Single pixel compressive imaging uses a single detector instead of array of detectors and reconstructs a complete image after several measurements. In this research we examined compressive imaging for different
Squeezing quadrature rotation in the acoustic band via optomechanics
NASA Astrophysics Data System (ADS)
Guccione, Giovanni; Slatyer, Harry J.; Carvalho, André R. R.; Buchler, Ben C.; Lam, Ping Koy
2016-03-01
We examine the use of optomechanically generated squeezing to obtain a sensitivity enhancement for interferometers in the gravitational-wave band. The intrinsic dispersion characteristics of optomechanical squeezing around the mechanical frequency are able to produce squeezing at different quadratures over the spectrum, a feature required by gravitational-wave interferometers to beat the standard quantum limit over an extended frequency range. Under realistic assumptions we show that the amount of available squeezing and the intrinsic quadrature rotation may provide, compared to similar amounts of fixed-quadrature squeezing, a detection advantage. A significant challenge for this scheme, however, is the amount of excess noise that is generated in the unsqueezed quadrature at frequencies near the mechanical resonance.
Note: Design of a laser feedback interferometer with double diffraction system
Guo, Dongmei Wang, Ming
2015-09-15
A novel laser feedback interferometer with double diffraction system is proposed in this paper. A beam emitted from the laser is incident onto a transmission grating. The mth order beam is vertically reflected by a mirror and diffracted again by the grating. The double-diffracted beam returns into the laser cavity and mixes with the light inside the active cavity, thus generating a modulation of both the amplitude and the frequency of the lasing field. Theoretical analysis and experimental observations show that the output signal of the proposed system depends on the grating pitch and the direction of the phase movement can be obtained from inclination of the interference signal. It provides a potential displacement sensor with high stability and quite a compact configuration.
Laser interferometer/Preston tube skin-friction comparison in shock/boundary-layer interaction
NASA Technical Reports Server (NTRS)
Kim, K.-S.; Lee, Y.; Settles, G. S.
1991-01-01
An evaluation is conducted of the accuracy of the 'Preston tube' surface pitot-pressure skin friction measurement method relative to the already proven laser interferometer skin-friction meter in a swept shock wave/turbulent boundary-layer interaction. The Preston tube was used to estimate the total shear-stress distribution in a fin-generated swept shock-wave/turbulent boundary-layer interaction. The Keener-Hopkins calibration method using the isentropic relation to calculate the Preston-tube Mach number produces the best results.
NASA Astrophysics Data System (ADS)
Dotti, D.; Famiglietti, M.; Scandolo, D.
1981-08-01
The electrical impedance and the most important electromechanical parameters of the commercial piezoelectric device PZT-5A were measured. With a high resolution laser interferometer, the amplitude of the vibrations at the surface were determined and related to the electric load characteristics. The experimental results were compared with estimations from a one dimensional theoretical model. The measurements are part of a program for defining vibration control techniques for ultrasound transducers and for absolute calibration of sensors. The characteristics and sensibility of an experimental piezoelectronic acoustical transducer are also presented.
A mode-locked fibre laser using a Sagnac interferometer and nonlinear polarization rotation
NASA Astrophysics Data System (ADS)
Ibarra-Escamilla, B.; Kuzin, E. A.; Gomez-Garcia, D. E.; Gutierrez-Zainos, F.; Mendoza-Vazquez, S.; Haus, J. W.
2003-09-01
We analyse the propagation in a nonlinear, birefringent optical fibre with twist. The results show that the polarization evolution is periodic and they are applied to the analysis of a Sagnac interferometer. We give a useful way to visualize the behaviour of the nonlinear optical loop mirror (NOLM) (as a function of birefringence, twist, length, and input polarization) in terms of the Poincaré sphere. We describe a new fibre laser configuration based on the NOLM with a symmetrical coupler, quarter-wave retarder, and highly twisted, birefringent fibre in the loop. We achieved stable generation of subpicosecond pulses with milliwatts of average output power.
NASA Astrophysics Data System (ADS)
Klipstein, William; McKenzie, Kirk; Grace Follow-On LASER Ranging Interferometer Team
2016-03-01
GRACE Follow-On will replace the Gravity Recovery and Climate Experiment (GRACE) mission, which has been measuring Earth's gravity field since 2002. Like GRACE, GRACE Follow-On will use a microwave link as its primary instrument to measure micron-level changes in the 200km separation of a pair of satellites in a following polar orbit. GRACE Follow-On will also include a 2-way laser-link, the Laser Ranging Interferometer (LRI), as a technology demonstrator package. The LRI is an NASA/German partnership and will demonstrate inter-spacecraft laser interferometry with a goal of 10 times better precision than the microwave instrument, or about 90 nm/ √(Hz) between 10 and 100 mHz. The similarities between the LRI and a single arm of Laser Interferometer Space Antenna (LISA) mean many of the required technologies will be the same. This talk will give an overview of the LRI and the status of the LRI instruments, and implications for LISA.
NASA Astrophysics Data System (ADS)
Ruden, E. L.; Camacho, J. F.; Lynn, A. G.
2013-10-01
The interference signal visibility V (difference to sum ratio of intensities at maximum and minimum interference) of an interferometer that uses a multimode laser is here derived for a given laser gain profile and spectral mode separation as a function of the difference ZS between the probe and reference beam optical path lengths and the spectral separation kS between the center of the laser gain profile and the nearest laser mode of higher frequency. kS has a significant effect on V for a given ZS. This parameter, in lasers where it sweeps freely across the gain profile, and other effects, such as various misalignments and optical coupling inefficiencies, render V alone an unreliable parameter for quantifying ZS (for the purpose of reducing it, say). However, the difference to sum ratio of the maximum and minimum V due to variations in kS for a given ZS is an intrinsic property of the laser insensitive to configurational details. Parameter W so defined, therefore, proves very useful for balancing path lengths. This is of particular importance for systems where probe and/or reference beams are transmitted via long single mode optical fibers, so this application is detailed. Optical path lengths within such fibers often cannot be measured to sufficient accuracy by spatial path length measurements due to fiber nonuniformity resulting in variations in the mode's group velocity (needed to convert to optical path length). Two examples are provided using different makes and models of 0.633 μm HeNe lasers with similar specifications. In the first case, the function W(ZS) is calculated directly from the laser's published gain profile and mode separation. In the second case, W is determined empirically for a range of ZS values for a laser with an unknown gain profile in a (heterodyned) interferometer whose interference signal oscillates between maximum and minimum intensity at 80 MHz due to the reference beam's optical frequency being acousto-optically upshifted by that
Q-switching of a high-power solid-state laser by a fast scanning Fabry-Perot interferometer
Baburin, N V; Borozdov, Yu V; Danileiko, Yu K; Denker, B I; Ivanov, A D; Osiko, Vyacheslav V; Sverchkov, S E; Sidorin, A V; Chikov, V A; Ifflander, R; Hack, R; Kertesz, I; Kroo, N
1998-07-31
An investigation was made of the suitability of a Q-switch, based on a piezoelectrically scanned short-base Fabry-Perot interferometer, for an Nd{sup 3+}:YAG laser with an average output radiation power up to 2 kW. The proposed switch made it possible to generate of giant pulses of 60 - 300 ns duration at a repetition rate of 20 - 100 kHz. Throughout the investigated range of the pulse repetition rates the average power was at least equal to that obtained by cw lasing. Special requirements to be satisfied by the interferometer, essential for efficient Q-switching, were considered. (control of laser radiation parameters)
Application of a laser interferometer skin-friction meter in complex flows
NASA Technical Reports Server (NTRS)
Monson, D. J.; Driver, D. M.; Szodruch, J.
1981-01-01
The application of a nonintrusive laser-interferometer skin-friction meter, which measures skin friction with a remotely located laser interferometer that monitors the thickness change of a thin oil film, is extended both experimentally and theoretically to several complex wind-tunnel flows. These include two-dimensional seperated and reattached subsonic flows with large pressure and shear gradients, and two and three-dimensional supersonic flows at high Reynolds number, which include variable wall temperatures and cross-flows. In addition, it is found that the instrument can provide an accurate location of the mean reattachment length for separated flows. Results show that levels up to 120 N/sq m, or 40 times higher than previous tests, can be obtained, despite encountering some limits to the method for very high skin-friction levels. It is concluded that these results establish the utility of this instrument for measuring skin friction in a wide variety of flows of interest in aerodynamic testing.
NASA Technical Reports Server (NTRS)
Folkner, W. M.; Bender, P. L.; Stebbins, R. T.
1998-01-01
This document presents the results of a design feasibility study for LISA (Laser Interferometer Space Antenna). The goal of LISA is to detect and study low-frequency astrophysical gravitational radiation from strongly relativistic regions. Astrophysical sources potentially visible to LISA include extra-galactic massive black hole binaries at cosmological distances, binary systems composed of a compact star and a massive black hole, galactic neutron star-black hole binaries, and background radiation from the Big Bang. The LISA mission will comprise three spacecraft located five million kilometers apart forming an equilateral triangle in an Earth-trailing orbit. Fluctuations in separation between shielded test masses located within each spacecraft will be determined by optical interferometry which determines the phase shift of laser light transmitted between the test masses.
Internal structure of laser supported detonation waves by two-wavelength Mach-Zehnder interferometer
Shimamura, Kohei; Kawamura, Koichi; Fukuda, Akio; Wang Bin; Yamaguchi, Toshikazu; Komurasaki, Kimiya; Hatai, Keigo; Fukui, Akihiro; Arakawa, Yoshihiro
2011-04-15
Characteristics of the internal structure of the laser supported detonation (LSD) waves, such as the electron density n{sub e} and the electron temperature T{sub e} profiles behind the shock wave were measured using a two-wavelength Mach-Zehnder interferometer along with emission spectroscopy. A TEA CO{sub 2} laser with energy of 10 J/pulse produced explosive laser heating in atmospheric air. Results show that the peak values of n{sub e} and T{sub e} were, respectively, about 2 x 10{sup 24} m{sup -3} and 30 000 K, during the LSD regime. The temporal variation of the laser absorption coefficient profile estimated from the measured properties reveals that the laser energy was absorbed perfectly in a thin layer behind the shock wave during the LSD regime, as predicted by Raizer's LSD model. However, the absorption layer was much thinner than a plasma layer, the situation of which was not considered in Raizer's model. The measured n{sub e} at the shock front was not zero while the LSD was supported, which implies that the precursor electrons exist ahead of the shock wave.
The three-wave laser polarimeter-interferometer on J-TEXT tokamak
NASA Astrophysics Data System (ADS)
Zhuang, G.; Liu, Y.; Chen, J.; Gao, L.; Li, Q.; Xiong, C. Y.; Shi, P.; Zhou, Y. N.
2016-02-01
Motivated by increasing demands on high-quality measurement of interior magnetic field in tokamak plasma, a far-infrared laser-based polarimeter-interferometer system has been developed on J-TEXT. Three formic acid lasers separately pumped by three CO2 lasers are used as sources, providing more than 90 mW output power in total. High laser power along with usage of newly developed planar Schottky diode mixer enable high phase resolution < 1 mrad. Collinearity and polarization calibrations have been carefully done to improve the measurement reliability. Meanwhile, real-time feedback control of three-wave laser source has been realized for the first time, to fulfill the system stability. Based on three-wave technique, Faraday angle and integrated density phase along the laser path are simultaneously measured with high temporal resolution. In addition, the laser beam is expanded to cover the entire cross-section of the plasma to provide high spatial resolution measurement. With this system, MHD equilibrium of the J-TEXT plasma has been reconstructed. Obscure perturbations on magnetic topology and electron density associated with MHD instabilities, e.g. sawteeth and tearing modes have also been observed. In particular, some interesting features of disruptions in high-density discharges are identified by carefully interpreting the measured polarimeter-interferometer data. In the density ramp-up phase of a high density discharge, asymmetry in both electron density and current density profiles between the Low-Field-Side (LFS) edge (r > 0.8a) and the High-Field-Side (HFS) edge (r < -0.8a) would appear and extend gradually toward the center region. At the same time, a low-frequency (< 1 kHz) density perturbation suddenly occurs at the HFS edge and also gradually propagates into the center region. The disruption takes place when the electron density asymmetry/perturbation reaches the location nearly the m/n = 2/1 (where m and n are the toroidal mode number and the poloidal
NASA Technical Reports Server (NTRS)
Martino, Anthony J.; Cornwell, Donald M.
1998-01-01
A combination of a single mode AlGaAs laser diode and broadband LED was used in a Michelson interferometer to provide reference signals in a Fourier transform spectrometer, the Composite Infrared Spectrometer, on the Cassini mission to Saturn. The narrowband light from the laser produced continuous fringes throughout the travel of the interferometer, which were used to control the velocity of the scan mechanism and to trigger data sampling. The broadband light from the LED produced a burst of fringes at zero path difference, which was used as a fixed position reference. The system, including the sources, the interferometer, and the detectors, was designed to work both at room temperature and instrument operating temperature of 170 Kelvin. One major challenge that was overcome was preservation, from room temperature to 170 K, of alignment sufficient for high modulation of fringes from the broadband source. Another was the shift of the source spectra about 30 nm toward shorter wavelengths upon cooldown.
NASA Astrophysics Data System (ADS)
Giaime, Joseph Anthony
1995-01-01
Two techniques are developed that are needed in the design of an interferometric gravitational wave (GW) detector such as the LIGO, or Long-baseline Interferometric Gravitational-wave Observatory. The detector sensitivity of a long-baseline instrument is studied. A multi-layer mechanical isolation stack to filter seismic noise from test masses is designed, modeled and tested in vacuum. This is a four-stage elastomer (spring) and stainless steel (mass) stack, consisting of a table resting on three separate legs of three layers each. The visco-elastic properties of elastomer springs are exploited to damp the stack's normal modes while providing rapid roll-off of stack transmission above these modal frequencies. The stack's transmission of base motion to top motion is measured in vacuum and compared with 3-D finite-element models. In one tested configuration, at 100 Hz, horizontal transmission is 10^{-7}, vertical transmission is 3 times 10^{-6}, and the cross-coupling terms are between these values. A length detection scheme using RF phase modulated light and synchronous detection is developed for Fabry -Perot arm power-recycled Michelson interferometer GW detectors. This scheme uses an external Mach-Zehnder interferometer to measure the GW signal, and a frequency-shifted subcarrier to measure ancillary interferometer degrees of freedom. Use of the Mach-Zehnder allows rejection of laser source amplitude noise from the output, as well as the ability to exploit well-balanced Fabry-Perot arms to reject frequency noise from the output. A long baseline GW detector using these techniques should meet the LIGO initial goal sensitivity to GW strain of h_{rm RMS} = 10^ {-21} at 100 Hz. (Copies available exclusively from MIT Libraries, Rm. 14-0551, Cambridge, MA 02139-1307. Ph. 617 -253-5668; Fax 617-253-1690.).
Positive quadrature formulas III
NASA Astrophysics Data System (ADS)
Peherstorfer, Franz
2008-12-01
First we discuss briefly our former characterization theorem for positive interpolation quadrature formulas (abbreviated qf), provide an equivalent characterization in terms of Jacobi matrices, and give links and applications to other qf, in particular to Gauss-Kronrod quadratures and recent rediscoveries. Then for any polynomial t_n which generates a positive qf, a weight function (depending on n ) is given with respect to which t_n is orthogonal to mathbb{P}_{n-1} . With the help of this result an asymptotic representation of the quadrature weights is derived. In general the asymptotic behaviour is different from that of the Gaussian weights. Only under additional conditions do the quadrature weights satisfy the so-called circle law. Corresponding results are obtained for positive qf of Radau and Lobatto type.
Fabry-Perot interferometer embedded in a glass chip fabricated by femtosecond laser.
Lin, Cheng-Hsiang; Jiang, Lan; Xiao, Hai; Chai, Yen-Hsin; Chen, Shean-Jen; Tsai, Hai-Lung
2009-08-15
We report a simple Fabry-Perot interferometer (FPI) embedded in a glass chip, which is capable of precisely measuring the refractive indices of liquid samples. The microdevice is the integration of a single-mode optical fiber and a microchannel in the photosensitive glass fabricated by femtosecond laser followed by thermal treatment, wet etching, and annealing. The function of the FPI is demonstrated by measuring the refractive indices of water and methanol. The interference visibility is more than 4.0 dB, which is sufficient for most sensing applications. This refractive index sensor with rigid structure could be further integrated to become a more complex 3D lab-on-a-chip for reliable biomedical applications. PMID:19684798
Smith, Tristan L.; Peiris, Hiranya V.; Cooray, Asantha
2006-06-15
A detection of the primordial gravitational wave background is considered to be the 'smoking-gun' evidence for inflation. While superhorizon waves are probed with cosmic microwave background (CMB) polarization, the relic background will be studied with laser interferometers. The long lever arm spanned by the two techniques improves constraints on the inflationary potential and validation of consistency relations expected under inflation. If gravitational waves with a tensor-to-scalar amplitude ratio greater than 0.01 are detected by the CMB, then a direct-detection experiment with a sensitivity consistent with current concept studies should be pursued vigorously. If no primordial tensors are detected by the CMB, a direct-detection experiment to understand the simplest form of inflation must have a sensitivity improved by two to 3 orders of magnitude over current plans.
Test environments for the GRACE follow-on laser ranging interferometer
NASA Astrophysics Data System (ADS)
Görth, A.; Sanjuan, J.; Gohlke, M.; Rasch, S.; Abich, K.; Braxmaier, C.; Heinzel, G.
2016-05-01
In the year 2017 a follow-on mission to the very successful joint German/US mission GRACE (Gravity Recovery And Climate Experiment) will be launched. As of this day the two GRACE satellites have successfully been mapping the spatial and temporal varitations of the gravitational field of the Earth by satellite-to-satellite tracking for over a decade. The main science instrument on GRACE and its successor mission GRACE Follow-On which is used to measure the inter-satellite distance changes is a microwave link sensor. However, an additional instrument, the laser ranging interferometer (LRI), will be implemented into the architecture of the GRACE Follow-On satellites as a technology demonstrator. In this paper we will give a brief overview of a fiber-based test environment which is currently used during the assembly, integration and test of the LRI flight hardware.
Application of a Laser Interferometer Skin-Friction Meter in Complex Flows
NASA Technical Reports Server (NTRS)
Monson, D. J.; Driver, D. M.; Szodruch, J.
1981-01-01
A nonintrusive skin-friction meter has been found useful for a variety of complex wind-tunnel flows. This meter measures skin friction with a remotely located laser interferometer that monitors the thickness change of a thin oil film. Its accuracy has been proven in a low-speed flat-plate flow. The wind-tunnel flows described here include sub-sonic separated and reattached flow over a rearward-facing step, supersonic flow over a flat plate at high Reynolds numbers, and supersonic three - dimensional vortical flow over the lee of a delta wing at angle of attack. The data-reduction analysis was extended to apply to three-dimensional flows with unknown flow direction, large pressure and shear gradients, and large oil viscosity changes with time. The skin friction measurements were verified, where possible, with results from more conventional techniques and also from theoretical computations.
Two-color interferometer for the study of laser filamentation triggered electric discharges in air
Point, Guillaume Brelet, Yohann; Arantchouk, Leonid; Carbonnel, Jérôme; Prade, Bernard; Mysyrowicz, André; Houard, Aurélien
2014-12-15
We present a space and time resolved interferometric plasma diagnostic for use on plasmas where neutral-bound electron contribution to the refractive index cannot be neglected. By recording simultaneously the plasma optical index at 532 and 1064 nm, we are able to extract independently the neutral and free electron density profiles. We report a phase resolution of 30 mrad, corresponding to a maximum resolution on the order of 4×10{sup 22} m{sup −3} for the electron density, and of 10{sup 24} m{sup −3} for the neutral density. The interferometer is demonstrated on centimeter-scale sparks triggered by laser filamentation in air with typical currents of a few tens of A.
Liao, C R; Hu, T Y; Wang, D N
2012-09-24
We demonstrate a fiber in-line Fabry-Perot interferometer cavity sensor for refractive index measurement. The interferometer cavity is formed by drilling a micro-hole at the cleaved fiber end facet, followed by fusion splicing. A micro-channel is inscribed by femtosecond laser micromachining to vertically cross the cavity to allow liquid to flow in. The refractive index sensitivity obtained is ~994 nm/RIU (refractive index unit). Such a device is simple in configuration, easy for fabrication and reliable in operation due to extremely low temperature cross sensitivity of ~4.8 × 10(-6) RIU/°C. PMID:23037431
Monitoring Rotational Components of Seismic Waves with a Ring Laser Interferometer
NASA Astrophysics Data System (ADS)
Gakundi, Jackson; Dunn, Robert
2015-04-01
It has been known for decades that seismic waves can introduce rotation in the surface of the Earth. There are historic records of tombstones in Japan being rotated after large earthquakes. Until fairly recently, the primary way to detect ground rotation from earthquakes was with an array of several seismographs. The development of large ring laser interferometers has provided a way for a single instrument to make extremely sensitive measurements of ground motion. In this poster, a diagram of a large ring laser will be presented. For comparison, seismograms recorded with a ring laser and a collocated standard seismograph will be presented. A major thrust of this research is the detection and analysis of seismic responses from directional drilling sites in Arkansas and Oklahoma. There are suggestions that the injection of pressurized water used to fracture gas bearing shale may cause small earthquakes. The Arkansas Oil and Gas Commission ordered the closing of certain waste water disposal wells in North Central Arkansas. Apparently, these wells injected waste water into a previously unknown fault causing it to slip. An attempt is being made to determine if the seismic wave patterns from earthquakes generated near directional drilling sites differ from those generated miles away.
Stability of fringe counting interferometers
NASA Technical Reports Server (NTRS)
Edgerton, J. W.; Andrew, K. L.
1974-01-01
Two configurations of an automatic bidirectional, fringe-counting corner-cube interferometer are compared. They differ only in the method of quadrature phase introduction. The one using polarization coding has good phase stability at optical path differences as large as 955 mm, the one using adjacent beams has such poor phase stability as to render it useless at path differences greater than 700 mm. A useful well-defined alignment procedure is given for the corner-cube interferometer.
Buchenauer, C.J.
1981-09-23
The quadrature phase angle phi (t) of a pair of quadrature signals S/sub 1/(t) and S/sub 2/(t) is digitally encoded on a real time basis by a quadrature digitizer for fractional phi (t) rotational excursions and by a quadrature up/down counter for full phi (t) rotations. The pair of quadrature signals are of the form S/sub 1/(t) = k(t) sin phi (t) and S/sub 2/(t) = k(t) cos phi (t) where k(t) is a signal common to both. The quadrature digitizer and the quadrature up/down counter may be used together or singularly as desired or required. Optionally, a digital-to-analog converter may follow the outputs of the quadrature digitizer and the quadrature up/down counter to provide an analog signal output of the quadrature phase angle phi (t).
Buchenauer, C. Jerald
1984-01-01
The quadrature phase angle .phi.(t) of a pair of quadrature signals S.sub.1 (t) and S.sub.2 (t) is digitally encoded on a real time basis by a quadrature digitizer for fractional .phi.(t) rotational excursions and by a quadrature up/down counter for full .phi.(t) rotations. The pair of quadrature signals are of the form S.sub.1 (t)=k(t) sin .phi.(t) and S.sub.2 (t)=k(t) cos .phi.(t) where k(t) is a signal common to both. The quadrature digitizer and the quadrature up/down counter may be used together or singularly as desired or required. Optionally, a digital-to-analog converter may follow the outputs of the quadrature digitizer and the quadrature up/down counter to provide an analog signal output of the quadrature phase angle .phi.(t).
Tunable double-clad ytterbium-doped fiber laser based on a double-pass Mach-Zehnder interferometer
NASA Astrophysics Data System (ADS)
Meng, Yichang; Zhang, Shumin; Wang, Xinzhan; Du, Juan; Li, Hongfei; Hao, Yanping; Li, Xingliang
2012-03-01
We have demonstrated an adjustable double-clad Yb 3+-doped fiber laser using a double-pass Mach-Zehnder interferometer. The laser is adjustable over a range of 40 nm from 1064 nm to 1104 nm. By adjusting the state of the polarization controller, which is placed in the double-pass Mach-Zehnder interferometer, we obtained central lasing wavelengths that can be accurately tuned with controllable spacing between different tunable wavelengths. The laser has a side mode suppression ratio of 42 dB, the 3 dB spectral width is less than 0.2 nm, and the slope efficiencies at 1068 nm, 1082 nm and 1098 nm are 23%, 32% and 26%, respectively. In addition, we have experimentally observed tunable multi-wavelengths lasing output.
Kawata, Yoshiyuki; Hyashi, Kyohei; Aoto, Tomohiro
2015-06-15
A two-wavelength interferometer (TWI) based on a sinusoidal-phase-modulation method with an acetylene stabilized laser and a second harmonic generation (SHG) was developed. The periodic non-linearity error for the TWI was estimated to be ± 0.1 µm at a dead path of 0.32 m. A long-term measurement showed that the TWI stability was ± 3 × 10(-7) at a dead path of 1.00 m for 12 hours with an ambient pressure variation of 3 hPa under controlled conditions of ambient temperature and humidity. Finally, we confirmed that the TWI has substantially better stability than a single-wavelength interferometer by comparing both interferometers with large temporal and spatial temperature variations. PMID:26193576
Quadrature Mixer LO Leakage Suppression Through Quadrature DC Bias
BALDWIN, JESSE G; DUBBERT, DALE F.
2002-05-01
A new concept has been developed which allows direct-to-RF conversion of digitally synthesized waveforms. The concept named Quadrature Error Corrected Digital Waveform Synthesis (QECDWS) employs quadrature amplitude and phase predistortion to the complex waveform to reduce the undesirable quadrature image. Another undesirable product of QECDWS-based RF conversion is the Local Oscillator (LO) leakage through the quadrature upconverter (mixer). A common technique for reducing this LO leakage is to apply a quadrature bias to the mixer I and Q inputs. This report analyzes this technique through theory, lab measurement, and data analysis for a candidate quadrature mixer for Synthetic Aperture Radar (SAR) applications.
NASA Astrophysics Data System (ADS)
Liu, H. Q.; Jie, Y. X.; Ding, W. X.; Brower, D. L.; Zou, Z. Y.; Qian, J. P.; Li, W. M.; Yang, Y.; Zeng, L.; Zhang, S. B.; Lan, T.; Wang, S. X.; Hanada, K.; Wei, X. C.; Hu, L. Q.; Wan, B. N.
2016-01-01
A multi-channel far-infrared laser-based POlarimeter-INTerferometer (POINT) system utilizing the three-wave technique has been implemented for fully diagnosing the internal magnetic field in the EAST tokamak. Double-pass, horizontal, radially-viewing chords access the plasma via an equatorial port. The laser source consists of three CW formic acid (HCOOH) FIR lasers at nominal wavelength 432.5 μm which are optically pumped by independent infrared CO2 lasers. Output power is more than 30 mW of per cavity. Novel molybdenum retro-reflectors, can with withstand baking temperature up to 350°C and discharge duration more than 1000 s, are mounted in the inside wall for the double-pass optical arrangement. A Digital Phase Detector with 250 kHz bandwidth, which provide real-time Faraday rotation angle and density phase shift output for plasma control, have been developed for the POINT system. Reliability of both polarimetric and interferometric measurement are obtained in 22 s long pulse H mode discharge and 8 s NBI H mode discharge, indicating the POINT system works for any heating scheme on EAST so far. The electron line-integrated density resolution of POINT is less than 1 × 1016 m-2 (< 1°), and the Faraday rotation angle rms phase noise is < 0.1°. With the high temporal (~ 1 μsec) and phase resolution (< 0.1°), perturbations associated with the sawtooth cycle and MHD activity have been observed. The current profile, density profile and safety factor (q) profile are reconstructed by using EFIT code from the external magnetic and the validation POINT data. Realtime EFIT with Faraday angle and density phase shift constraints will be implemented in the plasma control system in the future.
Correlated quadratures of resonance fluorescence and the generalized uncertainty relation
NASA Technical Reports Server (NTRS)
Arnoldus, Henk F.; George, Thomas F.; Gross, Rolf W. F.
1994-01-01
Resonance fluorescence from a two-state atom has been predicted to exhibit quadrature squeezing below the Heisenberg uncertainty limit, provided that the optical parameters (Rabi frequency, detuning, laser linewidth, etc.) are chosen carefully. When the correlation between two quadratures of the radiation field does not vanish, however, the Heisenberg limit for quantum fluctuations might be an unrealistic lower bound. A generalized uncertainty relation, due to Schroedinger, takes into account the possible correlation between the quadrature components of the radiation, and it suggests a modified definition of squeezing. We show that the coherence between the two levels of a laser-driven atom is responsible for the correlation between the quadrature components of the emitted fluorescence, and that the Schrodinger uncertainty limit increases monotonically with the coherence. On the other hand, the fluctuations in the quadrature field diminish with an increasing coherence, and can disappear completely when the coherence reaches 1/2, provided that certain phase relations hold.
Skin Friction Measurements by a Dual-Laser-Beam Interferometer Technique
NASA Technical Reports Server (NTRS)
Monson, D. J.; Higuchi, H.
1981-01-01
A portable dual-laser-beam interferometer that nonintrusively measures skin friction by monitoring the thickness change of an oil film subject to shear stress is described. The method is an advance over past versions in that the troublesome and error-introducing need to measure the distance to the oil leading edge and the starting time for the oil flow has been eliminated. The validity of the method was verified by measuring oil viscosity in the laboratory, and then using those results to measure skin friction beneath the turbulent boundary layer in a low speed wind tunnel. The dual-laser-beam skin friction measurements are compared with Preston tube measurements, with mean velocity profile data in a "law-of-the-well" coordinate system, and with computations based on turbulent boundary-layer theory. Excellent agreement is found in all cases. (This validation and the aforementioned improvements appear to make the present form of the instrument usable to measure skin friction reliably and nonintrusively in a wide range of flow situations in which previous methods are not practical.)
NASA Astrophysics Data System (ADS)
Fox, Maik; Beuth, Thorsten; Streck, Andreas; Stork, Wilhelm
2015-09-01
Homodyne laser interferometers for velocimetry are well-known optical systems used in many applications. While the detector power output signal of such a system, using a long coherence length laser and a single target, is easily modelled using the Doppler shift, scenarios with a short coherence length source, e.g. an unstabilized semiconductor laser, and multiple weak targets demand a more elaborated approach for simulation. Especially when using fiber components, the actual setup is an important factor for system performance as effects like return losses and multiple way propagation have to be taken into account. If the power received from the targets is in the same region as stray light created in the fiber setup, a complete system simulation becomes a necessity. In previous work, a phasor based signal simulation approach for interferometers based on short coherence length laser sources has been evaluated. To facilitate the use of the signal simulation, a fiber component ray tracer has since been developed that allows the creation of input files for the signal simulation environment. The software uses object oriented MATLAB code, simplifying the entry of different fiber setups and the extension of the ray tracer. Thus, a seamless way from a system description based on arbitrarily interconnected fiber components to a signal simulation for different target scenarios has been established. The ray tracer and signal simulation are being used for the evaluation of interferometer concepts incorporating delay lines to compensate for short coherence length.
Quadrature wavelength scanning interferometry.
Moschetti, Giuseppe; Forbes, Alistair; Leach, Richard K; Jiang, Xiang; O'Connor, Daniel
2016-07-10
A novel method to double the measurement range of wavelength scanning interferometery (WSI) is described. In WSI the measured optical path difference (OPD) is affected by a sign ambiguity, that is, from an interference signal it is not possible to distinguish whether the OPD is positive or negative. The sign ambiguity can be resolved by measuring an interference signal in quadrature. A method to obtain a quadrature interference signal for WSI is described, and a theoretical analysis of the advantages is reported. Simulations of the advantages of the technique and of signal errors due to nonideal quadrature are discussed. The analysis and simulation are supported by experimental measurements to show the improved performances. PMID:27409307
NASA Astrophysics Data System (ADS)
Gutierrez-Gutierrez, J.; Rojas-Laguna, R.; Estudillo-Ayala, J. M.; Sierra-Hernández, J. M.; Jauregui-Vazquez, D.; Vargas-Treviño, M.; Tepech-Carrillo, L.; Grajales-Coutiño, R.
2016-09-01
In this manuscript, switchable and multi-wavelength erbium-doped fiber laser arrangement, based on Fabry-Perot (FPI) and Mach-Zehnder (MZI) interferometers is presented. Here, the FPI is composed by two air-microcavities set into the tip of conventional single mode fiber, this one is used as a partially reflecting mirror and lasing modes generator. And the MZI fabricated by splicing a segment of photonic crystal fiber (PCF) between a single-mode fiber section, was set into an optical fiber loop mirror that acts as full-reflecting and wavelength selective filter. Both interferometers, promotes a cavity oscillation into the fiber laser configuration, besides by curvature applied over the MZI, the fiber laser generates: single, double, triple and quadruple laser emissions with a signal to noise ratio (SNR) of 30 dB. These laser emissions can be switching between them from 1525 nm to 1534 nm by adjusting the curvature radius over the MZI. This laser fiber offers a wavelength and power stability at room temperature, compactness and low implementation cost. Moreover the linear laser proposed can be used in several fields such as spectroscopy, telecommunications and fiber optic sensing systems.
NASA Astrophysics Data System (ADS)
Monien, H.
2010-04-01
Gaussian quadrature is a well-known technique for numerical integration. Recently Gaussian quadrature with respect to discrete measures corresponding to finite sums has found some new interest. In this paper we apply these ideas to infinite sums in general and give an explicit construction for the weights and abscissae of Gaussian formulas. The abscissae of the Gaussian summation have a very interesting asymptotic distribution function with a kink singularity. We apply the Gaussian summation technique to two problems which have been discussed in the literature. We find that the Gaussian summation has a very rapid convergence rate for the Hardy-Littlewood sum for a large range of parameters.
NASA Technical Reports Server (NTRS)
Ni, Wei-Tou; Shy, Jow-Tsong; Tseng, Shiao-Min; Shao, Michael
1992-01-01
A propasal to study the second order light deflection in the solar gravitational field is presented. It is proposed to use 1 to 2 W frequency stabilized lasers on two microspacecraft about 0.25 degree apart in the sky with apparent positions near the Sun, and observe the relative angle of two spacecraft using ground based fiber linked interferometers with 10 km baseline to determine the second order relativistic light deflection effects. The first two years of work would emphasize the establishment of a prototype stabilized laser system and fiber linked interferometer. The first year, a prototype fiber linked interferometer would be set up to study the phase noise produced by external perturbations to fiber links. The second year, a second interferometer would be set up. The cancellation of phase drift due to fiber links of both interferometers in the same environment would be investigated.
NASA Astrophysics Data System (ADS)
Chen, Weiguo; Lou, Shuqin; Wang, Liwen; Li, Honglei; Guo, Tieying; Jian, Shuisheng
2009-08-01
The compact Mach-Zehnder interferometer is proposed by splicing a section of photonic crystal fiber (PCF) and two pieces of single mode fiber (SMF) with the air-holes of PCF intentionally collapsed in the vicinity of the splices. The depedence of the fringe spacing on the length of PCF is investigated. Based on the Mach-Zehnder interferometer as wavelength-selective filter, a switchable dual-wavelength fiber ring laser is demonstrated with a homemade erbiumdoped fiber amplifier (EDFA) as the gain medium at room temperature. By adjusting the states of the polarization controller (PC) appropriately, the laser can be switched among the stable single-and dual -wavelength lasing operations by exploiting polarization hole burning (PHB) effect.
Observation of a critically refracted converted SP wave using laser Doppler interferometer
NASA Astrophysics Data System (ADS)
Gurevich, Boris; Lebedev, Maxim; Madadi, Mahyar; Bona, Andrej; Pevzner, Roman
2015-04-01
Laboratory measurements of elastic properties of rocks are important for calibration of seismic data and for corroboration of theoretical models of rocks. The most common way of determining the elastic properties of rock samples in laboratory settings is to estimate the velocities of ultrasonic waves propagating in different directions. The wave velocities are usually obtained from the travel times of waves generated and recorded by ultrasonic piezoelectric transducers. This approach has a large uncertainty associated with shear-wave travel time estimation and separation of differently polarised shear waves, as well as uncertainty as to whether phase or group velocity is measured. The problems are caused by the relatively large size and small number of transducers. One way to address some of these issues is by using laser Doppler interferometer, which records a particle like movement that can serve to separate the waves and to pick the travel times from which the ray velocities cab be estimated reliably, and with a huge data redundancy. In this paper, laser Doppler interferometer is used to record wave propagation in an anisotropic rock sample by measuring three orthogonal components of particle velocity on the sample surface. These measurements allow a clear separation of different wave types. The travel time of these waves are used for estimation of anisotropy parameters of the sample. A key observation is the very strong wave which at small offsets has traveltimes equal to those of the S-wave, but at large offsets travels with a velocity close to that of the P-wave. We interpret this wave as a converted SP wave critically refracted at the free surface. The nature and characteristics of this wave are confirmed by numerical simulations in both isotropic and anisotropic media. These simulations show the same traveltimes as measured in the experiment, but the amplitude of the converted SP wave is much stronger in the measured data. Analysis of this inconsistency is
Modeling of optical quadrature microscopy for imaging mouse embryos
NASA Astrophysics Data System (ADS)
Warger, William C., II; DiMarzio, Charles A.
2008-02-01
Optical quadrature microscopy (OQM) has been shown to provide the optical path difference through a mouse embryo, and has led to a novel method to count the total number of cells further into development than current non-toxic imaging techniques used in the clinic. The cell counting method has the potential to provide an additional quantitative viability marker for blastocyst transfer during in vitro fertilization. OQM uses a 633 nm laser within a modified Mach-Zehnder interferometer configuration to measure the amplitude and phase of the signal beam that travels through the embryo. Four cameras preceded by multiple beamsplitters record the four interferograms that are used within a reconstruction algorithm to produce an image of the complex electric field amplitude. Here we present a model for the electric field through the primary optical components in the imaging configuration and the reconstruction algorithm to calculate the signal to noise ratio when imaging mouse embryos. The model includes magnitude and phase errors in the individual reference and sample paths, fixed pattern noise, and noise within the laser and detectors. This analysis provides the foundation for determining the imaging limitations of OQM and the basis to optimize the cell counting method in order to introduce additional quantitative viability markers.
Mansfield, D.K.; Park, H.K.; Johnson, L.C.; Anderson, H.M.; Chouinard, R.; Foote, V.S.; Ma, C.H.; Clifton, B.J.
1987-07-01
A ten-channel far-infrared laser interferometer which is routinely used to measure the spatial and temporal behavior of the electron density profile on the TFTR tokamak is described and representative results are presented. This system has been designed for remote operation in the very hostile environment of a fusion reactor. The possible expansion of the system to include polarimetric measurements is briefly outlined. 13 refs., 8 figs.
NASA Astrophysics Data System (ADS)
Grabovskii, V. V.; Prokhorenko, V. I.; Yatskiv, D. Ya
1996-04-01
An experimental investigation was made of the spatial structure of the output beam profile of an Nd3+:YAG laser with a cavity based on a Sagnac interferometer. In a wide range of pulse repetition frequencies, the output beam profile could be described by a Gaussian curve with a confidence probability of at least 99%. This corresponded to the emission of the TEM00 mode from a stable cavity. The diameter of the radiation spot was inversely proportional to the shift of a thermo-optical lens from the centre of the Sagnac interferometer.
NASA Astrophysics Data System (ADS)
Grabovskii, V. V.; Prokhorenko, V. I.; Yatskiv, D. Ya
1995-04-01
An experimental investigation was made of the attainment of highly reproducible single-frequency emission from a pulsed laser based on a Sagnac interferometer. The probability of single-frequency emission was found to be related to the position of a passive switch relative to the optical centre of the interferometer. A qualitative theoretical analysis showed that the observed effect can be due to the scattering of waves on a grating induced in the switch. Experiments indicated that the probability of single-frequency emission was 99.99%.
Tanaka, K.; Sanin, A.L.; Vyacheslavov, L.N.; Akiyama, T.; Kawahata, K.; Tokuzawa, T.; Ito, Y.; Okajima, S.
2004-10-01
A multichannel imaging heterodyne interferometer is installed on Large Helical Device. Two branches of the interferometer: one with CO{sub 2} laser (wavelength {lambda}{sub i}=10.6 {mu}m) and other with diode pumped yttritium-aluminium-garnet (YAG) laser (wavelength {lambda}{sub i}=1.06 {mu}m) are used for electron density measurements and vibration compensation. Two slab beams and one circular beam are injected vertically. Liquid nitrogen cooled linear detector arrays are used for detection of slab beams and room temperature detector is employed for single circular beam for CO{sub 2} beams. Avalanche photodiodes record signals for YAG beams. Chordal resolution is from 15 to 22.5 mm, phase resolution is 10{sup -3} of CO{sub 2} fringe, which is determined by the electrical noise of phase counter. About 5x10{sup -3}{approx}10{sup -2} of CO{sub 2} fringe uncompensated signal remains after vibration compensation. Combining existing 13-channels far infrared (wavelength {lambda}{sub i}=119 {mu}m) laser interferometer, profile measurements at high density (>10{sup 20} m{sup -3}) become possible.
Xiong, C. Y.; Chen, J. Li, Q.; Liu, Y.; Gao, L.
2014-12-15
A three-wave laser polarimeter-interferometer, equipped with three independent far-infrared laser sources, has been developed on Joint-TEXT (J-TEXT) tokamak. The diagnostic system is capable of high-resolution temporal and phase measurement of the Faraday angle and line-integrated density. However, for long-term operation (>10 min), the free-running lasers can lead to large drifts of the intermediate frequencies (∼100–∼500 kHz/10 min) and decay of laser power (∼10%–∼20%/10 min), which act to degrade diagnostic performance. In addition, these effects lead to increased maintenance cost and limit measurement applicability to long pulse/steady state experiments. To solve this problem, a real-time feedback control method of the laser source is proposed. By accurately controlling the length of each laser cavity, both the intermediate frequencies and laser power can be simultaneously controlled: the intermediate frequencies are controlled according to the pre-set values, while the laser powers are maintained at an optimal level. Based on this approach, a real-time feedback control system has been developed and applied on J-TEXT polarimeter-interferometer. Long-term (theoretically no time limit) feedback of intermediate frequencies (maximum change less than ±12 kHz) and laser powers (maximum relative power change less than ±7%) has been successfully achieved.
Xiong, C Y; Chen, J; Li, Q; Liu, Y; Gao, L
2014-12-01
A three-wave laser polarimeter-interferometer, equipped with three independent far-infrared laser sources, has been developed on Joint-TEXT (J-TEXT) tokamak. The diagnostic system is capable of high-resolution temporal and phase measurement of the Faraday angle and line-integrated density. However, for long-term operation (>10 min), the free-running lasers can lead to large drifts of the intermediate frequencies (∼100-∼500 kHz/10 min) and decay of laser power (∼10%-∼20%/10 min), which act to degrade diagnostic performance. In addition, these effects lead to increased maintenance cost and limit measurement applicability to long pulse/steady state experiments. To solve this problem, a real-time feedback control method of the laser source is proposed. By accurately controlling the length of each laser cavity, both the intermediate frequencies and laser power can be simultaneously controlled: the intermediate frequencies are controlled according to the pre-set values, while the laser powers are maintained at an optimal level. Based on this approach, a real-time feedback control system has been developed and applied on J-TEXT polarimeter-interferometer. Long-term (theoretically no time limit) feedback of intermediate frequencies (maximum change less than ±12 kHz) and laser powers (maximum relative power change less than ±7%) has been successfully achieved. PMID:25554288
NASA Astrophysics Data System (ADS)
Xiong, C. Y.; Chen, J.; Li, Q.; Liu, Y.; Gao, L.
2014-12-01
A three-wave laser polarimeter-interferometer, equipped with three independent far-infrared laser sources, has been developed on Joint-TEXT (J-TEXT) tokamak. The diagnostic system is capable of high-resolution temporal and phase measurement of the Faraday angle and line-integrated density. However, for long-term operation (>10 min), the free-running lasers can lead to large drifts of the intermediate frequencies (˜100-˜500 kHz/10 min) and decay of laser power (˜10%-˜20%/10 min), which act to degrade diagnostic performance. In addition, these effects lead to increased maintenance cost and limit measurement applicability to long pulse/steady state experiments. To solve this problem, a real-time feedback control method of the laser source is proposed. By accurately controlling the length of each laser cavity, both the intermediate frequencies and laser power can be simultaneously controlled: the intermediate frequencies are controlled according to the pre-set values, while the laser powers are maintained at an optimal level. Based on this approach, a real-time feedback control system has been developed and applied on J-TEXT polarimeter-interferometer. Long-term (theoretically no time limit) feedback of intermediate frequencies (maximum change less than ±12 kHz) and laser powers (maximum relative power change less than ±7%) has been successfully achieved.
NASA Technical Reports Server (NTRS)
Garrison, T. J.; Settles, G. S.; Narayanswami, N.; Knight, D. D.
1994-01-01
Wall shear stress measurements beneath crossing-shock-wave/turbulent boundary-layer interactions have been made for three interactions of different strengths. The interactions are generated by two sharp fins at symetric angles of attack mounted on a flat plate. The shear stress measurements were made for fin angles of 7 and 11 deg at Mach 3 and 15 deg at Mach 3.85. The measurements were made using a laser interferometer skin-friction meter, a device that determines the wall shear by optically measuring the time rate of thinning of an oil film placed on the test model surface. Results of the measurements reveal high skin-friction coefficients in the vicinity of the fin/plate junction and the presence of quasi-two-dimensional flow separation on the interaction center line. Additionally, two Navier-Stokes computations, one using a Baldwin-Lomax turbulence model and one using a k-epsilon model, are compared with the experimental results for the Mach 3.85, 15-deg interaction case. Although the k-epsilon model did a reasonable job of predicting the overall trend in portions of the skin-friction distribution, neither computation fully captured the physics of the near-surface flow in this complex interaction.
Quadrature, Interpolation and Observability
NASA Technical Reports Server (NTRS)
Hodges, Lucille McDaniel
1997-01-01
Methods of interpolation and quadrature have been used for over 300 years. Improvements in the techniques have been made by many, most notably by Gauss, whose technique applied to polynomials is referred to as Gaussian Quadrature. Stieltjes extended Gauss's method to certain non-polynomial functions as early as 1884. Conditions that guarantee the existence of quadrature formulas for certain collections of functions were studied by Tchebycheff, and his work was extended by others. Today, a class of functions which satisfies these conditions is called a Tchebycheff System. This thesis contains the definition of a Tchebycheff System, along with the theorems, proofs, and definitions necessary to guarantee the existence of quadrature formulas for such systems. Solutions of discretely observable linear control systems are of particular interest, and observability with respect to a given output function is defined. The output function is written as a linear combination of a collection of orthonormal functions. Orthonormal functions are defined, and their properties are discussed. The technique for evaluating the coefficients in the output function involves evaluating the definite integral of functions which can be shown to form a Tchebycheff system. Therefore, quadrature formulas for these integrals exist, and in many cases are known. The technique given is useful in cases where the method of direct calculation is unstable. The condition number of a matrix is defined and shown to be an indication of the the degree to which perturbations in data affect the accuracy of the solution. In special cases, the number of data points required for direct calculation is the same as the number required by the method presented in this thesis. But the method is shown to require more data points in other cases. A lower bound for the number of data points required is given.
NASA Astrophysics Data System (ADS)
Lee, Won-Kyu; Park, Chang Yong; Mun, Jongchul; Yu, Dai-Hyuk
2011-07-01
The linewidth of a distributed-feedback (DFB) diode laser at 1156 nm, of which free-running linewidth was 3 MHz, was reduced to 15 kHz using an all-fiber interferometer with 5-m-long path imbalance. Optical power loss and bandwidth limitation were negligible with this short optical fiber patch cord. This result was achieved without acoustic and vibration isolations, and the frequency lock could be maintained over weeks. In addition to its simplicity, compactness, robustness, and cost-effectiveness, this technique can be applied at any wavelength owing to the availability of DFB diode lasers and fiber-optic components.
Lee, Won-Kyu; Park, Chang Yong; Mun, Jongchul; Yu, Dai-Hyuk
2011-07-01
The linewidth of a distributed-feedback (DFB) diode laser at 1156 nm, of which free-running linewidth was 3 MHz, was reduced to 15 kHz using an all-fiber interferometer with 5-m-long path imbalance. Optical power loss and bandwidth limitation were negligible with this short optical fiber patch cord. This result was achieved without acoustic and vibration isolations, and the frequency lock could be maintained over weeks. In addition to its simplicity, compactness, robustness, and cost-effectiveness, this technique can be applied at any wavelength owing to the availability of DFB diode lasers and fiber-optic components. PMID:21806170
Mandula, Gábor; Kis, Zsolt; Lengyel, Krisztián
2015-12-01
We report on a method for real-time dynamic calibration of a tunable external cavity diode laser by using a partially mode-matched plano-concave Fabry-Pérot interferometer in reflection geometry. Wide range laser frequency scanning is carried out by piezo-driven tilting of a diffractive grating playing the role of a frequency selective mirror in the laser cavity. The grating tilting system has a considerable mechanical inertness, so static laser frequency calibration leads to false results. The proposed real-time dynamic calibration based on the identification of primary- and Gouy-effect type secondary interference peaks with known frequency and temporal history can be used for a wide scanning range (from 0.2 GHz to more than 1 GHz). A concave spherical mirror with a radius of R = 100 cm and a plain 1% transmitting mirror was used as a Fabry-Pérot interferometer with various resonator lengths to investigate and demonstrate real-time calibration procedures for two kinds of laser frequency scanning functions. PMID:26724003
Real-time dynamic calibration of a tunable frequency laser source using a Fabry-Pérot interferometer
Mandula, Gábor Kis, Zsolt; Lengyel, Krisztián
2015-12-15
We report on a method for real-time dynamic calibration of a tunable external cavity diode laser by using a partially mode-matched plano-concave Fabry-Pérot interferometer in reflection geometry. Wide range laser frequency scanning is carried out by piezo-driven tilting of a diffractive grating playing the role of a frequency selective mirror in the laser cavity. The grating tilting system has a considerable mechanical inertness, so static laser frequency calibration leads to false results. The proposed real-time dynamic calibration based on the identification of primary- and Gouy-effect type secondary interference peaks with known frequency and temporal history can be used for a wide scanning range (from 0.2 GHz to more than 1 GHz). A concave spherical mirror with a radius of R = 100 cm and a plain 1% transmitting mirror was used as a Fabry-Pérot interferometer with various resonator lengths to investigate and demonstrate real-time calibration procedures for two kinds of laser frequency scanning functions.
Wang, Anting; Ming, Hai; Xie, Jianping; Chen, Xiyao; Lv, Liang; Huang, Wencai; Xu, Lixin
2003-06-20
A single-frequency Q-switched erbium-doped fiber ring laser is implemented for producing a single frequency with 25-Mhz laser linewidth, Q-switched pulses with a high peak power at 1557.5 mm. The single longitudinal mode is selected by a distributed Bragg reflector fiber laser, and a fiberoptic Mach-Zehnder interferometer acts as a Q-switch. The peak power and the average power of the Q-switched pulses vary with the repetition rate. PMID:12833955
Feldman, M.; Mockler, D.J.; English, R.E. Jr.; Byrd, J.L.; Salmon, J.T.
1991-02-01
We are incorporating a novel self-referencing Mach-Zehnder interferometer into a large scale laser system as a real time, interactive diagnostic tool for wavefront measurement. The instrument is capable of absolute wavefront measurements accurate to better than {lambda}/10 pv over a wavelength range > 300 nm without readjustment of the optical components. This performance is achieved through the design of both refractive optics and catadioptric collimator to achromatize the Mach-Zehnder reference arm. Other features include polarization insensitivity through the use of low angles of incidence on all beamsplitters as well as an equal path length configuration that allows measurement of either broad-band or closely spaced laser-line sources. Instrument accuracy is periodically monitored in place by means of a thermally and mechanically stable wavefront reference source that is calibrated off-line with a phase conjugate interferometer. Video interferograms are analyzed using Fourier transform techniques on a computer that includes dedicated array processor. Computer and video networks maintain distributed interferometers under the control of a single analysis computer with multiple user access. 7 refs., 11 figs.
NASA Astrophysics Data System (ADS)
Pustovoyt, V. I.
2016-07-01
In this paper we describe an observational method for determining black holes masses. The study shows that the knowledge of the recorded low-frequency gravitational waves and the period from the beginning of registration till the moment of black holes collapse is sufficient and even preferable in determining the possible values of collapsing black holes masses. The reason for this is that the proportion of the period in the measured interval containing relativistic corrections (i.e. those ones in which the black hole speed is comparable to the speed of light), is smaller if the observed and measured time interval is longer. The values of black holes masses and the measured time interval, obtained as a result of the first observations, according to this model are in a very good agreement.We examine the problem of mirror heating in Fabry-Perot cavity of Michelson interferometer, by incident laser radiation, and we conclude that the problem of heat removal can be solved by a different approach, without use of multilayer reflective openings. As an alternative approach to the creation of highly reflective structures, we suggest using a spatially extended structure with a sinusoidal variation of the refractive index. We consider some of the possible technological methods for producing such structures based on heterogeneous media.The article describes the effects of the incident laser radiation exposure on the periodic structure, and it shows that the volume ponderomotive force may lead to a mirror polarization due to the radiation, and consequently, to appearance of an additional mechanical connection of the mirror with the surrounding mirror suspension design. The article examines the impact of the surface ponderomotive forces on the media boundary with different dielectric permeability and it shows that pressure spatial variables arising at the same time lead to deformation of the media layers, and the deformation and pressure values depend on the difference in the
Liu, Yi; Qu, Shiliang
2014-01-20
The Fabry-Perot interferometer (FPI) cavity in a single-mode fiber with two open faces was fabricated by using the method of femtosecond laser-induced water breakdown. Then the FPI cavity was annealed by the arc discharge to greatly smooth its internal surface. The whole fabrication process features simple operation and high efficiency. The fabricated FPI cavity exhibits a perfect interferometer spectrum with reflection loss of only -3 dB and fringe visibility of almost 30 dB. It can be used as a perfectly reliable liquid refractive index sensor, as it exhibits high sensitivity (1147.48 nm/RIU), good linearity (99.93%), good repeatability, high actual measurement accuracy (1.29×10(-4)RIU), large measurement range, and good temperature insensitive characteristic. PMID:24514135
What Can be Expected from the GRACE-FO Laser Ranging Interferometer for Earth Science Applications?
NASA Astrophysics Data System (ADS)
Flechtner, Frank; Neumayer, Karl-Hans; Dahle, Christoph; Dobslaw, Henryk; Fagiolini, Elisa; Raimondo, Jean-Claude; Güntner, Andreas
2016-03-01
The primary objective of the gravity recovery and climate experiment follow-on (GRACE-FO) satellite mission, due for launch in August 2017, is to continue the GRACE time series of global monthly gravity field models. For this, evolved versions of the GRACE microwave instrument, GPS receiver, and accelerometer will be used. A secondary objective is to demonstrate the effectiveness of a laser ranging interferometer (LRI) in improving the satellite-to-satellite tracking measurement performance. In order to investigate the expected enhancement for Earth science applications, we have performed a full-scale simulation over the nominal mission lifetime of 5 years using a realistic orbit scenario and error assumptions both for instrument and background model errors. Unfiltered differences between the synthetic input and the finally recovered time-variable monthly gravity models show notable improvements with the LRI, on a global scale, of the order of 23 %. The gain is realized for wavelengths smaller than 240 km in case of Gaussian filtering but decreases to just a few percent when anisotropic filtering is applied. This is also confirmed for some typical regional Earth science applications which show randomly distributed patterns of small improvements but also degradations when using DDK4-filtered LRI-based models. Analysis of applied error models indicates that accelerometer noise followed by ocean tide and non-tidal mass variation errors are the main contributors to the overall GRACE-FO gravity model error. Improvements in these fields are therefore necessary, besides optimized constellations, to make use of the increased LRI accuracy and to significantly improve gravity field models from next-generation gravity missions.
Laser Interferometer Space Antenna double black holes: dynamics in gaseous nuclear discs
NASA Astrophysics Data System (ADS)
Dotti, Massimo; Colpi, Monica; Haardt, Francesco
2006-03-01
We study the inspiral of double black holes, with masses in the Laser Interferometer Space Antenna (LISA) window of detectability, orbiting inside a massive circumnuclear, rotationally supported gaseous disc. Using high-resolution smoothed particle hydrodynamics simulations, we follow the black hole dynamics in the early phase when gas-dynamical friction acts on the black holes individually, and continue our simulation until they form a close binary. We find that in the early sinking the black holes lose memory of their initial orbital eccentricity if they corotate with the gaseous disc. As a consequence, the massive black holes bind forming a binary with a low eccentricity, consistent with zero within our numerical resolution limit. The cause of circularization resides in the rotation present in the gaseous background where dynamical friction operates. Circularization may hinder gravitational waves from taking over and leading the binary to coalescence. In the case of counter-rotating orbits, the initial eccentricity (if present) does not decrease, and the black holes may bind forming an eccentric binary. When dynamical friction has subsided, for equal mass black holes and regardless their initial eccentricity, angular momentum loss, driven by the gravitational torque exerted on the binary by surrounding gas, is nevertheless observable down to the smallest scale probed (~=1 pc). In the case of unequal masses, dynamical friction remains efficient down to our resolution limit, and there is no sign of formation of any ellipsoidal gas distribution that may further harden the binary. During inspiral, gravitational capture of gas by the black holes occurs mainly along circular orbits; eccentric orbits imply high relative velocities and weak gravitational focusing. Thus, the active galactic nucleus activity may be excited during the black hole pairing process and double active nuclei may form when circularization is completed, on distance scales of tens of parsecs.
NASA Astrophysics Data System (ADS)
Coffer, J. G.; Bernard, J. M.; Chodzko, R. A.; Turner, E. B.; Gross, R. W. F.; Warren, W. R.
1983-01-01
Active phase matching of multiline HF laser beams by means of a phase-locked Mach-Zehnder interferometer was demonstrated by locking the interferometer to the central interference fringe at zero optical path length difference. The central fringe could be found by varying the spectral content of the input beam. Laser amplification in one leg of the interferometer decreased fringe visibility without adversely affecting locking. Single-line fringe patterns produced by an array spectrometer (while the interferometer was operated in its scanning mode) were analyzed to show that no significant dispersion occurred in the amplifier. The techniques developed have potential for measuring dispersion mismatch between larger parallel amplifiers. These experiments demonstrated in principle that a number of multiline HF amplified beams can be recombined and phase-matched to produce a high beam quality output beam.
VISAR (Velocity Interferometer System for Any Reflector): Line-imaging interferometer
Hemsing, W.F.; Mathews, A.R.; Warnes, R.H.; Whittemore, G.R.
1990-01-01
This paper describes a Velocity Interferometer System for Any Reflector (VISAR) technique that extends velocity measurements from single points to a line. Single-frequency argon laser light was focused through a cylindrical lens to illuminate a line on a surface. The initially stationary, flat surface was accelerated unevenly during the experiment. Motion produced a Doppler-shift of light reflected from the surface that was proportional to the velocity at each point. The Doppler-shifted image of the illuminated line was focused from the surface through a push-pull VISAR interferometer where the light was split into four quadrature-coded images. When the surface accelerated, the Doppler-shift caused the interference for each point on each line image to oscillate sinusoidally. Coherent fiber optic bundles transmitted images from the interferometer to an electronic streak camera for sweeping in time and recording on film. Data reduction combined the images to yield a continuous velocity and displacement history for all points on the surface that reflected sufficient light. The technique was demonstrated in an experiment where most of the surface was rapidly driven to a saddle shape by an exploding foil. Computer graphics were used to display the measured velocity history and to aid visualization of the surface motion. 6 refs., 8 figs.
NASA Astrophysics Data System (ADS)
Warger, William C., II; Newmark, Judith A.; Chang, ChihChing; Brooks, Dana H.; Warner, Carol M.; DiMarzio, Charles A.
2005-03-01
The Multifunctional Staring Mode Microscope was developed to permit three modes of imaging for cell counting in mouse embryos: Optical Quadrature, Differential Interference Contrast (DIC), and Fluorescence Imaging. The Optical Quadrature Microscope, consisting of a modified Mach-Zender Interferometer, uses a 632.8 nm laser to measure the amplitude and phase of the signal beam that travels through the embryo. Four cameras, preceded by multiple beamsplitters, are used to read the four interferograms, which are then combined to produce an image of the complex electric field amplitude. The phase of the complex amplitude is then unwrapped using a 2-D phase unwrap algorithm and images of optical path length are produced. To combine the additional modes of DIC and Fluorescence Imaging with the Optical Quadrature Microscope, a 632.8 nm narrow bandpass beamsplitter was placed at the output of the microscope. This allows the laser light to continue through the Mach-Zender while all other wavelengths are reflected at 90 degrees to another camera. This was effective in combining the three modes as the fluorescence wavelength for the Hoechst stain is well below the bandpass window of the beamsplitter. Both live and fixed samples have been successfully imaged in all three modes. Accuracy in cell counting was achieved by using the DIC image for detecting cell boundaries and the Optical Quadrature image for phase mapping to determine where cells overlap. The final results were verified by Hoechst fluorescence imaging to count the individual nuclei. Algorithms are currently being refined so larger cell counts can be done more efficiently.
NASA Astrophysics Data System (ADS)
Bouyer, P.
2015-12-01
Since its first demonstration in 1991, Atomic Interferometry (AI) has shown to be an extremely performing probe of inertial forces. More recently, AI has revealed sensitivities to acceleration or rotation competing with or even beating state-of-the art sensors based on other technologies. The high stability and accuracy of AI sensors relying on cold atoms is at the basis of several applications ranging from fundamental physics (e.g. tests of general relativity and measurements of fundamental constants), geophysics (gravimetry, gradiometry) and inertial navigation. We are currently building a large scale matter-wave detector which will open new applications in geoscience and fundamental physics. In contrast to standard AI based sensors, our matter-wave laser interferometer gravitation antenna (MIGA) exploits the superb seismic environment of a low noise underground laboratory. This new infrastructure is embedded into the LSBB underground laboratory, in France, ideally located away from major anthropogenic disturbances and benefitting from very low background noise. MIGA combines atom and laser interferometry techniques, manipulating an array of atomic ensembles distributed along the antenna to simultanously read out seismic effects, inertial effects and eventually the passage of a gravity wave. The first version uses a set of three atomic sensors placed along an optical cavity. The spatial resolution obtained with this configuration will enable the separation of the seismic, inertial and GW contributions. This technique will bring unprecedented sensitivities to gravity gradients variations and open new perspectives for sub Hertz gravity wave and geodesic detection. MIGA will provide measurements of gravity gradients variations limited only by the AI shot noise, which will allow sensitivities of about 10-13 s-2Hz-1/2@ 2Hz. This instrument will then be capable to spatially resolve 1 m3 of water a distances of about 100 m, which opens important potential applications
Bamford, D. J.; Cummings, E. A.; Panasenko, D.; Fenner, D. B.; Hensley, J. M.; Boivin, R. L.; Carlstrom, T. N.; Van Zeeland, M. A.
2013-09-15
A dispersion interferometer based on the second-harmonic generation of a carbon dioxide laser in orientation-patterned gallium arsenide has been developed for measuring electron density in plasmas. The interferometer includes two nonlinear optical crystals placed on opposite sides of the plasma. This instrument has been used to measure electron line densities in a pulsed radio-frequency generated argon plasma. A simple phase-extraction technique based on combining measurements from two successive pulses of the plasma has been used. The noise-equivalent line density was measured to be 1.7 × 10{sup 17} m{sup −2} in a detection bandwidth of 950 kHz. One of the orientation-patterned crystals produced 13 mW of peak power at the second-harmonic wavelength from a carbon dioxide laser with 13 W of peak power. Two crystals arranged sequentially produced 58 mW of peak power at the second-harmonic wavelength from a carbon dioxide laser with 37 W of peak power.
NASA Astrophysics Data System (ADS)
Zou, Hui; Lou, Shuqin; Su, Wei; Wang, Xin
2013-09-01
A dual-pass Mach-Zehnder interferometer filter using a section of twin-core fiber (TCF) loop mirror is proposed. The filter is theoretically and experimentally studied for various interferometer arm difference when TCF length is constant. Theoretical results are validated by the experimental demonstration and in good agreement with the experimental results. And then, by using the filter in a ring fiber laser, a stable and switchable dual-wavelength lasing is obtained experimentally. The 3-dB bandwidth and the SMSR of the output laser are 0.015 nm and higher than 62.4 dB, respectively. The peak power fluctuation and wavelength shift are also monitored to be less than 0.04 dB and 0.02 nm over an hour at room temperature. Furthermore, the output laser can be switched between single and dual wavelength by carefully adjusting the PCs. The experimental results show that the filter can suppress mode competition effectively, improve the SMSR availably, and enhance the stability of the output lasing.
Bamford, D J; Cummings, E A; Panasenko, D; Fenner, D B; Hensley, J M; Boivin, R L; Carlstrom, T N; Van Zeeland, M A
2013-09-01
A dispersion interferometer based on the second-harmonic generation of a carbon dioxide laser in orientation-patterned gallium arsenide has been developed for measuring electron density in plasmas. The interferometer includes two nonlinear optical crystals placed on opposite sides of the plasma. This instrument has been used to measure electron line densities in a pulsed radio-frequency generated argon plasma. A simple phase-extraction technique based on combining measurements from two successive pulses of the plasma has been used. The noise-equivalent line density was measured to be 1.7 × 10(17) m(-2) in a detection bandwidth of 950 kHz. One of the orientation-patterned crystals produced 13 mW of peak power at the second-harmonic wavelength from a carbon dioxide laser with 13 W of peak power. Two crystals arranged sequentially produced 58 mW of peak power at the second-harmonic wavelength from a carbon dioxide laser with 37 W of peak power. PMID:24089824
Hilbert, V.; Rödel, C.; Zastrau, U.; Brenner, G.; Düsterer, S.; Dziarzhytski, S.; Harmand, M.; Przystawik, A.; Redlin, H.; Toleikis, S.; Döppner, T.; Ma, T.; Fletcher, L.; Förster, E.; Glenzer, S. H.; Lee, H. J.; Hartley, N. J.; Kazak, L.; Komar, D.; Skruszewicz, S.; and others
2014-09-08
A key feature of extreme ultraviolet (XUV) radiation from free-electron lasers (FELs) is its spatial and temporal coherence. We measured the spatio-temporal coherence properties of monochromatized FEL pulses at 13.5 nm using a Michelson interferometer. A temporal coherence time of (59±8) fs has been determined, which is in good agreement with the spectral bandwidth given by the monochromator. Moreover, the spatial coherence in vertical direction amounts to about 15% of the beam diameter and about 12% in horizontal direction. The feasibility of measuring spatio-temporal coherence properties of XUV FEL radiation using interferometric techniques advances machine operation and experimental studies significantly.
NASA Astrophysics Data System (ADS)
Sun, Xiao-Yan; Chu, Dong-Kai; Dong, Xin-Ran; Zhou, Chu; Li, Hai-Tao; Luo-Zhi; Hu, You-Wang; Zhou, Jian-Ying; Cong-Wang; Duan, Ji-An
2016-03-01
A High sensitive refractive index (RI) sensor based on Mach-Zehnder interferometer (MZI) in a conventional single-mode optical fiber is proposed, which is fabricated by femtosecond laser transversal-scanning inscription method and chemical etching. A rectangular cavity structure is formed in part of fiber core and cladding interface. The MZI sensor shows excellent refractive index sensitivity and linearity, which exhibits an extremely high RI sensitivity of -17197 nm/RIU (refractive index unit) with the linearity of 0.9996 within the refractive index range of 1.3371-1.3407. The experimental results are consistent with theoretical analysis.
Bravo, Mikel; Fernández-Vallejo, Montserrat; Echapare, Mikel; López-Amo, Manuel; Kobelke, J; Schuster, K
2013-02-11
This work experimentally demonstrates a long-range optical fiber sensing network for the multiplexing of fiber sensors based on photonic crystal fibers. Specifically, six photonic crystal fiber sensors which are based on a Sagnac interferometer that includes a suspended-core fiber have been used. These sensors offer a high sensitivity for micro-displacement measurements. The fiber sensor network presents a ladder structure and its operation mode is based on a fiber ring laser which combines Raman and Erbium doped fiber amplification. Thus, we show the first demonstration of photonic crystal fiber sensors for remote measurement applications up to 75 km. PMID:23481755
Digital quadrature phase detection
Smith, J.A.; Johnson, J.A.
1992-05-26
A system for detecting the phase of a frequency or phase modulated signal that includes digital quadrature sampling of the frequency or phase modulated signal at two times that are one quarter of a cycle of a reference signal apart, determination of the arctangent of the ratio of a first sampling of the frequency or phase modulated signal to the second sampling of the frequency or phase modulated signal, and a determination of quadrant in which the phase determination is increased by 2[pi] when the quadrant changes from the first quadrant to the fourth quadrant and decreased by 2[pi] when the quadrant changes from the fourth quadrant to the first quadrant whereby the absolute phase of the frequency or phase modulated signal can be determined using an arbitrary reference convention. 6 figs.
Digital quadrature phase detection
Smith, James A.; Johnson, John A.
1992-01-01
A system for detecting the phase of a frequency of phase modulated signal that includes digital quadrature sampling of the frequency or phase modulated signal at two times that are one quarter of a cycle of a reference signal apart, determination of the arctangent of the ratio of a first sampling of the frequency or phase modulated signal to the second sampling of the frequency or phase modulated signal, and a determination of quadrant in which the phase determination is increased by 2.pi. when the quadrant changes from the first quadrant to the fourth quadrant and decreased by 2.pi. when the quadrant changes from the fourth quadrant to the first quadrant whereby the absolute phase of the frequency or phase modulated signal can be determined using an arbitrary reference convention.
Optimized quadrature surface coil designs
Kumar, Ananda; Bottomley, Paul A.
2008-01-01
Background Quadrature surface MRI/MRS detectors comprised of circular loop and figure-8 or butterfly-shaped coils offer improved signal-to-noise-ratios (SNR) compared to single surface coils, and reduced power and specific absorption rates (SAR) when used for MRI excitation. While the radius of the optimum loop coil for performing MRI at depth d in a sample is known, the optimum geometry for figure-8 and butterfly coils is not. Materials and methods The geometries of figure-8 and square butterfly detector coils that deliver the optimum SNR are determined numerically by the electromagnetic method of moments. Figure-8 and loop detectors are then combined to create SNR-optimized quadrature detectors whose theoretical and experimental SNR performance are compared with a novel quadrature detector comprised of a strip and a loop, and with two overlapped loops optimized for the same depth at 3 T. The quadrature detection efficiency and local SAR during transmission for the three quadrature configurations are analyzed and compared. Results The SNR-optimized figure-8 detector has loop radius r8 ∼ 0.6d, so r8/r0 ∼ 1.3 in an optimized quadrature detector at 3 T. The optimized butterfly coil has side length ∼ d and crossover angle of ≥ 150° at the center. Conclusions These new design rules for figure-8 and butterfly coils optimize their performance as linear and quadrature detectors. PMID:18057975
Fiber-optic gas pressure sensing with a laser-heated silicon-based Fabry-Perot interferometer.
Liu, Guigen; Han, Ming
2015-06-01
We report a novel fiber-optic sensor for measurement of static gas pressure based on the natural convection of a heated silicon pillar attached to a fiber tip functioning as a Fabry-Perot interferometer (FPI). A visible laser beam is guided by the fiber to efficiently heat the silicon pillar, while an infrared whitelight source, also guided by the fiber, is used to measure the temperature of the FPI, which is influenced both by the laser power and the pressure through natural convection. We theoretically and experimentally show that, by monitoring the fringe shift caused by the laser heating, air pressure sensing with little temperature cross-sensitivity can be achieved. The pressure sensitivity can be easily tuned by adjusting the heating laser power. In our experiment, the sensor performance within the temperature range from 20°C to 50°C and the pressure range from 0 to 1400 psi has been characterized, showing an average sensitivity of -0.52 pm/psi. Compared to the passive version of the sensor, the pressure sensitivity was ∼15 times larger, and the temperature cross-sensitivity was ∼100 times smaller. PMID:26030532
Pardue, Robert M.; Williams, Richard R.
1982-01-01
A double-pass interferometer is provided which allows direct measurement of relative displacement between opposed surfaces. A conventional plane mirror interferometer may be modified by replacing the beam-measuring path cube-corner reflector with an additional quarter-wave plate. The beam path is altered to extend to an opposed plane mirrored surface and the reflected beam is placed in interference with a retained reference beam split from dual-beam source and retroreflected by a reference cube-corner reflector mounted stationary with the interferometer housing. This permits direct measurement of opposed mirror surfaces by laser interferometry while doubling the resolution as with a conventional double-pass plane mirror laser interferometer system.
Pardue, R.M.; Williams, R.R.
1980-09-12
A double-pass interferometer is provided which allows direct measurement of relative displacement between opposed surfaces. A conventional plane mirror interferometer may be modified by replacing the beam-measuring path cube-corner reflector with an additional quarterwave plate. The beam path is altered to extend to an opposed plane mirrored surface and the reflected beam is placed in interference with a retained reference beam split from dual-beam source and retroreflected by a reference cube-corner reflector mounted stationary with the interferometer housing. This permits direct measurement of opposed mirror surfaces by laser interferometry while doubling the resolution as with a conventional double-pass plane mirror laser interferometer system.
The Palomar Testbed Interferometer
NASA Technical Reports Server (NTRS)
Colavita, M. M.; Wallace, J. K.; Hines, B. E.; Gursel, Y.; Malbet, F.; Palmer, D. L.; Pan, X. P.; Shao, M.; Yu, J. W.; Boden, A. F.
1999-01-01
The Palomar Testbed Interferometer (PTI) is a long-baseline infrared interferometer located at Palomar Observatory, California. It was built as a testbed for interferometric techniques applicable to the Keck Interferometer. First fringes were obtained in 1995 July. PTI implements a dual-star architecture, tracking two stars simultaneously for phase referencing and narrow-angle astrometry. The three fixed 40 cm apertures can be combined pairwise to provide baselines to 110 m. The interferometer actively tracks the white-light fringe using an array detector at 2.2 microns and active delay lines with a range of +/-38 m. Laser metrology of the delay lines allows for servo control, and laser metrology of the complete optical path enables narrow-angle astrometric measurements. The instrument is highly automated, using a multiprocessing computer system for instrument control and sequencing.
Thrane, Eric; Mandic, Vuk; Ballmer, Stefan; Romano, Joseph D.; Mitra, Sanjit; Talukder, Dipongkar; Bose, Sukanta
2009-12-15
We present a maximum-likelihood analysis for estimating the angular distribution of power in an anisotropic stochastic gravitational-wave background using ground-based laser interferometers. The standard isotropic and gravitational-wave radiometer searches (optimal for point sources) are recovered as special limiting cases. The angular distribution can be decomposed with respect to any set of basis functions on the sky, and the single-baseline, cross-correlation analysis is easily extended to a network of three or more detectors--that is, to multiple baselines. A spherical-harmonic decomposition, which provides maximum-likelihood estimates of the multipole moments of the gravitational-wave sky, is described in detail. We also discuss (i) the covariance matrix of the estimators and its relationship to the detector response of a network of interferometers, (ii) a singular-value decomposition method for regularizing the deconvolution of the detector response from the measured sky map, (iii) the expected increase in sensitivity obtained by including multiple baselines, and (iv) the numerical results of this method when applied to simulated data consisting of both pointlike and diffuse sources. Comparisons between this general method and the standard isotropic and radiometer searches are given throughout, to make contact with the existing literature on stochastic background searches.
Output power stability of a HCN laser using a stepping motor for the EAST interferometer system
NASA Astrophysics Data System (ADS)
Zhang, J. B.; Wei, X. C.; Liu, H. Q.; Shen, J. J.; Zeng, L.; Jie, Y. X.
2015-11-01
The HCN laser on EAST is a continuous wave glow discharge laser with 3.4 m cavity length and 120 mW power output at 337 μ m wavelength. Without a temperature-controlled system, the cavity length of the laser is very sensitive to the environmental temperature. An external power feedback control system is applied on the HCN laser to stabilize the laser output power. The feedback system is composed of a stepping motor, a PLC, a supervisory computer, and the corresponding control program. One step distance of the stepping motor is 1 μ m and the time response is 0.5 s. Based on the power feedback control system, a stable discharge for the HCN laser is obtained more than eight hours, which satisfies the EAST experiment.
NASA Astrophysics Data System (ADS)
Kitano, Takuya; Nasir Uddin, Mohammad; Hong, Bingzhou; Tajima, Akio; Jiang, Haisong; Hamamoto, Kiichi
2016-08-01
The recent rapid growth of data traffic is leading to high-speed communication for local areas, such as the fiber-to-the-home service. A semiconductor laser is used for such a purpose; however, there is the difficulty that an even higher frequency response occurs in only carrier-photon resonance. For this reason, it is effective to use a second resonance, such as a photon–photon resonance (PPR), for enhancing the frequency response, and the active multimode interferometer laser diode (active-MMI LD) is one of the candidates for achieving a high PPR frequency. In order to obtain an even higher PPR frequency, we have investigated the control scheme of enhancing PPR. In this work, we compared two types of active-MMI waveguide structures to confirm the scheme. As a result, a 3.8 GHz enhancement of the PPR peak, resulting in a 3 dB lower frequency response of 17 GHz, has been successfully achieved by waveguide geometry modification.
NASA Astrophysics Data System (ADS)
Sanjuán, J.; Preston, A.; Korytov, D.; Spector, A.; Freise, A.; Dixon, G.; Livas, J.; Mueller, G.
2011-12-01
The laser interferometer space antenna (LISA) is a mission designed to detect low frequency gravitational waves. In order for LISA to succeed in its goal of direct measurement of gravitational waves, many subsystems must work together to measure the distance between proof masses on adjacent spacecraft. One such subsystem, the telescope, plays a critical role as it is the laser transmission and reception link between spacecraft. Not only must the material that makes up the telescope support structure be strong, stiff, and light, but it must have a dimensional stability of better than 1 pm Hz-1/2 at 3 mHz and the distance between the primary and the secondary mirrors must change by less than 2.5 μm over the mission lifetime. Carbon fiber reinforced polymer is the current baseline material; however, it has not been tested to the pico meter level as required by the LISA mission. In this paper, we present dimensional stability results, outgassing effects occurring in the cavity and discuss its feasibility for use as the telescope spacer for the LISA spacecraft.
NASA Astrophysics Data System (ADS)
Ishige, Masashi; Aketagawa, Masato; Banh Quoc, Tuan; Hoshino, Yuta
2009-08-01
We present a method for air-refractive-index (nair) fluctuation measurement using a laser interferometer. The method is based on a combination of a phase modulation homodyne interferometer (PMHI), an external cavity laser diode (ECLD) and an ultralow thermal expansion material (ULTEM). The PMHI utilizes a Michelson interferometer which is constructed on the ULTEM plate under the condition of an air temperature fluctuation of less than 10 mK, so that the optical path change or the air-refractive-index fluctuation (Δnair) caused by the thermal disturbance can be neglected. Meanwhile, the ECLD is controlled by adjusting its frequency to track some of the dark fringes of the interferometer, so that Δnair can be derived from the ECLD frequency change. The uncertainty of the Δnair measurement in the experiment is of 10-8 order. However, it will be possible to decrease the uncertainty to 10-9 or less if the signal-to-noise ratio (SNR) of the control system is improved.
The improved pyroelectric detectors for far-infrared laser interferometer measuring
NASA Astrophysics Data System (ADS)
Xiang, Gao
1990-05-01
In this paper, the application of the pyroelectric detectors for Far-Infrared laser diagnostics on TOKAMAK plasma is described. We discovered experimentally that the Fabry-Perot interference could affect the performance of the pyroelectric detectors (PED). The improved pyroelectric detector (IPD) was developed for FIR laser coheront measuring. Some designing considerations about the pyroelectric detectors used in high temperature plasma conditions are mentioned.
Mode-locked fiber laser using the Sagnac interferometer and the nonlinear polarization rotation
NASA Astrophysics Data System (ADS)
Ibarra-Escamilla, Baldeamr; Kuzin, Evgueni A.; Haus, Joseph W.; Pottiez, Olivier; Gomez-Garcia, Dario E.; Gutierrez-Zainos, Francisco; Mendoza-Vazquez, Sergio; Grajales-Coutico, Ruben
2003-07-01
In figure-eight lasers (F8L) mode locking is achieved through a nonlinear fiber amplifier loop mirror (NALM) or an asymmetrical nonlinear optical loop mirror (NOLM). Recently, we have theoretically shown that the symmetrical NOLM with a twisted fiber is useful for passive mode locking of fiber lasers. In this work we experimentally demonstrate the operation of a F8L based on the symmetrical NOLM with a twisted low-birefringence fiber in the loop. The modelocking operation is achieved by nonlinear polarization rotation. We found that the counter-propagating beams accumulate a differential nonlinear phase shift when they have different As (where As is the Stokes parameter). At the input NOLM, we used a polarizer controller to adjust the clockwise beam to be circularly polarized, As=1. In the loop of the NOLM, we used a quarter-wave retarder to transform the counter-clockwise beam to linear polarization, As=0. The quarter-wave retarder was the only element that we adjust to achieve modelocking. The pulse repetition frequency was 0.8 MHz. The FWHM of the autocorrelation function was 0.7 ps. We used a pump power of 80 mW to get the modelocking operation. The modelocked laser ran in stable operation for hours. Even in this first experiments the laser shown several advantages. The adjustment procedure was straightforward. The laser shows stable operation and exhibits high pulse energy. We achieved stable generation of subpicosecond pulses with milliwatts of average output power.
Rotatable shear plate interferometer
Duffus, Richard C.
1988-01-01
A rotatable shear plate interferometer comprises a transparent shear plate mounted obliquely in a tubular supporting member at 45.degree. with respect to its horizontal center axis. This tubular supporting member is supported rotatably around its center axis and a collimated laser beam is made incident on the shear plate along this center axis such that defocus in different directions can be easily measured.
NASA Technical Reports Server (NTRS)
Garrison, T. J.; Settles, G. S.
1993-01-01
Wall shear stress measurements beneath crossingshock wave/turbulent boundary-layer interactions have been made for three interactions of different strengths. The interactions are generated by two sharp fins at symmetric angles of attack mounted on a flat plate. The shear stress measurements were made for fin angles of 7 and 11 degrees at Mach 3 and 15 degrees at Mach 4. The measurements were made using a Laser Interferometer Skin Friction (LISF) meter; a device which determines the wail shear by optically measuring the time rate of thinning of an oil film placed on the test model surface. Results of the measurements reveal high skin friction coefficients in the vicinity of the fin/plate junction and the presence of quasi-two-dimensional flow separation on the interaction centerline. Additionally, two Navier-Stokes computations, one using a Baldwin-Lomax turbulence model and one using a k- model, are compared to the experimental results for the Mach 4, 15 degree interaction case. While the k- model did a reasonable job of predicting the overall trend in portions of the skin friction distribution, neither computation fully captured the physics of the near surface flow in this complex interaction.
Sanjuán, J; Lobo, A; Ramos-Castro, J
2009-11-01
Low-noise temperature measurements at frequencies in the millihertz range are required in the laser interferometer space antenna (LISA) and LISA PathFinder missions. The required temperature stability for LISA is around 10 microK Hz(-1/2) at frequencies down to 0.1 mHz. In this paper we focus on the identification and reduction in a source of excess noise detected when measuring time-varying temperature signals. This is shown to be due to nonidealities in the analog-to-digital converter (ADC) transfer curve, and degrades the measurement by about one order of magnitude in the measurement bandwidth when the measured temperature drifts by a few approximately microK s(-1). In a suitable measuring system for the LISA mission, this noise needs to be reduced. Two different methods based on the same technique have been implemented, both consisting in the addition of dither signals out of band to mitigate the ADC nonideality errors. Excess noise of this nature has been satisfactorily reduced by using these methods when measuring temperature ramps up to 10 microK s(-1). PMID:19947750
Zheng, Mei C. Gmachl, Claire F.; Liu, Peter Q.; Wang, Xiaojun; Fan, Jen-Yu; Troccoli, Mariano
2013-11-18
We report on the experimental demonstration of a widely tunable single mode quantum cascade laser with Asymmetric Mach-Zehnder (AMZ) interferometer type cavities with separately biased arms. Current and, consequently, temperature tuning of the two arms of the AMZ type cavity resulted in a single mode tuning range of 20 cm{sup −1} at 80 K in continuous-wave mode operation, a ten-fold improvement from the lasers under a single bias current. In addition, we also observed a five fold increase in the tuning rate as compared to the AMZ cavities controlled by one bias current.
Bunkenburg, J.; Kessler, T.J.; Skulski, W.; Huang, H.
2006-04-27
Multikilojoule petawatt lasers using chirped-pulse amplification are being constructed worldwide. Several systems have adopted a tiled-grating approach to meet the size requirements for the compression gratings. Grating tiles need to be precisely phased to ensure a transform-limited focal spot when focusing high-energy laser pulses in the target plane. A computer-control test system that uses a Mach-Zehnder interferometer capable of monitoring and correcting drift in the tiled-grating assemblies within the compressor is described.
NASA Astrophysics Data System (ADS)
Feng, Suchun; Xu, Ou; Lu, Shaohua; Ning, Tigang; Jian, Shuisheng
2008-12-01
A switchable multi-wavelength erbium-doped fiber (EDF) ring laser based on cascaded polarization maintaining fiber Bragg gratings (PMFBGs) in a Sagnac loop interferometer as the wavelength-selective filter at room temperature is proposed. Due to the polarization hole burning (PHB) enhanced by the PMFBGs, stable single-, dual-, three- and four-wavelength lasing operations can be obtained. The laser can be switched among the stable single-, dual-, three- and four-wavelength lasing operations by adjusting the polarization controllers (PCs). The optical signal-to-noise ratio (OSNR) is over 50 dB.
NASA Astrophysics Data System (ADS)
Chen, W. G.; Lou, S. Q.; Feng, S. C.; Wang, L. W.; Li, H. L.; Guo, T. Y.; Jian, S. S.
2009-11-01
Switchable multi-wavelength fiber ring laser with an in-fiber Mach-Zehnder interferometer incorporated into the ring cavity serving as wavelength-selective filter at room temperature is demonstrated. The filter is formed by splicing a section of few-mode photonic crystal fiber (PCF) and two segments of single mode fiber (SMF) with the air-holes on the both sides of PCF intentionally collapsed in the vicinity of the splices. By adjusting the states of the polarization controller (PC) appropriately, the laser can be switched among the stable single-, dual- and triple-wavelength lasing operations by exploiting polarization hole burning (PHB) effect.
NASA Astrophysics Data System (ADS)
Feng, Suchun; Li, Honglei; Xu, Ou; Lu, Shaohua; Mao, Xiangqiao; Ning, Tigang; Jian, Shuisheng
2008-11-01
Dual-wavelength with orthogonal polarizations erbium-doped fiber ring laser at room temperature is proposed. One polarization-maintaining fiber Bragg grating (PMFBG) in a Sagnac loop interferometer is used as the wavelength-selective filter. Due to the polarization hole burning (PHB) enhanced by the PMFBG, the laser can operate in stable dual-wavelength operation with wavelength spacing of 0.336 nm at room temperature by adjusting a polarization controller (PC). The optical signal-to-noise ratio (OSNR) is over 52 dB. The amplitude variation in nearly one and half an hour is less than 0.6 dB for both wavelengths.
Fiber laser strain sensor based in the measurement of a Sagnac interferometer optical power spectrum
NASA Astrophysics Data System (ADS)
Durán Sánchez, M.; Álvarez Tamayo, R. I.; Pottiez, O.; Kuzin, E. A.; Ibarra-Escamilla, B.; Barcelata Pinzón, A.
2014-06-01
In this paper a linear cavity Erbium doped fiber (EDF) laser based in a fiber Bragg grating (FBG) and a fiber optical loop mirror with a high birefringence fiber in the loop (Hi-Bi FOLM) is used as a strain sensor. The Fabry-Perot cavity is formed by the FBG and the Hi-Bi FOLM, used as a measurement system of strain variations produced on the FBG, used as a strain sensor device. Usually, fiber laser sensor experimental setups determine the measured variable magnitude by using of an optical spectrum analyzer (OSA). Hi-Bi FOLM transmission spectrum wavelength displacement by fiber loop temperature variations measurement can be an attractive application exploiting the characteristics of FOLM transmission spectrum behavior due to Hi-Bi fiber loop temperature variations to determine the FBG strain applied through the maximal optical power monitoring by simple use of a photodetector and a temperature meter.
Length Scales in Bayesian Automatic Adaptive Quadrature
NASA Astrophysics Data System (ADS)
Adam, Gh.; Adam, S.
2016-02-01
Two conceptual developments in the Bayesian automatic adaptive quadrature approach to the numerical solution of one-dimensional Riemann integrals [Gh. Adam, S. Adam, Springer LNCS 7125, 1-16 (2012)] are reported. First, it is shown that the numerical quadrature which avoids the overcomputing and minimizes the hidden floating point loss of precision asks for the consideration of three classes of integration domain lengths endowed with specific quadrature sums: microscopic (trapezoidal rule), mesoscopic (Simpson rule), and macroscopic (quadrature sums of high algebraic degrees of precision). Second, sensitive diagnostic tools for the Bayesian inference on macroscopic ranges, coming from the use of Clenshaw-Curtis quadrature, are derived.
NASA Technical Reports Server (NTRS)
2003-01-01
At the summit of Mauna Kea, Hawaii, NASA astronomers have linked the two 10-meter (33-foot) telescopes at the W. M. Keck Observatory. The linked telescopes, which together are called the Keck Interferometer, make up the world's most powerful optical telescope system. The Keck Interferometer will search for planets around nearby stars and study dust clouds around those stars that may hamper future space-based searches for habitable, Earthlike planets. The Keck Interferometer is part of NASA's Origins program, which seeks to answer two fundamental questions: How did we get here? Are we alone?
Heterodyne Interferometer Angle Metrology
NASA Technical Reports Server (NTRS)
Hahn, Inseob; Weilert, Mark A.; Wang, Xu; Goullioud, Renaud
2010-01-01
A compact, high-resolution angle measurement instrument has been developed that is based on a heterodyne interferometer. The common-path heterodyne interferometer metrology is used to measure displacements of a reflective target surface. In the interferometer setup, an optical mask is used to sample the measurement laser beam reflecting back from a target surface. Angular rotations, around two orthogonal axes in a plane perpendicular to the measurement- beam propagation direction, are determined simultaneously from the relative displacement measurement of the target surface. The device is used in a tracking telescope system where pitch and yaw measurements of a flat mirror were simultaneously performed with a sensitivity of 0.1 nrad, per second, and a measuring range of 0.15 mrad at a working distance of an order of a meter. The nonlinearity of the device is also measured less than one percent over the measurement range.
Hitachi, K. Ishizawa, A.; Mashiko, H.; Sogawa, T.; Gotoh, H.; Tadanaga, O.; Nishikawa, T.
2015-06-08
We report the stabilization of the carrier-envelope offset (CEO) frequency of an Er-doped fiber laser with a collinear 2f-to-3f self-referencing interferometer. The interferometer is implemented by a dual-pitch periodically poled lithium niobate ridge waveguide with two different quasi-phase matching pitch sizes. We obtain a 52-dB signal-to-noise ratio in the 100-kHz resolution bandwidth of a heterodyne beat signal, which is sufficient for frequency stabilization. We also demonstrate that the collinear geometry is robust against environmental perturbation by comparing in-loop and out-of-loop Allan deviations when the in-loop CEO frequency is stabilized with a phase-locked loop circuit.
NASA Astrophysics Data System (ADS)
Hitachi, K.; Ishizawa, A.; Tadanaga, O.; Nishikawa, T.; Mashiko, H.; Sogawa, T.; Gotoh, H.
2015-06-01
We report the stabilization of the carrier-envelope offset (CEO) frequency of an Er-doped fiber laser with a collinear 2f-to-3f self-referencing interferometer. The interferometer is implemented by a dual-pitch periodically poled lithium niobate ridge waveguide with two different quasi-phase matching pitch sizes. We obtain a 52-dB signal-to-noise ratio in the 100-kHz resolution bandwidth of a heterodyne beat signal, which is sufficient for frequency stabilization. We also demonstrate that the collinear geometry is robust against environmental perturbation by comparing in-loop and out-of-loop Allan deviations when the in-loop CEO frequency is stabilized with a phase-locked loop circuit.
Quadrature formulas for Fourier coefficients
NASA Astrophysics Data System (ADS)
Bojanov, Borislav; Petrova, Guergana
2009-09-01
We consider quadrature formulas of high degree of precision for the computation of the Fourier coefficients in expansions of functions with respect to a system of orthogonal polynomials. In particular, we show the uniqueness of a multiple node formula for the Fourier-Tchebycheff coefficients given by Micchelli and Sharma and construct new Gaussian formulas for the Fourier coefficients of a function, based on the values of the function and its derivatives.
NASA Astrophysics Data System (ADS)
Ray Pitambar Mohapatra, Satyanarayan
This is an exciting time for Gravitational Wave (GW) theory and observations. From a theoretical standpoint, the grand-challenge problem of the full evolution of a Binary Black Hole (BBH) system has been solved numerically, and a variety of source simulations are made available steadfastly. On the observational side, the first generation of state-of-the-art GW detectors, LIGO and Virgo, have achieved their design goal, collected data and provided astrophysically meaningful limits. The second generation of detectors are expected to start running by 2015. Inspired by this zeitgeist, this thesis focuses on the detection of potential GW signatures from the coalescence of BBH in ground-based laser interferometers. The LIGO Scientific Collaboration has implemented different algorithms to search for transient GW signatures, targeting different portions of the BBH coalescence waveform. This thesis has used the existing algorithms to study the detection potential of GW from colliding BBH in LIGO in a wide range of source parameters, such as mass and spin of the black holes, using a sample of data from the last two months of the S5 LIGO science run (14 Aug 2007 to 30 Sept 2007). This thesis also uses numerical relativity waveforms made available via the Numerical INJection Analysis project (NINJA). Methods such as the Chirplet based analysis and the use of multivariate classifiers to optimize burst search algorithms have been introduced in this thesis. These performance studies over a wide parameter space were designed to optimize the discovery potential of ground-based GW detectors and defining strategies for the search of BBH signatures in advanced LIGO data, as a step towards the realization of GW astronomy.
Recording of Phase Transition in Tin in Shock and Release Waves Using Laser Interferometer
NASA Astrophysics Data System (ADS)
Finyushin, Stanislav; Fedorov, Alexey; Mikhailov, Anatoliy; Nazarov, Dmitriy; Chudakov, Evgeniy; Kalashnikov, Denis; Trunin, Ivan; Tereshkina, Irina
2013-06-01
In this paper the authors present results of experiments, concerning the recording of free surface velocity and particle velocity of tin/LiF window interface of tin sample at the dynamic pressure in the range 10 to 70 GPa, using laser interferometry methods Fabry-Perot and PDV. The features of recorded velocity histories interpreted the polymorphous transitions (the direct β - γ phase transition, the reverse γ - β phase transition) and some points on tin melting curve. The tin samples were loaded by impactor accelerated by explosion products of HE cartridge (the wave with the rectangular profile) or by direct detonation wave of HE layer (the wave with the decaying profile). With the help of PDV method, velocity of particles cloud was recorded with the free surface velocity at the moment when the shock wave front arrived to the tin free surface with the pressure amplitude of P >= 19 GPa. This could is corresponded to appearing of the liquid-solid phase at the release wave. The tin melting in the shock wave is recorded at the loading pressure of P >= 51 GPa. The pressure and temperature numerical simulations were performed for shock compression and the further releasing of tin.
Corral, V; Guzmán, R; Gordón, C; Leijtens, X J M; Carpintero, G
2016-05-01
We report the demonstration of an optical-frequency comb generator based on a monolithically integrated ring laser fabricated in a multiproject wafer run in an active/passive integration process in a generic foundry using standardized building blocks. The device is based on a passive mode-locked ring laser architecture, which includes a Mach-Zehnder interferometer to flatten the spectral shape of the comb output. This structure allows monolithic integration with other optical components, such as optical filters for wavelength selection, or dual wavelength lasers for their stabilization. The results show a -10 dB span of the optical comb of 8.7 nm (1.08 THz), with comb spacing of 10.16 GHz. We also obtain a flatness of 44 lines within a 1.8 dB power variation. PMID:27128043
NASA Technical Reports Server (NTRS)
Seasholtz, R. G.; Goldman, L. J.
1982-01-01
A technique for measuring a small optical axis velocity component in a flow with a large transverse velocity component is presented. Experimental results are given for a subsonic free jet operating in a laboratory environment, and for a 0.508 meter diameter turbine stator cascade. Satisfactory operation of the instrument was demonstrated in the stator cascade facility with an ambient acoustic noise level during operation of about 105 dB. In addition, the turbulence intensity measured with the interferometer was consistent with previous measurements taken with a fringe type laser anemometer.
Compact portable diffraction moire interferometer
Deason, V.A.; Ward, M.B.
1988-05-23
A compact and portable moire interferometer used to determine surface deformations of an object. The improved interferometer is comprised of a laser beam, optical and fiber optics devices coupling the beam to one or more evanescent wave splitters, and collimating lenses directing the split beam at one or more specimen gratings. Observations means including film and video cameras may be used to view and record the resultant fringe patterns. 7 figs.
Compact portable diffraction moire interferometer
Deason, Vance A.; Ward, Michael B.
1989-01-01
A compact and portable moire interferometer used to determine surface deformations of an object. The improved interferometer is comprised of a laser beam, optical and fiber optics devices coupling the beam to one or more evanescent wave splitters, and collimating lenses directing the split beam at one or more specimen gratings. Observation means including film and video cameras may be used to view and record the resultant fringe patterns.
Semiconductor laser self-mixing micro-vibration measuring technology based on Hilbert transform
NASA Astrophysics Data System (ADS)
Tao, Yufeng; Wang, Ming; Xia, Wei
2016-06-01
A signal-processing synthesizing Wavelet transform and Hilbert transform is employed to measurement of uniform or non-uniform vibrations in self-mixing interferometer on semiconductor laser diode with quantum well. Background noise and fringe inclination are solved by decomposing effect, fringe counting is adopted to automatic determine decomposing level, a couple of exact quadrature signals are produced by Hilbert transform to extract vibration. The tempting potential of real-time measuring micro vibration with high accuracy and wide dynamic response bandwidth using proposed method is proven by both simulation and experiment. Advantages and error sources are presented as well. Main features of proposed semiconductor laser self-mixing interferometer are constant current supply, high resolution, simplest optical path and much higher tolerance to feedback level than existing self-mixing interferometers, which is competitive for non-contact vibration measurement.
A Thermal-beam Calcium Interferometer
NASA Astrophysics Data System (ADS)
Erickson, Christopher; van Zjill, Marshall; Washburn, Matthew; Archibald, James; Christensen, Dan; Birrell, Jeremiah; Burdett, Adam; Durfee, Dallin
2007-06-01
We report on the construction of a next-generation atom interferometer. Our research includes developing passive stabilization techniques, low-noise laser current drivers, high-speed scan-balancing lock circuits, and high-speed low-noise photo-detecting units. Our efforts have lead to developing an extremely stable laser locked to an ultra-high finesse optical cavity for use in a Ramsey-Bord'e interferometer scheme. The interferometer itself is based on a thermal calcium beam and will be upgraded in the future to a dual species Ca/Sr interferometer sensitive enough to improve measurements of possible time variance of the fine structure constant.
NASA Astrophysics Data System (ADS)
Khalili, Farid Ya.; Tarabrin, Sergey P.; Hammerer, Klemens; Schnabel, Roman
2016-07-01
We analyze the radiation-pressure-induced interaction of mirror motion and light fields in Michelson-type interferometers used for the detection of gravitational waves and for fundamental research in tabletop quantum optomechanical experiments, focusing on the asymmetric regime with a (slightly) unbalanced beam splitter and a (small) offset from the dark port. This regime, as it was shown recently, provides new interesting features, in particular a stable optical spring and optical cooling on cavity resonance. We show that, generally, the nature of optomechanical coupling in Michelson-type interferometers does not fit into the standard dispersive-dissipative dichotomy. In particular, a symmetric Michelson interferometer with signal-recycling but without power-recycling cavity is characterized by a purely dissipative optomechanical coupling; only in the presence of asymmetry, additional dispersive coupling arises. In gravitational waves detectors possessing signal- and power-recycling cavities, yet another coherent type of optomechanical coupling takes place. We develop here a generalized framework for the analysis of asymmetric Michelson-type interferometers, which also covers the possibility of the injection of carrier light into both ports of the interferometer. Using this framework, we analyze in depth the anomalous features of the Michelson-Sagnac interferometer, which have been discussed and observed experimentally previously [A. Xuereb et al., Phys. Rev. Lett. 107, 213604 (2011), 10.1103/PhysRevLett.107.213604; S. P. Tarabrin et al., Phys. Rev. A 88, 023809 (2013);, 10.1103/PhysRevA.88.023809 A. Sawadsky et al., Phys. Rev. Lett. 114, 043601 (2015), 10.1103/PhysRevLett.114.043601].
NASA Astrophysics Data System (ADS)
Yin, Guolu; Lou, Shuqin; Zou, Hui
2013-04-01
A multiwavelength Er-doped fiber (EDF) laser was proposed and experimentally demonstrated using a nonlinear optical loop mirror (NOLM) and a twin-core fiber-based Mach-Zehnder interferometer (TCF MZI). A total of 40 lasing wavelengths around the wavelength of 1560 nm with an average signal-to-noise ratio (SNR) of 28 dB and 30 lasing wavelengths around the wavelength of 1530 nm with an average SNR of 20 dB were achieved when the EDF length was 6 and 2 m, respectively. To the best of our knowledge, this is the first time that multiwavelength oscillations have been obtained in the wavelength range of 1530 nm by using a NOLM. The wavelength spacing is ˜0.19 nm which is determined by the TCF MZI. In addition, the effects of the pump power and the state of polarization on the performance of the multiwavelength EDF laser were investigated.
NASA Astrophysics Data System (ADS)
Feng, Suchun; Xu, Ou; Lu, Shaohua; Mao, Xiangqiao; Ning, Tigang; Jian, Shuisheng
2009-04-01
A switchable erbium-doped fiber-ring laser providing dual-wavelength outputs with orthogonal polarizations when operating at room temperature is proposed. One polarization-maintaining fiber Bragg grating (PMFBG) in a Sagnac loop interferometer is used as the wavelength-selective filter. Due to the enhancement of the polarization hole burning (PHB) by the PMFBG, the laser can be designed to operate in stable dual-wavelength or wavelength-switching modes with a wavelength spacing of 0.336 nm at room temperature by adjusting a polarization controller (PC). The optical signal-to-noise ratio (OSNR) is over 42 dB. The amplitude variation over 90 min is less than 0.6 dB for both wavelengths.
Error Analysis of Quadrature Rules. Classroom Notes
ERIC Educational Resources Information Center
Glaister, P.
2004-01-01
Approaches to the determination of the error in numerical quadrature rules are discussed and compared. This article considers the problem of the determination of errors in numerical quadrature rules, taking Simpson's rule as the principal example. It suggests an approach based on truncation error analysis of numerical schemes for differential…
Liquid-Crystal Point-Diffraction Interferometer
NASA Technical Reports Server (NTRS)
Mercer, Carolyn R.
1996-01-01
Liquid-crystal point-diffraction interferometer (LCPDI) invented to combine flexible control of liquid-crystal phase-shifts with robustness of point-diffraction interferometers. Produces interferograms indicative of shapes of wavefronts of laser beams having passed through or reflected from objects of interest. Interferograms combined in computers to produce phase maps describing wavefronts.
A generalized discrepancy and quadrature error bound
NASA Astrophysics Data System (ADS)
Hickernell, F. J.
1998-01-01
An error bound for multidimensional quadrature is derived that includes the Koksma-Hlawka inequality as a special case. This error bound takes the form of a product of two terms. One term, which depends only on the integrand, is defined as a generalized variation. The other term, which depends only on the quadrature rule, is defined as a generalized discrepancy. The generalized discrepancy is a figure of merit for quadrature rules and includes as special cases the L-p-star discrepancy and P-alpha that arises in the study of lattice rules.
Gaussian quadrature formulae for arbitrary positive measures.
Fernandes, Andrew D; Atchley, William R
2006-01-01
We present computational methods and subroutines to compute Gaussian quadrature integration formulas for arbitrary positive measures. For expensive integrands that can be factored into well-known forms, Gaussian quadrature schemes allow for efficient evaluation of high-accuracy and -precision numerical integrals, especially compared to general ad hoc schemes. In addition, for certain well-known density measures (the normal, gamma, log-normal, Student's t, inverse-gamma, beta, and Fisher's F) we present exact formulae for computing the respective quadrature scheme. PMID:19455218
Lockerbie, N. A.; Tokmakov, K. V.
2014-11-15
This paper describes the design and performance of an extremely low-noise differential transimpedance amplifier, which takes its two inputs from separate photodiodes. The amplifier was planned to serve as the front-end electronics for a highly sensitive shadow-displacement sensing system, aimed at detecting very low-level “Violin-Mode” (VM) oscillations in 0.4 mm diameter by 600 mm long fused-silica suspension fibres. Four such highly tensioned fibres support the 40 kg test-masses/mirrors of the Advanced Laser Interferometer Gravitational wave Observatory interferometers. This novel design of amplifier incorporates features which prevent “noise-gain peaking” arising from large area photodiode (and cable) capacitances, and which also usefully separate the DC and AC photocurrents coming from the photodiodes. In consequence, the differential amplifier was able to generate straightforwardly two DC outputs, one per photodiode, as well as a single high-gain output for monitoring the VM oscillations—this output being derived from the difference of the photodiodes’ two, naturally anti-phase, AC photocurrents. Following a displacement calibration, the amplifier's final VM signal output was found to have an AC displacement responsivity at 500 Hz of (9.43 ± 1.20) MV(rms) m{sup −1}(rms), and, therefore, a shot-noise limited sensitivity to such AC shadow- (i.e., fibre-) displacements of (69 ± 13) picometres/√Hz at this frequency, over a measuring span of ±0.1 mm.
NASA Astrophysics Data System (ADS)
Iwamoto, D.; Sakuma, I.; Kitagawa, Y.; Kikuchi, Y.; Fukumoto, N.; Nagata, M.
2012-10-01
In next step of fusion devices such as ITER, lifetime of plasma-facing materials (PFMs) is strongly affected by transient heat and particle loads during type I edge localized modes (ELMs) and disruption. To clarify damage characteristics of the PFMs, transient heat and particle loads have been simulated by using a plasma gun device. We have performed simulation experiments by using a magnetized coaxial plasma gun (MCPG) device at University of Hyogo. The line-averaged electron density measured by a He-Ne interferometer is 2x10^21 m-3 in a drift tube. The plasma velocity measured by a time of flight technique and ion Doppler spectrometer was 70 km/s, corresponding to the ion energy of 100 eV for helium. Thus, the ion flux density is 1.4x10^26 m-2s-1. On the other hand, the MCPG is connected to a target chamber for material irradiation experiments. It is important to measure plasma parameters in front of target materials in the target chamber. In particular, a vapor cloud layer in front of the target material produced by the pulsed plasma irradiation has to be characterized in order to understand surface damage of PFMs under ELM-like plasma bombardment. In the conference, preliminary results of application of the He-Ne laser interferometer for the above experiment will be shown.
Lockerbie, N A; Tokmakov, K V
2014-11-01
This paper describes the design and performance of an extremely low-noise differential transimpedance amplifier, which takes its two inputs from separate photodiodes. The amplifier was planned to serve as the front-end electronics for a highly sensitive shadow-displacement sensing system, aimed at detecting very low-level "Violin-Mode" (VM) oscillations in 0.4 mm diameter by 600 mm long fused-silica suspension fibres. Four such highly tensioned fibres support the 40 kg test-masses/mirrors of the Advanced Laser Interferometer Gravitational wave Observatory interferometers. This novel design of amplifier incorporates features which prevent "noise-gain peaking" arising from large area photodiode (and cable) capacitances, and which also usefully separate the DC and AC photocurrents coming from the photodiodes. In consequence, the differential amplifier was able to generate straightforwardly two DC outputs, one per photodiode, as well as a single high-gain output for monitoring the VM oscillations-this output being derived from the difference of the photodiodes' two, naturally anti-phase, AC photocurrents. Following a displacement calibration, the amplifier's final VM signal output was found to have an AC displacement responsivity at 500 Hz of (9.43 ± 1.20) MV(rms) m(-1)(rms), and, therefore, a shot-noise limited sensitivity to such AC shadow- (i.e., fibre-) displacements of (69 ± 13) picometres/√Hz at this frequency, over a measuring span of ±0.1 mm. PMID:25430131
NASA Astrophysics Data System (ADS)
Lockerbie, N. A.; Tokmakov, K. V.
2014-11-01
This paper describes the design and performance of an extremely low-noise differential transimpedance amplifier, which takes its two inputs from separate photodiodes. The amplifier was planned to serve as the front-end electronics for a highly sensitive shadow-displacement sensing system, aimed at detecting very low-level "Violin-Mode" (VM) oscillations in 0.4 mm diameter by 600 mm long fused-silica suspension fibres. Four such highly tensioned fibres support the 40 kg test-masses/mirrors of the Advanced Laser Interferometer Gravitational wave Observatory interferometers. This novel design of amplifier incorporates features which prevent "noise-gain peaking" arising from large area photodiode (and cable) capacitances, and which also usefully separate the DC and AC photocurrents coming from the photodiodes. In consequence, the differential amplifier was able to generate straightforwardly two DC outputs, one per photodiode, as well as a single high-gain output for monitoring the VM oscillations—this output being derived from the difference of the photodiodes' two, naturally anti-phase, AC photocurrents. Following a displacement calibration, the amplifier's final VM signal output was found to have an AC displacement responsivity at 500 Hz of (9.43 ± 1.20) MV(rms) m-1(rms), and, therefore, a shot-noise limited sensitivity to such AC shadow- (i.e., fibre-) displacements of (69 ± 13) picometres/√Hz at this frequency, over a measuring span of ±0.1 mm.
NASA Astrophysics Data System (ADS)
Zarroli, John M.; Karchnak, Martin; Goodenow, Theodore C.
2004-07-01
Correct sensing provides the availability of information for health monitoring interests, including those associated with unit health, stress, diagnostics and prognostics. Data is presented from measurement projects employing a 'highly linear,' wideband (0 to hundreds of kilohertz), large dynamic range (over 180 dB demonstrated in both displacement and acceleration), non-contact, vibration and Acoustic Emissions (AE) measurement capability. The measurement capability is currently embodied in an Epoch Engineering, Inc. Research and Development (R&D) Robust Laser Interferometer (RLI). Existing data from previous measurement projects is assessed in terms of its potential to provide meaningful information for health monitoring needs in situations involving civil, mechanical and aerospace infrastructure elements. In addition to performance considerations, implementation alternatives, integration benefits, cost considerations and remaining work that would need to be accomplished for commercializing the RLI capability is presented.
NASA Astrophysics Data System (ADS)
Debowska, Anna K.; Koba, Marcin; Janik, Monika; Bock, Wojtek J.; Śmietana, Mateusz
2016-05-01
In this paper we focus on refractive index (RI) sensing properties of a micro-size In-fiber Mach-Zehnder Interferometer (μIMZI). The μIMZI structure was fabricated as a precisely controlled side opening of a single-mode fiber using a femtosecond laser. The sensitivity to RI change in the micro-cavity has been measured and two RI sensitivity regions have been found for RI 1.33-1.36 and 1.37-1.40 RIU. The sensitivity in the first region is over 12,000 nm/RIU, and in the higher RI region is close to 50% higher. The obtained structures are an excellent solution for RI sensing with negligible temperature cross-sensitivity, especially where small amounts of liquid are available, e.g. in lab-on-chip, microfluidics.
Antenna-array, phase quadrature tracking system
NASA Technical Reports Server (NTRS)
Cubley, H. D.
1970-01-01
Phase relationship between input signals appearing on widely-spaced parallel connected antenna elements in array is automatically adjusted in phase quadrature tracking system. Compact and lightweight design permit use in wide variety of airborne communications networks.
Structured eigenvalue problems for rational gauss quadrature
NASA Astrophysics Data System (ADS)
Fasino, Dario; Gemignani, Luca
2007-08-01
The connection between Gauss quadrature rules and the algebraic eigenvalue problem for a Jacobi matrix was first exploited in the now classical paper by Golub and Welsch (Math. Comput. 23(106), 221?230, 1969). From then on many computational problems arising in the construction of (polynomial) Gauss quadrature formulas have been reduced to solving direct and inverse eigenvalue problems for symmetric tridiagonals. Over the last few years (rational) generalizations of the classical Gauss quadrature formulas have been studied, i.e., formulas integrating exactly in spaces of rational functions. This paper wants to illustrate that stable and efficient procedures based on structured numerical linear algebra techniques can also be devised for the solution of the eigenvalue problems arising in the field of rational Gauss quadrature.
Calculates Angular Quadrature Weights and Cosines.
Energy Science and Technology Software Center (ESTSC)
1988-02-18
DSNQUAD calculates the angular quadrature weights and cosines for use in CCC-254/ANISN-ORNL. The subroutines in DSNQUAD were lifted from the XSDRN-PM code, which is supplied with the CCC-475/ SCALIAS-77 package.
Wavelength independent interferometer
NASA Technical Reports Server (NTRS)
Hochberg, Eric B. (Inventor); Page, Norman A. (Inventor)
1991-01-01
A polychromatic interferometer utilizing a plurality of parabolic reflective surfaces to properly preserve the fidelity of light wavefronts irrespective of their wavelengths as they pass through the instrument is disclosed. A preferred embodiment of the invention utilizes an optical train which comprises three off-axis parabolas arranged in conjunction with a beam-splitter and a reference mirror to form a Twyman-Green interferometer. An illumination subsystem is provided and comprises a pair of lasers at different preselected wavelengths in the visible spectrum. The output light of the two lasers is coaxially combined by means of a plurality of reflectors and a grating beam combiner to form a single light source at the focal point of the first parabolic reflection surface which acts as a beam collimator for the rest of the optical train. By using visible light having two distinct wavelengths, the present invention provides a long equivalent wavelength interferogram which operates at visible light wherein the effective wavelength is equal to the product of the wavelengths of the two laser sources divided by their difference in wavelength. As a result, the invention provides the advantages of what amounts to long wavelength interferometry but without incurring the disadvantage of the negligible reflection coefficient of the human eye to long wavelength frequencies which would otherwise defeat any attempt to form an interferogram at that low frequency using only one light source.
Improved Skin Friction Interferometer
NASA Technical Reports Server (NTRS)
Westphal, R. V.; Bachalo, W. D.; Houser, M. H.
1986-01-01
An improved system for measuring aerodynamic skin friction which uses a dual-laser-beam oil-film interferometer was developed. Improvements in the optical hardware provided equal signal characteristics for each beam and reduced the cost and complexity of the system by replacing polarization rotation by a mirrored prism for separation of the two signals. An automated, objective, data-reduction procedure was implemented to eliminate tedious manual manipulation of the interferometry data records. The present system was intended for use in two-dimensional, incompressible flows over a smooth, level surface without pressure gradient, but the improvements discussed are not limited to this application.
Infrared lateral shearing interferometers
NASA Astrophysics Data System (ADS)
Kwon, O.
1980-04-01
Recently IR interferometry has received much attention for its special capabilities of testing IR materials, diamond-turned metal mirrors, deep aspherics, unpolished rough surface optics, and other unconventional optics. A CW CO2 laser is used as a coherent light source at 10.6 microns, and germanium and zinc selenide optics are used for lenses and beam splitters. A pyroelectric vidicon (PEV) detects the modulated interference pattern through a TV monitor and video recorder-player. This paper presents three methods of IR lateral shear interferometry using (1) a germanium plane-parallel plate, (2) a Ronchi ruling, and (3) a double-grating lateral shear interferometer.
NASA Technical Reports Server (NTRS)
Rogers, Ryan
2007-01-01
The Michelson Interferometer is a device used in many applications, but here it was used to measure small differences in distance, in the milli-inch range, specifically for defects in the Orbiter windows. In this paper, the method of using the Michelson Interferometer for measuring small distances is explained as well as the mathematics of the system. The coherence length of several light sources was calculated in order to see just how small a defect could be measured. Since white light is a very broadband source, its coherence length is very short and thus can be used to measure small defects in glass. After finding the front and back reflections from a very thin glass slide with ease and calculating the thickness of it very accurately, it was concluded that this system could find and measure small defects on the Orbiter windows. This report also discusses a failed attempt for another use of this technology as well as describes an area of promise for further analysis. The latter of these areas has applications for finding possible defects in Orbiter windows without moving parts.
NASA Astrophysics Data System (ADS)
Cliche, Jean-François; Allard, Martin; Têtu, Michel
2006-05-01
In many sensing systems, a highly coherent laser source is necessary to perform sensitive interferometric or coherent measurements. At TeraXion, we have built a compact laser system that provides a stable laser frequency with a very narrow linewidth using a 60 mW DFB semiconductor laser. The linewidth reduction system uses a frequency discriminator to measure the laser frequency noise and provides an electrical feedback to reduce this noise over a given bandwidth. Experimental work shows that the phase noise of the DFB semiconductor laser can be reduced by more than 4 orders of magnitude from 10 Hz to 100 kHz. We analyzed the effect of the particular frequency noise spectrum of such a laser on its degree of coherence, its linewidth and the resulting interferometric noise. The laser linewidth computed from the power spectral density of frequency noise of the laser is reduced from 570 kHz down to an equivalent of 1.8 kHz when the output signal is observed for 30 ms, and from 370 kHz to 18 Hz for 1 ms. Similarly, the coherence length is increased from 145 m up to 45 km for fringes observed over 30 ms. Each result is compared with those obtained with a fiber laser.
NASA Astrophysics Data System (ADS)
Koerdt, Michael; Vollertsen, Frank
2011-04-01
It is known that deep ultraviolet (UV) radiation induces a refractive index increase in the surface layer of polymethylmethacrylate (PMMA) samples. This effect can be used for the fabrication of integrated optical waveguides. PMMA is of considerable interest for bio and chemical sensing applications because it is biocompatible and can be micromachined by several methods, e.g. structuring by photolithography, ablation and hot embossing. In the presented work direct UV irradiation of a common PMMA substrate by a krypton fluoride excimer laser beam through a contact mask has been used to write integrated optical Mach-Zehnder interferometers (MZI). MZI are used as sensitive bio and chemical sensors. The aim was to determine contact mask design and laser irradiation parameters for fabricating single-mode MZI for the infrared region from 1.30 μm to 1.62 μm. Straight and curved waveguides have been generated and characterized to determine the optical losses. The generation of channel waveguide structures has been optimized by a two step irradiation process to minimize the lithographic writing time and optical loss. By flood exposure to UV laser radiation in the first step the optical absorption of PMMA can be increased in the irradiated region. The required refractive index profile is then achieved with a second lithographic irradiation. The spectral behaviour of an unbalanced, integrated optical MZI fabricated by this excimer laser based contact mask method is shown for the first time. Further the optical intensity at the output port of a MZI has been measured while the optical path length difference was tuned by creating a temperature difference between the two arms of the MZI.
Goodenow, T.C.; Shipman, R.L.; Holland, H.M.
1995-06-01
Epoch Engineering, Incorporated (EEI) has completed a series of vibration measurements comparing their newly-developed Robust Laser Interferometer (RLI) with accelerometer-based instrumentation systems. EEI has successfully demonstrated, on several pieces of commonplace machinery, that non-contact, line-of-sight measurements are practical and yield results equal to or, in some cases, better than customary field implementations of accelerometers. The demonstration included analysis and comparison of such phenomena as nonlinearity, transverse sensitivity, harmonics, and signal-to-noise ratio. Fast Fourier Transformations were performed on the accelerometer and the laser system outputs to provide a comparison basis. The RLI was demonstrated, within the limits of the task, to be a viable, line-of-sight, non-contact alternative to accelerometer systems. Several different kinds of machinery were instrumented and compared, including a small pump, a gear-driven cement mixer, a rotor kit, and two small fans. Known machinery vibration sources were verified and RLI system output file formats were verified to be compatible with commercial computer programs used for vibration monitoring and trend analysis. The RLI was also observed to be less subject to electromagnetic interference (EMI) and more capable at very low frequencies.
Goodenow, T.C.; Shipman, R.L.; Holland, H.M.
1995-06-01
Epoch Engineering, Incorporated (EEI) has completed a series of vibration measurements comparing their newly-developed Robust Laser Interferometer (RLI) with accelerometer-based instrumentation systems. EEI has successfully demonstrated, on several pieces of commonplace machinery, that non-contact, line-of-sight measurements are practical and yield results equal to or, in some cases, better than customary field implementations of accelerometers. The demonstration included analysis and comparison of such phenomena as nonlinearity, transverse sensitivity, harmonics, and signal-to-noise ratio. Fast Fourier Transformations were performed on the accelerometer and the laser system outputs to provide a comparison basis. The RLI was demonstrated, within the limits o the task, to be a viable, line-of-sight, non-contact alternative to accelerometer systems. Several different kinds of machinery were instrumented and. compared, including a small pump, a gear-driven cement mixer, a rotor kit, and two small fans. Known machinery vibration sources were verified and RLI system output file formats were verified to be compatible with commercial computer programs used for vibration monitoring and trend analysis. The RLI was also observed to be less subject to electromagnetic interference (EMI) and more capable at very low frequencies. This document, Volume 2, provides the appendices to this report.
NASA Astrophysics Data System (ADS)
Yin, Guolu; Lou, Shuqin; Wang, Xin; Han, Bolin
2013-12-01
A tunable multi-wavelength erbium-doped fiber laser (MEDFL) based on a nonlinear optical loop mirror (NOLM) was proposed and experimentally demonstrated by cascading a standard Mach-Zehnder interferometer (MZI) and a twin-core fiber (TCF)-based filter. Due to the ‘blue shift’ of the transmission band of the TCF-based filter when the TCF was bent, a tunable lasing waveband was realized by moving the transmission band of the TCF-based filter to cover different channels provided by the standard MZI. Experimental results showed that the lasing waveband can be linearly tuned over a range of 24 nm from 1542 to 1566 nm with a channel spacing of 0.4 nm, a maximum lasing line amount of 19, and an optical signal to noise ratio (OSNR) of 39 dB. The stability of the laser spectra was verified with a wavelength drift of 0.04 nm and a power fluctuation of ±0.3 dB.
Past and Future SOHO-Ulysses Quadratures
NASA Technical Reports Server (NTRS)
Suess, Steven; Poletto, G.
2006-01-01
With the launch of SOHO, it again became possible to carry out quadrature observations. In comparison with earlier observations, the new capabilities of coronal spectroscopy with UVCS and in situ ionization state and composition with Ulysses/SWICS enabled new types of studies. Results from two studies serve as examples: (i) The acceleration profile of wind from small coronal holes. (ii) A high-coronal reconnecting current sheet as the source of high ionization state Fe in a CME at Ulysses. Generally quadrature observations last only for a few days, when Ulysses is within ca. 5 degrees of the limb. This means luck is required for the phenomenon of interest to lie along the radial direction to Ulysses. However, when Ulysses is at high southern latitude in winter 2007 and high northern latitude in winter 2008, there will be unusually favorable configurations for quadrature observations with SOHO and corresponding bracketing limb observations from STEREO A/B. Specifically, Ulysses will be within 5 degrees of the limb from December 2006 to May 2007 and within 10 degrees of the limb from December 2007 to May 2008. These long-lasting quadratures and bracketing STEREO A/B observations overcome the limitations inherent in the short observation intervals of typical quadratures. Furthermore, ionization and charge state measurements like those on Ulysses will also be made on STEREO and these will be essential for identification of CME ejecta - one of the prime objectives for STEREO.
Quadrature formulae for problems in mechanics
NASA Astrophysics Data System (ADS)
Milovanović, Gradimir V.; Igić, Tomislav; Tončev, Novica
2012-09-01
The fast progress in recent years in symbolic computation and variable-precision arithmetic provide a possibility for generating the recursion coefficients in the three-term recurrence relation for orthogonal polynomials with respect to several nonclassical weight functions, as well as the construction of the corresponding quadrature rules of Gaussian type. Such quadratures are very important in many applications in engineering (fracture mechanics, damage mechanics, etc.), as well as in other computational and applied sciences. The boundary element method (BEM), finite element method (FEM), methods for solving integral equations, etc. very often require the numerical evaluation of one dimensional or multiple integrals with singular or near singular integrands with a high precision. In this paper we give some improvements of quadrature rules of Gaussian type with logarithmic and/or algebraic singularities. A numerical examples is included.
NASA Technical Reports Server (NTRS)
Baker, John G.; Thorpe, J. I.
2012-01-01
We consider a class of proposed gravitational wave detectors based on multiple atomic interferometers separated by large baselines and referenced by common laser systems. We compute the sensitivity limits of these detectors due to intrinsic phase noise of the light sources, non-inertial motion of the light sources, and atomic shot noise and compare them to sensitivity limits for traditional light interferometers. We find that atom interferometers and light interferometers are limited in a nearly identical way by intrinsic phase noise and that both require similar mitigation strategies (e.g. multiple arm instruments) to reach interesting sensitivities. The sensitivity limit from motion of the light sources is slightly different and favors the atom interferometers in the low-frequency limit, although the limit in both cases is severe. Whether this potential advantage outweighs the additional complexity associated with including atom interferometers will require further study.
Summation Paths in Clenshaw-Curtis Quadrature
NASA Astrophysics Data System (ADS)
Adam, S.; Adam, Gh.
2016-02-01
Two topics concerning the use of Clenshaw-Curtis quadrature within the Bayesian automatic adaptive quadrature approach to the numerical solution of Riemann integrals are considered. First, it is found that the efficient floating point computation of the coefficients of the Chebyshev series expansion of the integrand is to be done within a mathematical structure consisting of the union of coefficient families ordered into complete binary trees. Second, the scrutiny of the decay rates of the involved even and odd rank Chebyshev expansion coefficients with the increase of their rank labels enables the definition of Bayesian decision paths for the advancement to the numerical output.
Saucedo-A, Tonatiuh; De la Torre-Ibarra, M H; Santoyo, F Mendoza; Moreno, Ivan
2010-09-13
The use of digital holographic interferometry for 3D measurements using simultaneously three illumination directions was demonstrated by Saucedo et al. (Optics Express 14(4) 2006). The technique records two consecutive images where each one contains three holograms in it, e.g., one before the deformation and one after the deformation. A short coherence length laser must be used to obtain the simultaneous 3D information from the same laser source. In this manuscript we present an extension of this technique now illuminating simultaneously with three different lasers at 458, 532 and 633 nm, and using only one high resolution monochrome CMOS sensor. This new configuration gives the opportunity to use long coherence length lasers allowing the measurement of large object areas. A series of digital holographic interferograms are recorded and the information corresponding to each laser is isolated in the Fourier spectral domain where the corresponding phase difference is calculated. Experimental results render the orthogonal displacement components u, v and w during a simple load deformation. PMID:20940878
Photoacoustic microscopy based on polydimethylsiloxane thin film Fabry-Perot optical interferometer
NASA Astrophysics Data System (ADS)
Park, Soongho; Eom, Jonghyun; Shin, Jun Geun; Rim, Sunghwan; Lee, Byeong Ha
2016-03-01
We present a photoacoustic microscopy (PAM) system based on a Fabry-Perot Interferometer (FPI) consisting of a transparent Polydimethylsiloxane (PDMS) thin film. Most of the PAM systems have limitations with the system alignment because the ultrasound transducers for detection are not transparent. Therefore, the excitation laser source should avoid the opaque transducer to illuminate the sample, which makes the system difficult to build-up. Especially, the system volume is highly limited to be compact. In our experiment, to solve these difficulties, a FPI based on the PDMS film has been implemented and applied to measure the acoustic wave signal. The system uses a FPI as an acoustic wave detector instead of a conventional ultrasound transducer. A tunable laser was used to choose the quadrature-point at which the signal has the highly sensitve and linear response to the acoustic wave. Also a 20Hz pulsed Nd:YAG laser was used to generate acoustic waves from a sample. When the acoustic waves arrive at the PDMS film, one of the surfaces of the film is modulated at the detecting point, which gives the tuned FPI interference signal. From the signal arriving time, the depth location of the sample is calculated. As a primary experiment using the PDMS thin film as an ultrasound transducer, a couple of narrow black friction tapes located in a water container were used as the samples. This proposed imaging method can be used in various applications for the detection and measurement of acoustic waves.
Composite Gauss-Legendre Quadrature with Error Control
ERIC Educational Resources Information Center
Prentice, J. S. C.
2011-01-01
We describe composite Gauss-Legendre quadrature for determining definite integrals, including a means of controlling the approximation error. We compare the form and performance of the algorithm with standard Newton-Cotes quadrature. (Contains 1 table.)
Heterodyne Interferometer with Angstrom-level Periodic Nonlinearity
Schmitz, Tony L.; Beckwith, John F.
2005-01-25
Displacement measuring interferometer systems and methods are disclosed. One or more acousto-optic modulators for receiving a laser light beam from a laser light source can be utilized to split the laser light beam into two or more laser light beams, while spatially separating frequencies thereof. One or more reflective mechanisms can be utilized to reflect one or more of the laser light beams back to the acoustooptic modulator. Interference of two or more of the laser light beams generally at the acousto-optic modulator can provide an interfered laser light beam thereof. A detector for receiving the interfered laser light beam can be utilized to provide interferometer measurement data.
Runge-Kutta based generalized convolution quadrature
NASA Astrophysics Data System (ADS)
Lopez-Fernandez, Maria; Sauter, Stefan
2016-06-01
We present the Runge-Kutta generalized convolution quadrature (gCQ) with variable time steps for the numerical solution of convolution equations for time and space-time problems. We present the main properties of the method and a convergence result.
Gauss-Laguerre interval quadrature rule
NASA Astrophysics Data System (ADS)
Milovanovic, Gradimir V.; Cvetkovic, Aleksandar S.
2005-10-01
In this paper we prove the existence and uniqueness of the Gaussian interval quadrature formula with respect to the generalized Laguerre weight function. An algorithm for numerical construction has also investigated and some suitable solutions are proposed. A few numerical examples are included.
A Strontium87 Ion Interferometer
NASA Astrophysics Data System (ADS)
Erickson, Christopher J.; Archibald, James L., II; Jackson, Jarom; Anderson, Dean; Hermansen, Michael; Cunningham, Mark; Durfee, Dallin S.
2011-05-01
We describe a matter-wave interferometer based on Sr87+. The ions are generated from a laser-cooled strontium beam that is photo-ionized using a two-photon transition to an auto- ionizing state in the continuum. The ionization occurs between two electrodes, allowing us to accelerate the ions to any desired energy from a few meV to 20 keV. Each ion's quantum wave is split and recombined using stimulated Raman transitions between the hyperfine ground states of Sr87+. The two required optical frequencies for this transition are created by frequency-shifting a master laser in opposite directions by half of the 5 GHz ground-state hyperfine splitting. We can then determine the interferometer phase from the fluorescence of one of the ground states. We will discuss the theory of operation, experimental methods, and potential applications of the device. NSF, NIST
NASA Astrophysics Data System (ADS)
Vannoni, M.; Freijo Martín, I.
2016-05-01
The European XFEL (X-ray Free Electron Laser) is a large facility under construction in Hamburg, Germany. It will provide a transversally fully coherent x-ray radiation with outstanding characteristics: high repetition rate (up to 2700 pulses with a 0.6 ms long pulse train at 10 Hz), short wavelength (down to 0.05 nm), short pulse (in the femtoseconds scale), and high average brilliance (1.6 ṡ 1025 (photons s-1 mm-2 mrad-2)/0.1% bandwidth). The beam has very high pulse energy; therefore, it has to be spread out on a relatively long mirror (about 1 m). Due to the very short wavelength, the mirrors need to have a high quality surface on their entire length, and this is considered very challenging even with the most advanced polishing methods. In order to measure the mirrors and to characterize their interaction with the mechanical mount, we equipped a metrology laboratory with a large aperture Fizeau interferometer. The system is a classical 100 mm diameter commercial Fizeau, with an additional expander providing a 300 mm diameter beam. Despite the commercial nature of the system, special care has been taken in the polishing of the reference flats and in the expander quality. We report the first commissioning of the instrument, its calibration, and performance characterization, together with some preliminary results with the measurement of a 950 mm silicon substrate. The intended application is to characterize the final XFEL mirrors with nanometer accuracy.
Single and double superimposing interferometer systems
Erskine, David J.
2000-01-01
Interferometers which can imprint a coherent delay on a broadband uncollimated beam are described. The delay value can be independent of incident ray angle, allowing interferometry using uncollimated beams from common extended sources such as lamps and fiber bundles, and facilitating Fourier Transform spectroscopy of wide angle sources. Pairs of such interferometers matched in delay and dispersion can measure velocity and communicate using ordinary lamps, wide diameter optical fibers and arbitrary non-imaging paths, and not requiring a laser.
Dual-beam skin friction interferometer
NASA Technical Reports Server (NTRS)
Monson, D. J. (Inventor)
1981-01-01
A portable dual-laser beam interferometer is described that nonintrusively measures skin friction by monitoring the thickness change of an oil film at two locations while said oil film is subjected to shear stress. An interferometer flat is utilized to develop the two beams. Light detectors sense the beam reflections from the oil film and the surface thereunder. The signals from the detectors are recorded so that the number of interference fringes produced over a given time span may be counted.
Error Bounds for Quadrature Methods Involving Lower Order Derivatives
ERIC Educational Resources Information Center
Engelbrecht, Johann; Fedotov, Igor; Fedotova, Tanya; Harding, Ansie
2003-01-01
Quadrature methods for approximating the definite integral of a function f(t) over an interval [a,b] are in common use. Examples of such methods are the Newton-Cotes formulas (midpoint, trapezoidal and Simpson methods etc.) and the Gauss-Legendre quadrature rules, to name two types of quadrature. Error bounds for these approximations involve…
Michelson interferometer vibrometer using self-correcting synthetic-heterodyne demodulation.
Connelly, Michael J; Galeti, José Henrique; Kitano, Cláudio
2015-06-20
Synthetic-heterodyne demodulation is a useful technique for dynamic displacement and velocity detection in interferometric sensors, as it can provide an output signal that is immune to interferometric drift. With the advent of cost-effective, high-speed real-time signal-processing systems and software, processing of the complex signals encountered in interferometry has become more feasible. In synthetic heterodyne, to obtain the actual dynamic displacement or vibration of the object under test requires knowledge of the interferometer visibility and also the argument of two Bessel functions. In this paper, a method is described for determining the former and setting the Bessel function argument to a set value, which ensures maximum sensitivity. Conventional synthetic-heterodyne demodulation requires the use of two in-phase local oscillators; however, the relative phase of these oscillators relative to the interferometric signal is unknown. It is shown that, by using two additional quadrature local oscillators, a demodulated signal can be obtained that is independent of this phase difference. The experimental interferometer is a Michelson configuration using a visible single-mode laser, whose current is sinusoidally modulated at a frequency of 20 kHz. The detected interferometer output is acquired using a 250 kHz analog-to-digital converter and processed in real time. The system is used to measure the displacement sensitivity frequency response and linearity of a piezoelectric mirror shifter over a range of 500 Hz to 10 kHz. The experimental results show good agreement with two data-obtained independent techniques: the signal coincidence and denominated n-commuted Pernick method. PMID:26193022
Long distance measurement with femtosecond pulses using a dispersive interferometer.
Cui, M; Zeitouny, M G; Bhattacharya, N; van den Berg, S A; Urbach, H P
2011-03-28
We experimentally demonstrate long distance measurements with a femtosecond frequency comb laser using dispersive interferometry. The distance is derived from the unwrapped spectral phase of the dispersed interferometer output and the repetition frequency of the laser. For an interferometer length of 50 m this approach has been compared to an independent phase counting laser interferometer. The obtained mutual agreement is better than 1.5 μm (3×10(-8)), with a statistical averaging of less than 200 nm. Our experiments demonstrate that dispersive interferometry with a frequency comb laser is a powerful method for accurate and non-incremental measurement of long distances. PMID:21451683
The POLIS interferometer for ponderomotive squeezed light generation
NASA Astrophysics Data System (ADS)
Calloni, Enrico; Conte, Andrea; De Laurentis, Martina; Naticchioni, Luca; Puppo, Paola; Ricci, Fulvio
2016-07-01
POLIS (POnderomotive LIght Squeezer) is a suspended interferometer, presently under construction, devoted to the generation of ponderomotive squeezed light and to the study of the interaction of non classical quantum states of light and macroscopic objects. The interferometer is a Michelson whose half-meter long arms are constituted by high-finesse cavities, suspended to a seismic isolation chain similar to the Virgo SuperAttenuator. The mass of the suspended cavity mirrors are chosen to be tens of grams: this value is sufficiently high to permit the use of the well-tested Virgo suspension techniques but also sufficiently small to generate the coupling among the two phase quadratures with a limited amount of light in the cavity, of the order of few tens of kW. In this short paper the main features of the interferometer are shown, together with the expected sensitivity and squeezing factor.
Special relativity and interferometers
NASA Technical Reports Server (NTRS)
Han, D.; Kim, Y. S.
1988-01-01
A new generation of gravitational wave detectors is expected to be based on interferometers. Yurke et al. (1986) introduced a class of interferometers characterized by SU(1,1) which can in principle achieve a phase sensitivity approaching 1/N, where N is thte total number of photons entering the interferometer. It is shown here that the SU(1,1) interferometer can serve as an analog computer for Wigner's little group of the Poincare\\'| group.
Uniform positive-weight quadratures for discrete ordinate transport calculations
Carew, J.F.; Zamonsky, G.
1999-02-01
Mechanical quadratures that allow systematic improvement and solution convergence are derived for application of the discrete ordinates method to the Boltzmann transport equation. the quadrature directions are arranged on n latitudinal levels, are uniformly distributed over the unit sphere, and have positive weights. Both a uniform and equal-weight quadrature set UE{sub n} and a uniform and Gauss-weight quadrature set UG{sub n} are derived. These quadratures have the advantage over the standard level-symmetric LQ{sub n} quadrature sets in that the weights are positive for all orders, and the solution may be systematically converged by increasing the order of the quadrature set. As the order of the quadrature is increased the points approach a uniform continuous distribution on the unit sphere and the quadrature is invariant with respect to spatial rotations. The numerical integrals converge for continuous functions as the order of the quadrature is increased. Numerical calculations were performed to evaluate the application of the UE{sub n} quadrature set. Comparisons of the exact moments and those calculated using the UE{sub n} quadrature set demonstrate that the moment integrals are performed accurately except for distributions that are very sharply peaked along the direction of the polar axis. A series of DORT transport calculations of the >1-Mev neutron flux for a typical reactor core/pressure vessel geometry were also carried out. These calculations employed the UE{sub n} (n = 6, 10, 12, 18, and 24) quadratures and indicate that the UE{sub n} solutions have converged to within {approximately}0.5%. The UE{sub 24} solutions were also found to be more accurate than the calculations performed with the S{sub 16} level-symmetric quadratures.
Letunov, A. A.; Logvinenko, V. P.; Zav'yalov, V. V.
2008-03-15
An upgraded diagnostics for measuring the electron density profile in the L-2M stellarator is proposed. The existing diagnostics employs an interferometer based on an HCN laser with a mechanical frequency shifter and unmagnetized InSb detectors cooled with liquid helium. It is proposed to replace the HCN laser with a water vapor laser operating simultaneously at two wavelengths (220 and 118 {mu}m). Being equipped with an anisotropic exit mirror, the water vapor laser allows the generation of orthogonally polarized, frequency-splitted modes at each of these wavelengths with a frequency difference of several tens of kilohertzs. Such a scheme makes it possible to get rid of the mechanical frequency shifter. Moreover, simultaneous measurements at two wavelengths allow one to reliably separate the phase increments introduced by the plasma electron component and by variations in the lengths of the interferometer arms. To take full advantage of this scheme, specially developed cryogenic receivers consisting of Ge and InSb photodetectors placed one after another will be used. To increase the response of the system near {lambda} = 220 {mu}m, the InSb detector is placed in a Almost-Equal-To 0.55-T magnetic field.
Optically controlled quadrature coupler on silicon substrate
NASA Astrophysics Data System (ADS)
Bhadauria, Avanish; Sharma, Sonia; Sonania, Shikha; Akhtar, Jamil
2016-03-01
In this paper, we have proposed and studied an optically controlled quadrature coupler fabricated on silicon substrate. The optically controlled quadrature coupler can be realized by terminating its coupled or through ports by optically induced load. Simulation and experimental results show that by varying optical intensity, we can control the phase and amplitude of output RF signal and can realize optically controlled reflection type attenuator, reflection type phase-shifter and ultrafast switches. The new kind of proposed device can be useful for ultra-fast signal processing and modulation schemes in high speed communication especially in QPSK modulation. The optical control has several advantages over conventional techniques such as MEMS and other semiconductor switching, which have several inherent disadvantages and limitations like low response time, low power handling capacity, device parasitic and non-linearity.
Baker, John G; Thorpe, J I
2012-05-25
We consider a class of proposed gravitational-wave detectors based on multiple atomic interferometers separated by large baselines and referenced by common laser systems. We compute the sensitivity limits of these detectors due to intrinsic phase noise of the light sources, noninertial motion of the light sources, and atomic shot noise and compare them to sensitivity limits for traditional light interferometers. We find that atom interferometers and light interferometers are limited in a nearly identical way by intrinsic phase noise and that both require similar mitigation strategies (e.g., multiple-arm instruments) to reach interesting sensitivities. The sensitivity limit from motion of the light sources is slightly different and, in principle, favors the atom interferometers in the low-frequency limit, although the limit in both cases is severe. PMID:23003235
NASA Technical Reports Server (NTRS)
Baker, John G.
2012-01-01
We consider a class of proposed gravitational wave detectors based on multiple atomic interferometers separated by large baselines and referenced by common laser systems. We compute the sensitivity limits of these detectors due to intrinsic phase noise of the light sources, non-inertial motion of the light sources, and atomic shot noise and compare them to sensitivity limits for traditional light interferometers. We find that atom interferometers and light interferometers are limited in a nearly identical way by intrinsic phase noise and that both require similar mitigation strategies (e.g. multiple arm instruments) to reach interesting sensitivities. The sensitivity limit from motion of the light sources is slightly different and favors the atom interferometers in the low-frequency limit, although the limit in both cases is severe.
Exponential fitting quadrature rule for functional equations
NASA Astrophysics Data System (ADS)
Cardone, A.; Paternoster, B.; Santomauro, G.
2012-09-01
A Gaussian quadrature rule for periodic integrand function is presented. The weights and nodes depend on the frequency of the problem and they are constructed by following the exponential fitting theory. The composite rule based on this formula is derived. The analysis of the error is carried out and it proves that the exponentially fitted Gaussian rule is more accurate than the classical Gauss-Legendre rule when oscillatory functions are treated. Some numerical tests are presented.
Sarailou, Edris; Delfyett, Peter
2016-07-01
A linearized intensity modulator for periodic and pulsed light is proposed and demonstrated. The free carrier plasma effect has been used to modulate the refractive index of the phase section of a three-section mode-locked laser. If injection locked, the modulation induces an arcsine phase response on the three-section mode-locked laser. By introducing this mode-locked laser into a Mach-Zehnder interferometer biased at quadrature, one can realize a true linear intensity modulation. This novel laser suppresses any unwanted amplitude modulation and increases the performance of the linearized intensity modulator. Experimental results have provided a record low static I_{π} of 0.39 mA and a spur-free dynamic range of 75 dB.Hz^{2/3}. PMID:27367083
Nulling at the Keck Interferometer
NASA Technical Reports Server (NTRS)
Colavita, M. Mark; Serabyn, Gene; Wizinowich, Peter L.; Akeson, Rachel L.
2006-01-01
The nulling mode of the Keck Interferometer is being commissioned at the Mauna Kea summit. The nuller combines the two Keck telescope apertures in a split-pupil mode to both cancel the on-axis starlight and to coherently detect the residual signal. The nuller, working at 10 um, is tightly integrated with the other interferometer subsystems including the fringe and angle trackers, the delay lines and laser metrology, and the real-time control system. Since first 10 um light in August 2004, the system integration is proceeding with increasing functionality and performance, leading to demonstration of a 100:1 on-sky null in 2005. That level of performance has now been extended to observations with longer coherent integration times. An overview of the overall system is presented, with emphasis on the observing sequence, phasing system, and differences with respect to the V2 system, along with a presentation of some recent engineering data.
A Slow Ion Strontium Interferometer
NASA Astrophysics Data System (ADS)
Erickson, Christopher; Durfee, Dallin
2009-10-01
I will discuss an interferometer centered around a laser-cooled source of ^87Sr^+ ions, which will be split and recombined using stimulated Raman transitions. This will take place inside a conducting cylinder allowing the interferometer to measure electric and magnetic fields with unprecedented precision. Practical applications for the device include the precision measurement of the evolution of fields near solids to reveal their electronic structure. It will also be used for fundamental tests of the basic laws of electromagnetism and the search for a non-zero photon rest mass. The device should detect possible photon rest mass more than 100 times smaller than previous laboratory experiments. Both the details of the device and the theory connecting deviations from Coulomb's inverse-square law to a theory of massive photons will be discussed.
Sommargren, G.E.
1999-08-03
An interferometer is disclosed which has the capability of measuring optical elements and systems with an accuracy of {lambda}/1000 where {lambda} is the wavelength of visible light. Whereas current interferometers employ a reference surface, which inherently limits the accuracy of the measurement to about {lambda}/50, this interferometer uses an essentially perfect spherical reference wavefront generated by the fundamental process of diffraction. Whereas current interferometers illuminate the optic to be tested with an aberrated wavefront which also limits the accuracy of the measurement, this interferometer uses an essentially perfect spherical measurement wavefront generated by the fundamental process of diffraction. This interferometer is adjustable to give unity fringe visibility, which maximizes the signal-to-noise, and has the means to introduce a controlled prescribed relative phase shift between the reference wavefront and the wavefront from the optics under test, which permits analysis of the interference fringe pattern using standard phase extraction algorithms. 11 figs.
Sommargren, Gary E.
1999-01-01
An interferometer which has the capability of measuring optical elements and systems with an accuracy of .lambda./1000 where .lambda. is the wavelength of visible light. Whereas current interferometers employ a reference surface, which inherently limits the accuracy of the measurement to about .lambda./50, this interferometer uses an essentially perfect spherical reference wavefront generated by the fundamental process of diffraction. Whereas current interferometers illuminate the optic to be tested with an aberrated wavefront which also limits the accuracy of the measurement, this interferometer uses an essentially perfect spherical measurement wavefront generated by the fundamental process of diffraction. This interferometer is adjustable to give unity fringe visibility, which maximizes the signal-to-noise, and has the means to introduce a controlled prescribed relative phase shift between the reference wavefront and the wavefront from the optics under test, which permits analysis of the interference fringe pattern using standard phase extraction algorithms.
Hori, Yasuaki; Hirai, Akiko; Minoshima, Kaoru
2011-03-10
A prism-pair interferometer comprising two homodyne interferometers with a common light source was developed for high-precision measurements of the refractive index of optical glasses with an uncertainty of the order of 10{sup -6}. The two interferometers measure changes in the optical path length in the glass sample and in air, respectively. Uncertainties in the absolute wavelength of the common light source are cancelled out by calculating a ratio between the results from the interferometers. Uncertainties in phase measurement are suppressed by a quadrature detection system. The combined standard uncertainty of the developed system is evaluated as 1.1x10{sup -6}.
Gauss Legendre Quadrature Formulae for Tetrahedra
NASA Astrophysics Data System (ADS)
Rathod, H. T.; Venkatesudu, B.; Nagaraja, K. V.
2005-09-01
In this paper we consider the Gauss Legendre quadrature method for numerical integration over the standard tetrahedron: {(x, y, z)|0 = x, y, z = 1, x + y + z = 1} in the Cartesian three-dimensional (x, y, z) space. The mathematical transformation from the (x, y, z) space to (?, ?, ?) space is described to map the standard tetrahedron in (x, y, z) space to a standard 2-cube: {(?, ?, ?)| - 1 = ?, ?,? = 1} in the (?, ?, ?) space. This overcomes the difficulties associated with the derivation of new weight co-efficients and sampling points. The effectiveness of the formulae is demonstrated by applying them to the integration of three nonpolynomial and three polynomial functions.
Acedo, Pablo; Pedreira, P.; Criado, A. R.; Lamela, Horacio; Sanchez, Miguel; Sanchez, Joaquin
2008-10-15
A high spatial resolution two-color (CO{sub 2}, {lambda}=10.6 {mu}m, He-Ne, {lambda}=633 nm) interferometer for density profile measurements in the TJ-II stellarator is under development and installation, based in the currently operational single channel two-color heterodyne interferometer. To achieve the objectives of 32 channels, with 4-5 mm lateral separation between plasma chords, careful design and calibration of the interferometric waveforms for both the measurement and vibration compensation wavelengths are undertaken. The first step has been to set up in our laboratories an expanded-beam heterodyne/homodyne interferometer to evaluate the quality of both interferometric wavefronts, a reported source of poor vibration compensation and thus low resolution in the density profile measurements. This novel interferometric setup has allowed us to calibrate the spatial resolution in the profile measurements resulting in {approx}2 mm lateral resolution in the reconstruction of the interferometric wavefront.
Coupled-Cavity Interferometer for the Optics Laboratory
ERIC Educational Resources Information Center
Peterson, R. W.
1975-01-01
Describes the construction of a flexible coupled-cavity interferometer for student use. A helium-neon laser and phonograph turntable are the main components. Lists activities which may be performed with the apparatus. (Author/CP)
Surface profiling interferometer
Takacs, Peter Z.; Qian, Shi-Nan
1989-01-01
The design of a long-trace surface profiler for the non-contact measurement of surface profile, slope error and curvature on cylindrical synchrotron radiation (SR) mirrors. The optical system is based upon the concept of a pencil-beam interferometer with an inherent large depth-of-field. The key feature of the optical system is the zero-path-difference beam splitter, which separates the laser beam into two colinear, variable-separation probe beams. A linear array detector is used to record the interference fringe in the image, and analysis of the fringe location as a function of scan position allows one to reconstruct the surface profile. The optical head is mounted on an air bearing slide with the capability to measure long aspheric optics, typical of those encountered in SR applications. A novel feature of the optical system is the use of a transverse "outrigger" beam which provides information on the relative alignment of the scan axis to the cylinder optic symmetry axis.
Modulator-free quadrature amplitude modulation signal synthesis
NASA Astrophysics Data System (ADS)
Liu, Zhixin; Kakande, Joseph; Kelly, Brian; O'Carroll, John; Phelan, Richard; Richardson, David J.; Slavík, Radan
2014-12-01
The ability to generate high-speed on-off-keyed telecommunication signals by directly modulating a semiconductor laser’s drive current was one of the most exciting prospective applications of the nascent field of laser technology throughout the 1960s. Three decades of progress led to the commercialization of 2.5 Gbit s-1-per-channel submarine fibre optic systems that drove the growth of the internet as a global phenomenon. However, the detrimental frequency chirp associated with direct modulation forced industry to use external electro-optic modulators to deliver the next generation of on-off-keyed 10 Gbit s-1 systems and is absolutely prohibitive for today’s (>)100 Gbit s-1 coherent systems, which use complex modulation formats (for example, quadrature amplitude modulation). Here we use optical injection locking of directly modulated semiconductor lasers to generate complex modulation format signals showing distinct advantages over current and other currently researched solutions.
Modulator-free quadrature amplitude modulation signal synthesis
Liu, Zhixin; Kakande, Joseph; Kelly, Brian; O’Carroll, John; Phelan, Richard; Richardson, David J.; Slavík, Radan
2014-01-01
The ability to generate high-speed on–off-keyed telecommunication signals by directly modulating a semiconductor laser’s drive current was one of the most exciting prospective applications of the nascent field of laser technology throughout the 1960s. Three decades of progress led to the commercialization of 2.5 Gbit s−1-per-channel submarine fibre optic systems that drove the growth of the internet as a global phenomenon. However, the detrimental frequency chirp associated with direct modulation forced industry to use external electro-optic modulators to deliver the next generation of on–off-keyed 10 Gbit s−1 systems and is absolutely prohibitive for today’s (>)100 Gbit s−1 coherent systems, which use complex modulation formats (for example, quadrature amplitude modulation). Here we use optical injection locking of directly modulated semiconductor lasers to generate complex modulation format signals showing distinct advantages over current and other currently researched solutions. PMID:25523757
Dispersion interferometer using modulation amplitudes on LHD (invited)
Akiyama, T. Yasuhara, R.; Kawahata, K.; Okajima, S.; Nakayama, K.
2014-11-15
Since a dispersion interferometer is insensitive to mechanical vibrations, a vibration compensation system is not necessary. The CO{sub 2} laser dispersion interferometer with phase modulations on the Large Helical Device utilizes the new phase extraction method which uses modulation amplitudes and can improve a disadvantage of the original dispersion interferometer: measurement errors caused by variations of detected intensities. The phase variation within ±2 × 10{sup 17} m{sup −3} is obtained without vibration compensation system. The measured line averaged electron density with the dispersion interferometer shows good agreement with that with the existing far infrared laser interferometer. Fringe jump errors in high density ranging up to 1.5 × 10{sup 20} m{sup −3} can be overcome by a sufficient sampling rate of about 100 kHz.
The AEI 10 m prototype interferometer
NASA Astrophysics Data System (ADS)
Goßler, S.; Bertolini, A.; Born, M.; Chen, Y.; Dahl, K.; Gering, D.; Gräf, C.; Heinzel, G.; Hild, S.; Kawazoe, F.; Kranz, O.; Kühn, G.; Lück, H.; Mossavi, K.; Schnabel, R.; Somiya, K.; Strain, K. A.; Taylor, J. R.; Wanner, A.; Westphal, T.; Willke, B.; Danzmann, K.
2010-04-01
A 10 m prototype interferometer facility is currently being set up at the AEI in Hannover, Germany. The prototype interferometer will be housed inside a 100 m3 ultra-high vacuum envelope. Seismically isolated optical tables inside the vacuum system will be interferometrically interconnected via a suspension platform interferometer. Advanced isolation techniques will be used, such as inverted pendulums and geometrical anti-spring filters in combination with multiple-cascaded pendulum suspensions, containing an all-silica monolithic last stage. The light source is a 35 W Nd:YAG laser, geometrically filtered by passing it through a photonic crystal fibre and a rigid pre-modecleaner cavity. Laser frequency stabilisation will be achieved with the aid of a high finesse suspended reference cavity in conjunction with a molecular iodine reference. Coating thermal noise will be reduced by the use of Khalili cavities as compound end mirrors. Data acquisition and control of the experiments is based on the AdvLIGO digital control and data system. The aim of the project is to test advanced techniques for GEO 600 as well as to conduct experiments in macroscopic quantum mechanics. Reaching standard quantum-limit sensitivity for an interferometer with 100 g mirrors and subsequently breaching this limit, features most prominently among these experiments. In this paper we present the layout and current status of the AEI 10 m Prototype Interferometer project.
A Synthetic Quadrature Phase Detector/Demodulator for Fourier Transform Transform Spectrometers
NASA Technical Reports Server (NTRS)
Campbell, Joel
2008-01-01
A method is developed to demodulate (velocity correct) Fourier transform spectrometer (FTS) data that is taken with an analog to digital converter that digitizes equally spaced in time. This method makes it possible to use simple low cost, high resolution audio digitizers to record high quality data without the need for an event timer or quadrature laser hardware, and makes it possible to use a metrology laser of any wavelength. The reduced parts count and simplicity implementation makes it an attractive alternative in space based applications when compared to previous methods such as the Brault algorithm.
Ultrasonic Interferometers Revisited
NASA Astrophysics Data System (ADS)
Greenslade, Thomas B.
2007-03-01
I have been tinkering with ultrasonic transducers once more. In earlier notes I reported on optics-like experiments performed with ultrasonics, described a number of ultrasonic interferometers,2 and showed how ultrasonic transducers can be used for Fourier analysis.3 This time I became interested in trying the technique of using two detectors in acoustic interferometers instead of the usual one.
NASA Astrophysics Data System (ADS)
Zhen, Shenglai; Chen, Bo; Yuan, Liang; Li, Min; Liang, Jing; Yu, Benli
2010-03-01
In-phase and quadrature-phase (I/Q) signals often need to be formed in the laser interferometric vibration measurement technique. To avoid the disadvantages of traditional I/Q signals forming methods such as effect of piezoelectric ceramic (PZT) for generating high frequency carrier, or optical configuration with complicated structure, a novel interferometric vibration measurement sensor with quadrature detection is proposed. The sensor utilizes simple optical configuration which contains 1/8 wave plate to generate two I/Q signals, then the signals were processed by arctangent algorithm which is compiled by Labview software through data acquisition card. Theoretical analysis and experimental Lissajous figures synthesis prove the phase orthogonality of the two signals. The experimental results indicate that the system can measure the vibration displacement accurately.
Photorefractive Interferometers for Ultrasonic Measurements on Paper
Lafond, E. F.; Brodeur, P. H.; Gerhardstein, J. P.; Habeger, C. C.; Telschow, Kenneth Louis
2002-12-01
Photorefractive interferometers have been employed for the detection of ultrasound in metals and composites since 1991 [1–4]. Instances of laser-generated ultrasound and laser-based detection in paper were reported in 1996 [5]. More recently, bismuth silicon oxide (BSO) photorefractive interferometers were adapted to detect ultrasound in paper [6]. In this article we discuss BSO and GaAs photorefractive detection of ultrasound on different paper grades and present the resulting waveforms. Compared to contact piezoelectric transducer methods, laser interferometry offers signifcant advantages. One of these is that it is a noncontact technique. This is especially important for on-line application to lightweight papers which could be marked or damaged by contact transducers. Broadband ultrasonic laser generation matched with the broadband sensitivity of laser interferometers is another beneft. This is important for obtaining narrow pulses in nondispersive time-of-fight determinations and for measuring the phase velocity of dispersive modes over a wide frequency band. Also, laser ultrasonic techniques provide a measure of bending stiffness through the analysis of low frequency A0 waves.
Positive interpolatory quadrature formulas and para-orthogonal polynomials
NASA Astrophysics Data System (ADS)
Bultheel, Adhemar; Daruis, Leyla; Gonzalez-Vera, Pablo
2005-07-01
We establish a relation between quadrature formulas on the interval [-1,1] that approximate integrals of the form and Szego quadrature formulas on the unit circle that approximate integrals of the form . The functions [mu](x) and [omega]([theta]) are assumed to be weight functions on [-1,1] and [-[pi],[pi
Progress Toward a Cold Ion Interferometer
NASA Astrophysics Data System (ADS)
Archibald, James; Christopher, Erickson; Jackson, Jarom; Durfee, Dallin
2012-06-01
We describe progress on a cold ion matter-wave interferometer. The ions are generated by laser-cooling strontium and then photo-ionizing the atoms with a two-photon transition to an auto- ionizing state in the continuum. Each ion's quantum wave will be split and recombined using stimulated Raman transitions between the hyperfine ground states of Sr^87+. The interferometer phase will be determined by measuring the fraction of ions exiting in each hyperfine state. We will discuss the theory of operation, experimental methods, and potential applications of the device.
Two-frequency-dependent Gauss quadrature rules
NASA Astrophysics Data System (ADS)
Kim, Kyung Joong
2005-02-01
We construct two-frequency-dependent Gauss quadrature rules which can be applied for approximating the integration of the product of two oscillatory functions with different frequencies [beta]1 and [beta]2 of the forms,yi(x)=fi,1(x) cos([beta]ix)+fi,2(x) sin([beta]ix), i=1,2,where the functions fi,j(x) are smooth. A regularization procedure is presented to avoid the singularity of the Jacobian matrix of nonlinear system of equations which is induced as one frequency approaches the other frequency. We provide numerical results to compare the accuracy of the classical Gauss rule and one- and two-frequency-dependent rules.
The May 1997 SOHO-Ulysses Quadrature
NASA Technical Reports Server (NTRS)
Suess, Steven T.; Poletto, G.; Romoli, M.; Neugebauer, M.; Goldstein, B. E.; Simnett, G.
2000-01-01
We present results from the May 1997 SOHO-Ulysses quadrature, near sunspot minimum. Ulysses was at 5.1 AU, 100 north of the solar equator, and off the east limb. It was, by chance, also at the very northern edge of the streamer belt. Nevertheless, SWOOPS detected only slow, relatively smooth wind and there was no direct evidence of fast wind from the northern polar coronal hole or of mixing with fast wind. LASCO images show that the streamer belt at 10 N was narrow and sharp at the beginning and end of the two week observation interval, but broadened in the middle. A corresponding change in density, but not flow speed, occurred at Ulysses. Coronal densities derived from UVCS show that physical parameters in the lower corona are closely related to those in the solar wind, both over quiet intervals and in transient events on the limb. One small transient observed by both LASCO and UVCS is analyzed in detail.
Power flow control using quadrature boosters
NASA Astrophysics Data System (ADS)
Sadanandan, Sandeep N.
A power system that can be controlled within security constraints would be an advantage to power planners and real-time operators. Controlling flows can lessen reliability issues such as thermal limit violations, power stability problems, and/or voltage stability conditions. Control of flows can also mitigate market issues by reducing congestion on some lines and rerouting power to less loaded lines or onto preferable paths. In the traditional control of power flows, phase shifters are often used. More advanced methods include using Flexible AC Transmission System (FACTS) Controllers. Some examples include Thyristor Controlled Series Capacitors, Synchronous Series Static Compensators, and Unified Power Flow Controllers. Quadrature Boosters (QBs) have similar structures to phase-shifters, but allow for higher voltage magnitude during real power flow control. In comparison with other FACTS controllers QBs are not as complex and not as expensive. The present study proposes to use QBs to control power flows on a power system. With the inclusion of QBs, real power flows can be controlled to desired scheduled values. In this thesis, the linearized power flow equations used for power flow analysis were modified for the control problem. This included modifying the Jacobian matrix, the power error vector, and calculating the voltage injected by the quadrature booster for the scheduled real power flow. Two scenarios were examined using the proposed power flow control method. First, the power flow in a line in a 5-bus system was modified with a QB using the method developed in this thesis. Simulation was carried out using Matlab. Second, the method was applied to a 30-bus system and then to a 118-bus system using several QBs. In all the cases, the calculated values of the QB voltages led to desired power flows in the designated line.
High resolution heterodyne interferometer based on time-to-digital converter.
Wang, Fei; Long, Zhangcai; Zhang, Bin; Zhao, Meirong
2012-04-01
A new heterodyne interferometer is presented, which adopts time-to-digital converter (TDC) measuring the time intervals of zero crossings of heterodyne signal for phase demodulation. Thanks to the 0.1 ns time resolution of TDC and linear phase demodulation, it can achieve high resolution and avoids nonlinear measuring distortion in other indirect high precise phase demodulation methods, such as pulse width modulation (PWM) and in-phase∕quadrature (I∕Q) method. PMID:22559581
High resolution heterodyne interferometer based on time-to-digital converter
NASA Astrophysics Data System (ADS)
Wang, Fei; Long, Zhangcai; Zhang, Bin; Zhao, Meirong
2012-04-01
A new heterodyne interferometer is presented, which adopts time-to-digital converter (TDC) measuring the time intervals of zero crossings of heterodyne signal for phase demodulation. Thanks to the 0.1 ns time resolution of TDC and linear phase demodulation, it can achieve high resolution and avoids nonlinear measuring distortion in other indirect high precise phase demodulation methods, such as pulse width modulation (PWM) and in-phase/quadrature (I/Q) method.
NASA Technical Reports Server (NTRS)
Zhao, Feng
2010-01-01
Sub-aperture interferometers -- also called wavefront-split interferometers -- have been developed for simultaneously measuring displacements of multiple targets. The terms "sub-aperture" and "wavefront-split" signify that the original measurement light beam in an interferometer is split into multiple sub-beams derived from non-overlapping portions of the original measurement-beam aperture. Each measurement sub-beam is aimed at a retroreflector mounted on one of the targets. The splitting of the measurement beam is accomplished by use of truncated mirrors and masks, as shown in the example below
Warm Vapor Atom Interferometer
NASA Astrophysics Data System (ADS)
Biedermann, Grant; Wheeler, David; Jau, Yuan-Yu; McGuinness, Hayden
2014-05-01
We present a light pulse atom interferometer using room temperature rubidium vapor. Doppler sensitive stimulated Raman transitions forming the atom optical elements inherently select a cold velocity group for the interferometer. The interferometer is configured to be sensitive to accelerations. Sandia National Laboratories is a multi-program laboratory managed and operated by Sandia Corporation, a wholly owned subsidiary of Lockheed Martin Corporation, for the U.S. Department of Energy's National Nuclear Security Administration under contract DE-AC04-94AL85000.
A heterodyne interferometer for angle metrology
Hahn, Inseob; Weilert, M.; Wang, X.; Goullioud, R.
2010-04-15
We have developed a compact, high-resolution, angle measurement instrument based on a heterodyne interferometer. Common-path heterodyne interferometer metrology is used to measure displacements of a reflective target surface. In the interferometer set up, an optical mask is used to sample the laser beam reflecting back from four areas on a target surface. From the relative displacement measurements of the target surface areas, we can simultaneously determine angular rotations around two orthogonal axes in a plane perpendicular to the measurement beam propagation direction. The device is used in a testbed for a tracking telescope system where pitch and yaw angle measurements of a flat mirror are performed. Angle noise measurement of the device shows 0.1 nrad/{radical}(Hz) at 1 Hz, at a working distance of 1 m. The operation range and nonlinearity of the device when used with a flat mirror is approximately {+-}0.15 mrad, and 3 {mu}rad rms, respectively.
Turner, A.
1999-03-01
The Omega laser is a system with many different parts that may cause imperfections. There are a multitude of lenses and mirrors, for example, that may not be polished correctly and can cause the laser wave front to have aberrations. The Liquid Crystal Point Diffraction Interferometer (L.C.P.D.I.) is a device whose main purpose is to read the wave front of the laser and measure any aberrations that may be on it. The way the L.C.P.D.I. reads the laser wave front and measures these aberrations is very complicated and has yet to be perfected. A ray-tracing model of the L.C.P.D.I. has been built, which calculates and models the ray trajectories, the optical paths of the rays, the O.P.D. between the object and reference beams, the absorption of the rays in the liquid crystal, and the intensities of each beam. It can predict an actual experiment by manipulating the different parameters of the program. It will be useful in optimization and further development of the L.C.P.D.I. Evidently, it is necessary to develop a liquid crystal solution with an O.D. greater than 0.3, and possibly as high as 2.0. This new solution would be able to reduce the intensity of the object beam sufficiently to make it comparable with the reference beam intensity. If this were achieved, the contrast, or visibility of the fringes would be better, and the interferogram could be used to diagnose the aberrations in the laser beam front. Then the cause of the aberrations could be fixed. This would result in a near-perfect laser front. If this were achieved, then it is possible that laser fusion could be made more efficient and possibly used as an energy source.
Furnace control apparatus using polarizing interferometer
Schultz, Thomas J.; Kotidis, Petros A.; Woodroffe, Jaime A.; Rostler, Peter S.
1995-01-01
A system for non-destructively measuring an object and controlling industrial processes in response to the measurement is disclosed in which an impulse laser generates a plurality of sound waves over timed increments in an object. A polarizing interferometer is used to measure surface movement of the object caused by the sound waves and sensed by phase shifts in the signal beam. A photon multiplier senses the phase shift and develops an electrical signal. A signal conditioning arrangement modifies the electrical signals to generate an average signal correlated to the sound waves which in turn is correlated to a physical or metallurgical property of the object, such as temperature, which property may then be used to control the process. External, random vibrations of the workpiece are utilized to develop discernible signals which can be sensed in the interferometer by only one photon multiplier. In addition the interferometer includes an arrangement for optimizing its sensitivity so that movement attributed to various waves can be detected in opaque objects. The interferometer also includes a mechanism for sensing objects with rough surfaces which produce speckle light patterns. Finally the interferometer per se, with the addition of a second photon multiplier is capable of accurately recording beam length distance differences with only one reading.
Process control system using polarizing interferometer
Schultz, T.J.; Kotidis, P.A.; Woodroffe, J.A.; Rostler, P.S.
1994-02-15
A system for nondestructively measuring an object and controlling industrial processes in response to the measurement is disclosed in which an impulse laser generates a plurality of sound waves over timed increments in an object. A polarizing interferometer is used to measure surface movement of the object caused by the sound waves and sensed by phase shifts in the signal beam. A photon multiplier senses the phase shift and develops an electrical signal. A signal conditioning arrangement modifies the electrical signals to generate an average signal correlated to the sound waves which in turn is correlated to a physical or metallurgical property of the object, such as temperature, which property may then be used to control the process. External, random vibrations of the workpiece are utilized to develop discernible signals which can be sensed in the interferometer by only one photon multiplier. In addition the interferometer includes an arrangement for optimizing its sensitivity so that movement attributed to various waves can be detected in opaque objects. The interferometer also includes a mechanism for sensing objects with rough surfaces which produce speckle light patterns. Finally the interferometer per se, with the addition of a second photon multiplier is capable of accurately recording beam length distance differences with only one reading. 38 figures.
Process control system using polarizing interferometer
Schultz, Thomas J.; Kotidis, Petros A.; Woodroffe, Jaime A.; Rostler, Peter S.
1994-01-01
A system for non-destructively measuring an object and controlling industrial processes in response to the measurement is disclosed in which an impulse laser generates a plurality of sound waves over timed increments in an object. A polarizing interferometer is used to measure surface movement of the object caused by the sound waves and sensed by phase shifts in the signal beam. A photon multiplier senses the phase shift and develops an electrical signal. A signal conditioning arrangement modifies the electrical signals to generate an average signal correlated to the sound waves which in turn is correlated to a physical or metallurgical property of the object, such as temperature, which property may then be used to control the process. External, random vibrations of the workpiece are utilized to develop discernible signals which can be sensed in the interferometer by only one photon multiplier. In addition the interferometer includes an arrangement for optimizing its sensitivity so that movement attributed to various waves can be detected in opaque objects. The interferometer also includes a mechanism for sensing objects with rough surfaces which produce speckle light patterns. Finally the interferometer per se, with the addition of a second photon multiplier is capable of accurately recording beam length distance differences with only one reading.
Furnace control apparatus using polarizing interferometer
Schultz, T.J.; Kotidis, P.A.; Woodroffe, J.A.; Rostler, P.S.
1995-03-28
A system for nondestructively measuring an object and controlling industrial processes in response to the measurement is disclosed in which an impulse laser generates a plurality of sound waves over timed increments in an object. A polarizing interferometer is used to measure surface movement of the object caused by the sound waves and sensed by phase shifts in the signal beam. A photon multiplier senses the phase shift and develops an electrical signal. A signal conditioning arrangement modifies the electrical signals to generate an average signal correlated to the sound waves which in turn is correlated to a physical or metallurgical property of the object, such as temperature, which property may then be used to control the process. External, random vibrations of the workpiece are utilized to develop discernible signals which can be sensed in the interferometer by only one photon multiplier. In addition the interferometer includes an arrangement for optimizing its sensitivity so that movement attributed to various waves can be detected in opaque objects. The interferometer also includes a mechanism for sensing objects with rough surfaces which produce speckle light patterns. Finally the interferometer per se, with the addition of a second photon multiplier is capable of accurately recording beam length distance differences with only one reading. 38 figures.
Michelson and His Interferometer
ERIC Educational Resources Information Center
Shankland, Robert S.
1974-01-01
Presents a brief historical account of Michelson's invention of his interferometer with some subsequent ingenious applications of its capabilities for precise measurement discussed in details, including the experiment on detrmination of the diameters for heavenly bodies. (CC)
Frank, A.M.
1980-01-01
This paper describes a technique by which the sensitivity of plasma interferometers can be increased. Stabilization and fractional fringe measurement techniques have improved to the point where additional optical sensitivity could be useful. (MOW)
Phase shifting diffraction interferometer
Sommargren, Gary E.
1996-01-01
An interferometer which has the capability of measuring optical elements and systems with an accuracy of .lambda./1000 where .lambda. is the wavelength of visible light. Whereas current interferometers employ a reference surface, which inherently limits the accuracy of the measurement to about .lambda./50, this interferometer uses an essentially perfect spherical reference wavefront generated by the fundamental process of diffraction. This interferometer is adjustable to give unity fringe visibility, which maximizes the signal-to-noise, and has the means to introduce a controlled prescribed relative phase shift between the reference wavefront and the wavefront from the optics under test, which permits analysis of the interference fringe pattern using standard phase extraction algorithms.
Phase shifting diffraction interferometer
Sommargren, G.E.
1996-08-29
An interferometer which has the capability of measuring optical elements and systems with an accuracy of {lambda}/1000 where {lambda} is the wavelength of visible light. Whereas current interferometers employ a reference surface, which inherently limits the accuracy of the measurement to about {lambda}/50, this interferometer uses an essentially perfect spherical reference wavefront generated by the fundamental process of diffraction. This interferometer is adjustable to give unity fringe visibility, which maximizes the signal-to-noise, and has the means to introduce a controlled prescribed relative phase shift between the reference wavefront and the wavefront from the optics under test, which permits analysis of the interference fringe pattern using standard phase extraction algorithms. 8 figs.
Heterodyne interferometer with angstrom-level periodic nonlinearity
Schmitz, Tony L.; Beckwith, John F.
2005-01-25
Displacement measuring interferometer systems and methods are disclosed. One or more acousto-optic modulators for receiving a laser light beam from a laser light source can be utilized to split the laser light beam into two or more laser light beams, while spatially separating frequencies thereof. One or more reflective mechanisms can be utilized to reflect one or more of the laser light beams back to the acousto-optic modulator. Interference of two or more of the laser light beams generally at the acousto-optic modulator can provide an interfered laser light beam thereof. A detector for receiving the interfered laser light beam can be utilized to provide interferometer measurement data.
Hybrid photonic chip interferometer for embedded metrology
NASA Astrophysics Data System (ADS)
Kumar, P.; Martin, H.; Maxwell, G.; Jiang, X.
2014-03-01
Embedded metrology is the provision of metrology on the manufacturing platform, enabling measurement without the removal of the work piece. Providing closer integration of metrology upon the manufacturing platform can lead to the better control and increased throughput. In this work we present the development of a high precision hybrid optical chip interferometer metrology device. The complete metrology sensor system is structured into two parts; optical chip and optical probe. The hybrid optical chip interferometer is based on a silica-on-silicon etched integrated-optic motherboard containing waveguide structures and evanescent couplers. Upon the motherboard, electro-optic components such as photodiodes and a semiconductor gain block are mounted and bonded to provide the required functionality. The key structure in the device is a tunable laser module based upon an external-cavity diode laser (ECDL). Within the cavity is a multi-layer thin film filter which is rotated to select the longitudinal mode at which the laser operates. An optical probe, which uses a blazed diffracting grating and collimating objective lens, focuses light of different wavelengths laterally over the measurand. Incident laser light is then tuned in wavelength time to effectively sweep an `optical stylus' over the surface. Wavelength scanning and rapid phase shifting can then retrieve the path length change and thus the surface height. We give an overview of the overall design of the final hybrid photonic chip interferometer, constituent components, device integration and packaging as well as experimental test results from the current version now under evaluation.
Algorithm 699 - A new representation of Patterson's quadrature formulae
NASA Technical Reports Server (NTRS)
Krogh, Fred T.; Van Snyder, W.
1991-01-01
A method is presented to reduce the number of coefficients necessary to represent Patterson's quadrature formulae. It also reduces the amount of storage necessary for storing function values, and produces slightly smaller error in evaluating the formulae.
Experimental study of quadrature spring rate at tuned dry gyro
NASA Astrophysics Data System (ADS)
Hayakawa, Yoshiaki; Murayama, Naoshi
A survey result on the mechanism of quadrature spring rate occurring at the tuned dry gyro is given. It is noted that the quadrature spring rate is a damping torque. This damping torque is similar to the spring reaction torque generated by the flexure displacement angles and drives the gyro rotor back to a balanced position. In order to investigate the mechanism of damping occurring at the gyro rotor, the relation between surrounding gas pressure and damping factor under gyro nonoperating was measured. Furthermore, the drag torque acting on the gyro rotor was measured by the back EMF method at different surrounding gas pressure. As a result of these testings, it was found out that the quadrature spring rate was generated by gas movement of the flexure around and drag forces due to bearing loss and windage loss, and the mechanism and magnitude of each damping torque which are contributor to the quadrature spring rate were extracted separately.
Optical encryption system using quadrature multiplexing
NASA Astrophysics Data System (ADS)
Islam, Mohammed Nazrul; Alam, Mohammad S.
2006-08-01
Optical security systems have attracted much research interest recently for information security and fraud deterrent applications. A number of encryption techniques have been proposed in the literature, which includes double random-phase encryption, polarization encoding, encryption and verification using a multiplexed minimum average correlation energy phase-encrypted filter. Most of these reports employ a pseudo-random code for each information to be encrypted, where it requires individual storage capacity or transmission channel for further processing of each information. The objective of this paper is to develop an optical encryption system employing quadrature multiplexing to enhance the storage/transmission capacity of the system. Two information signals are encrypted using the same code but employing two orthogonal functions and then they are multiplexed together in the same domain. As the orthogonal functions have zero cross-correlation between them, so the encrypted information are expected to be unaffected by each other. Each encryption and multiplexing process can accommodate two information signals for a single code and a single storage cell or transmission channel. The same process can be performed in multiple steps to increase the multiplexing capability of the system. For decryption purpose, the composite encoded signal is correlated using the appropriate code and the appropriate function. The proposed technique has been found to work excellent in computer simulation with binary as well as gray level images. It has also been verified that the encrypted images remain secure, because no unwanted reproduction is possible without having the appropriate code and function.
Two integrator loop quadrature oscillators: A review
Soliman, Ahmed M.
2012-01-01
A review of the two integrator loop oscillator circuits providing two quadrature sinusoidal output voltages is given. All the circuits considered employ the minimum number of capacitors namely two except one circuit which uses three capacitors. The circuits considered are classified to four different classes. The first class includes floating capacitors and floating resistors and the active building blocks realizing these circuits are the Op Amp or the OTRA. The second class employs grounded capacitors and includes floating resistors and the active building blocks realizing these circuits are the DCVC or the unity gain cells or the CFOA. The third class employs grounded capacitors and grounded resistors and the active building blocks realizing these circuits are the CCII. The fourth class employs grounded capacitors and no resistors and the active building blocks realizing these circuits are the TA. Transformation methods showing the generation of different classes from each other is given in details and this is one of the main objectives of this paper. PMID:25685396
Novel double path shearing interferometer in corneal topography measurements
NASA Astrophysics Data System (ADS)
Licznerski, Tomasz J.; Jaronski, Jaroslaw; Kosz, Dariusz
2005-09-01
The paper presents an approach for measurements of corneal topography by use of a patent pending double path shearing interferometer (DPSI). Laser light reflected from the surface of the cornea is divided and directed to the inputs of two interferometers. The interferometers use lateral shearing of wavefronts in two orthogonal directions. A tilt of one of the mirrors in each interferometric setup perpendicularly to the lateral shear introduces parallel carrier frequency fringes at the output of each interferometer. There is orthogonal linear polarization of the laser light used in two DPSI. Two images of fringe patters are recorded by a high resolution digital camera. The obtained fringe patterns are used for phase difference reconstruction. The phase of the wavefront was reconstructed by use of algorithms for a large grid based on discrete integration. The in vivo method can also be used for tear film stability measurement, artificial tears and contact lens tests.
Orthogonal rational functions and quadrature on an interval
NASA Astrophysics Data System (ADS)
van Deun, J.; Bultheel, A.
2003-04-01
Rational functions with real poles and poles in the complex lower half-plane, orthogonal on the real line, are well known. Quadrature formulas similar to the Gauss formulas for orthogonal polynomials have been studied. We generalize to the case of arbitrary complex poles and study orthogonality on a finite interval. The zeros of the orthogonal rational functions are shown to satisfy a quadratic eigenvalue problem. In the case of real poles, these zeros are used as nodes in the quadrature formulas.
An exponentially fitted quadrature rule over unbounded intervals
NASA Astrophysics Data System (ADS)
Conte, D.; Paternoster, B.; Santomauro, G.
2012-09-01
A new class of quadrature formulae for the computation of integrals over unbounded intervals with oscillating integrand is illustrated. Such formulae are a generalization of the gaussian quadrature formulae by exploiting the Exponential Fitting theory. The coefficients depend on the frequency of oscillation, in order to improve the accuracy of the solution. The construction of the methods with 1, 2 and 3 nodes is described, together with the comparison of the order of accuracy with respect to classical formulae.
Quadrature mixture LO suppression via DSW DAC noise dither
Dubbert, Dale F.; Dudley, Peter A.
2007-08-21
A Quadrature Error Corrected Digital Waveform Synthesizer (QECDWS) employs frequency dependent phase error corrections to, in effect, pre-distort the phase characteristic of the chirp to compensate for the frequency dependent phase nonlinearity of the RF and microwave subsystem. In addition, the QECDWS can employ frequency dependent correction vectors to the quadrature amplitude and phase of the synthesized output. The quadrature corrections cancel the radars' quadrature upconverter (mixer) errors to null the unwanted spectral image. A result is the direct generation of an RF waveform, which has a theoretical chirp bandwidth equal to the QECDWS clock frequency (1 to 1.2 GHz) with the high Spurious Free Dynamic Range (SFDR) necessary for high dynamic range radar systems such as SAR. To correct for the problematic upconverter local oscillator (LO) leakage, precision DC offsets can be applied over the chirped pulse using a pseudo-random noise dither. The present dither technique can effectively produce a quadrature DC bias which has the precision required to adequately suppress the LO leakage. A calibration technique can be employed to calculate both the quadrature correction vectors and the LO-nulling DC offsets using the radar built-in test capability.
Bose-Einstein-condensate interferometer with macroscopic arm separation
Garcia, O.; Deissler, B.; Hughes, K. J.; Reeves, J. M.; Sackett, C. A.
2006-09-15
A Michelson interferometer using Bose-Einstein condensates is demonstrated with coherence times of up to 44 ms and arm separations up to 180 {mu}m. This arm separation is larger than that observed for any previous atom interferometer. The device uses atoms weakly confined in a magnetic guide and the atomic motion is controlled using Bragg interactions with an off-resonant standing-wave laser beam.
Fast adaptive interferometer with a photorefractive GaP crystal
NASA Astrophysics Data System (ADS)
Kamshilin, Alexei A.; Prokofiev, Victor V.
2002-10-01
The performance of an adaptive interferometer based on mixing of light waves with different polarization states in a photorefractive GaP crystal at λ = 0.633 μm is described. Both high sensitivity and fast response time are achieved with a low-power He-Ne laser. The parameters of the interferometer are appropriate for remote ultrasound detection in an industrial environment.
Overview and Status of Advanced Interferometers for Gravitational Wave Detection
NASA Astrophysics Data System (ADS)
Grote, H.
2016-05-01
The world-wide network of km-scale laser interferometers is aiming at the detection of gravitational waves of astrophysical origin. The second generation of these instruments, called advanced detectors has been, or is in the process of being completed, and a first observational run with the Advanced LIGO interferometers has been performed late in 2015. The basic functionality of advanced detectors is discussed, along with specific features and status updates of the individual projects.
Optical tweezers based on polarization interferometer
NASA Astrophysics Data System (ADS)
Angelsky, Oleg V.; Maksimyak, Andrew P.; Maksimyak, Peter P.; Dominikov, Mykola M.
2013-06-01
In this paper, we propose optical tweezers based on a biaxial crystal. To control the movement of opaque particles, we use the shift polarization interferometer. The results of experimental study of laser tweezers are shown. We demonstrates movement of a microparticle of toner using singular-optical trap, rotate a particle due to orbital momentum, conversion of two traps when changing the plane of polarizer transmission and converging of two traps.
High-speed and high-resolution heterodyne interferometer
NASA Astrophysics Data System (ADS)
Yokoyama, Shuko; Nishihara, I.; Okamoto, A.; Araki, Tsutomu; Suzuki, Norihito
1990-07-01
In conventional heterodyne interferometer it Is necessary to provide a high-beat frequency laser when measurement for a high-speed target Is required1 . But use of the high-beat frequency laser makes the " nanometerdivisions" difficult. We have developed a novel interferometer system that has a sufficient response to high-speed movement of the target without Increase of the laser beat frequency. In this work a two frequency laser light passes through the same optical path of the interferometer so that two conjugate beat signals are obtained. By processing the multiple beat signals with a newly developed signal processor the above methodological contradiction is solved. l. OPTICS FOR LINEAR DISTANCE MEASUREMENT Optical system of the interferometer Is shown In Flg. l in which two frequency lights Fl and F2 (freq. f and f2) of orthogonally linear-polarized components of a two mode laser are used. A reference beat signal R (freq. r ) is generated from Fl and F2. Fl and F2 are passed through exactly the same path of the interferometer. A half power of Fl and F2 Is reflected and the rest Is transmitted In the beam splitter ( BS ). Polarization direction of the reflected beam is rotated for ir/2 after passing the X/4 plate twice. The transmitted beam Is reflected by a moving mirror (MM) mounted on the moving target so that frequency of the reflected beam Is shifted by Doppler
Shalloo, R J; Corner, L
2016-09-01
We present a double unbalanced Michelson interferometer producing up to four output pulses from a single input pulse. The interferometer is stabilized with the Hänsch-Couillaud method using an auxiliary low power continuous wave laser injected into the interferometer, allowing the stabilization of the temporal jitter of the output pulses to 0.02 fs. Such stabilized pulse trains would be suitable for driving multi-pulse laser wakefield accelerators, and the technique could be extended to include amplification in the arms of the interferometer. PMID:27607974
Joo, K; Ellis, J D; Buice, E S; Spronck, J W; Munnig Schmidt, R H
2010-02-05
This paper describes a novel heterodyne laser interferometer with no significant periodic nonlinearity for linear displacement measurements. Moreover, the optical configurations have the benefit of doubling the measurement resolution when compared to its respective traditional counterparts. Experimental results show no discernable periodic nonlinearity for a retro-reflector interferometer and plane mirror interferometer configurations with a noise level below 20 pm. The incoming laser beams of the interferometers are achieved by utilizing two single mode optical fibers. To determine the stability of the optical fiber couplers a fiber delivery prototype was also built and tested.
PDX multichannel interferometer
Bitzer, R.; Ernst, W.; Cutsogeorge, G.
1980-10-01
A 10 channel, 140 GHz homodyne interferometer is described for use on PDX. One feature of this interferometer is the separation of the signal source and electronics from the power splitters, delay line, and receiving systems. The latter is situated near the upper and lower vacuum ports between the toroidal field magnets. A second feature is the signal stabilization of the EIO source by means of an AFC system. The complete interferometer is described including block diagrams, circuit diagrams, test data, and magnetic field test conducted on the preamplifiers, microwave diodes, isolators, etc., to determine the extent of magnetic shielding required. The description of the tracking filters and digital phase display circuit is referenced to accompanying reports.
Optical interferometer testbed
NASA Technical Reports Server (NTRS)
Blackwood, Gary H.
1991-01-01
Viewgraphs on optical interferometer testbed presented at the MIT Space Research Engineering Center 3rd Annual Symposium are included. Topics covered include: space-based optical interferometer; optical metrology; sensors and actuators; real time control hardware; controlled structures technology (CST) design methodology; identification for MIMO control; FEM/ID correlation for the naked truss; disturbance modeling; disturbance source implementation; structure design: passive damping; low authority control; active isolation of lightweight mirrors on flexible structures; open loop transfer function of mirror; and global/high authority control.
Ultrasonic Interferometers Revisited
ERIC Educational Resources Information Center
Greenslade, Thomas B., Jr.
2007-01-01
I have been tinkering with ultrasonic transducers once more. In earlier notes I reported on optics-like experiments performed with ultrasonics, described a number of ultrasonic interferometers, and showed how ultrasonic transducers can be used for Fourier analysis. This time I became interested in trying the technique of using two detectors in…
Gaussian quadrature inference for continuous-variable quantum key distribution
NASA Astrophysics Data System (ADS)
Gyongyosi, L.; Imre, S.
2016-05-01
We propose the Gaussian quadrature inference (GQI) method for multicarrier continuous-variable quantum key distribution (CVQKD). A multicarrier CVQKD protocol utilizes Gaussian subcarrier quantum continuous variables (CV) for information transmission. The GQI framework provides a minimal error estimate of the quadratures of the CV quantum states from the discrete, measured noisy subcarrier variables. GQI utilizes the fundamentals of regularization theory and statistical information processing. We characterize GQI for multicarrier CVQKD, and define a method for the statistical modeling and processing of noisy Gaussian subcarrier quadratures. We demonstrate the results through the adaptive multicarrier quadrature division (AMQD) scheme. We introduce the terms statistical secret key rate and statistical private classical information, which quantities are derived purely by the statistical functions of GQI. We prove the secret key rate formulas for a multiple access multicarrier CVQKD via the AMQD-MQA (multiuser quadrature allocation) scheme. The framework can be established in an arbitrary CVQKD protocol and measurement setting, and are implementable by standard low-complexity statistical functions, which is particularly convenient for an experimental CVQKD scenario.
The Fall 2000 and Fall 2001 SOHO-Ulysses Quadratures
NASA Technical Reports Server (NTRS)
Suess, S. T.; Poletto, G.
2000-01-01
SOHO-Ulysses quadrature occurs when the SOHO-Sun-Ulysses included angle is 90 degrees. It is only at such times that the same plasma leaving the Sun in the direction of Ulysses can first be remotely analyzed with SOHO instruments and then later be sampled in situ by Ulysses instruments. The quadratures in December 2000 and 2001 are of special significance because Ulysses will be near the south and north heliographic poles, respectively, and the solar cycle will be near sunspot maximum. Quadrature geometry is sometimes confusing and observations are influenced by solar rotation. The Fall 2000 and 2001 quadratures are more complex than usual because Ulysses is not in a true polar orbit and the orbital speed of Ulysses about the Sun is becoming comparable to the speed of SOHO about the Sun. In 2000 Ulysses will always be slightly behind the pole but will appear to hang over the pole for over two months because it is moving around the Sun in the same direction as SOHO. In 20001, Ulysses will be slightly in front of the pole so that its footpoint will be directly observable. Detailed plots will be shown of the relative positions of SOHO and Ulysses will their relative positions. In neither case is true quadrature actually achieved, but this works to the observers advantage in 2001.
The Fall 2000 and Fall 2001 SOHO-Ulysses Quadratures
NASA Technical Reports Server (NTRS)
Suess, S. T.; Poletto, G.; Rose, M. Franklin (Technical Monitor)
2001-01-01
SOHO-Ulysses quadrature occurs when the SOHO-Sun-Ulysses included angle is 90 degrees. It is only at such times that the same plasma leaving the Sun in the direction of Ulysses can first be remotely analyzed with SOHO instruments and then later be sampled in situ by Ulysses instruments. The quadratures in December 2000 and 2001 are of special significance because Ulysses will be near the south and north heliographic poles, respectively, and the solar cycle will be near sunspot maximum. Quadrature geometry is sometimes confusing and observations are influenced by solar rotation. The Fall 2000 and 2001 quadratures are more complex than usual because Ulysses is not in a true polar orbit and the orbital speed of Ulysses about the Sun is becoming comparable to the speed of SOHO about the Sun. In 2000 Ulysses will always be slightly behind the pole but will appear to hang over the pole for over two months because it is moving around the Sun in the same direction as SOHO. In 2001 Ulysses will be slightly in front of the pole so that its footpoint will be directly observable. Detailed plots will be shown of the relative positions of SOHO and Ulysses will their relative positions. In neither case is true quadrature actually achieved, but this works to the observers advantage in 2001.
Gauss Quadratures - the Keystone of Lattice Boltzmann Models
NASA Astrophysics Data System (ADS)
Piaud, Benjamin; Blanco, Stéphane; Fournier, Richard; Ambruş, Victor Eugen; Sofonea, Victor
2014-01-01
In this paper, we compare two families of Lattice Boltzmann (LB) models derived by means of Gauss quadratures in the momentum space. The first one is the HLB(N;Qx,Qy,Qz) family, derived by using the Cartesian coordinate system and the Gauss-Hermite quadrature. The second one is the SLB(N;K,L,M) family, derived by using the spherical coordinate system and the Gauss-Laguerre, as well as the Gauss-Legendre quadratures. These models order themselves according to the maximum order N of the moments of the equilibrium distribution function that are exactly recovered. Microfluidics effects (slip velocity, temperature jump, as well as the longitudinal heat flux that is not driven by a temperature gradient) are accurately captured during the simulation of Couette flow for Knudsen number (kn) up to 0.25.
Achromatic self-referencing interferometer
Feldman, M.
1994-04-19
A self-referencing Mach-Zehnder interferometer is described for accurately measuring laser wavefronts over a broad wavelength range (for example, 600 nm to 900 nm). The apparatus directs a reference portion of an input beam to a reference arm and a measurement portion of the input beam to a measurement arm, recombines the output beams from the reference and measurement arms, and registers the resulting interference pattern ([open quotes]first[close quotes] interferogram) at a first detector. Optionally, subportions of the measurement portion are diverted to second and third detectors, which respectively register intensity and interferogram signals which can be processed to reduce the first interferogram's sensitivity to input noise. The reference arm includes a spatial filter producing a high quality spherical beam from the reference portion, a tilted wedge plate compensating for off-axis aberrations in the spatial filter output, and mirror collimating the radiation transmitted through the tilted wedge plate. The apparatus includes a thermally and mechanically stable baseplate which supports all reference arm optics, or at least the spatial filter, tilted wedge plate, and the collimator. The tilted wedge plate is mounted adjustably with respect to the spatial filter and collimator, so that it can be maintained in an orientation in which it does not introduce significant wave front errors into the beam propagating through the reference arm. The apparatus is polarization insensitive and has an equal path length configuration enabling measurement of radiation from broadband as well as closely spaced laser line sources. 3 figures.
Achromatic self-referencing interferometer
Feldman, Mark
1994-01-01
A self-referencing Mach-Zehnder interferometer for accurately measuring laser wavefronts over a broad wavelength range (for example, 600 nm to 900 nm). The apparatus directs a reference portion of an input beam to a reference arm and a measurement portion of the input beam to a measurement arm, recombines the output beams from the reference and measurement arms, and registers the resulting interference pattern ("first" interferogram) at a first detector. Optionally, subportions of the measurement portion are diverted to second and third detectors, which respectively register intensity and interferogram signals which can be processed to reduce the first interferogram's sensitivity to input noise. The reference arm includes a spatial filter producing a high quality spherical beam from the reference portion, a tilted wedge plate compensating for off-axis aberrations in the spatial filter output, and mirror collimating the radiation transmitted through the tilted wedge plate. The apparatus includes a thermally and mechanically stable baseplate which supports all reference arm optics, or at least the spatial filter, tilted wedge plate, and the collimator. The tilted wedge plate is mounted adjustably with respect to the spatial filter and collimator, so that it can be maintained in an orientation in which it does not introduce significant wave front errors into the beam propagating through the reference arm. The apparatus is polarization insensitive and has an equal path length configuration enabling measurement of radiation from broadband as well as closely spaced laser line sources.
Discrete Ordinate Quadrature Selection for Reactor-based Eigenvalue Problems
Jarrell, Joshua J; Evans, Thomas M; Davidson, Gregory G
2013-01-01
In this paper we analyze the effect of various quadrature sets on the eigenvalues of several reactor-based problems, including a two-dimensional (2D) fuel pin, a 2D lattice of fuel pins, and a three-dimensional (3D) reactor core problem. While many quadrature sets have been applied to neutral particle discrete ordinate transport calculations, the Level Symmetric (LS) and the Gauss-Chebyshev product (GC) sets are the most widely used in production-level reactor simulations. Other quadrature sets, such as Quadruple Range (QR) sets, have been shown to be more accurate in shielding applications. In this paper, we compare the LS, GC, QR, and the recently developed linear-discontinuous finite element (LDFE) sets, as well as give a brief overview of other proposed quadrature sets. We show that, for a given number of angles, the QR sets are more accurate than the LS and GC in all types of reactor problems analyzed (2D and 3D). We also show that the LDFE sets are more accurate than the LS and GC sets for these problems. We conclude that, for problems where tens to hundreds of quadrature points (directions) per octant are appropriate, QR sets should regularly be used because they have similar integration properties as the LS and GC sets, have no noticeable impact on the speed of convergence of the solution when compared with other quadrature sets, and yield more accurate results. We note that, for very high-order scattering problems, the QR sets exactly integrate fewer angular flux moments over the unit sphere than the GC sets. The effects of those inexact integrations have yet to be analyzed. We also note that the LDFE sets only exactly integrate the zeroth and first angular flux moments. Pin power comparisons and analyses are not included in this paper and are left for future work.
A continuous cold atomic beam interferometer
Xue, Hongbo; Feng, Yanying Yan, Xueshu; Jiang, Zhikun; Chen, Shu; Wang, Xiaojia; Zhou, Zhaoying
2015-03-07
We demonstrate an atom interferometer that uses a laser-cooled continuous beam of {sup 87}Rb atoms having velocities of 10–20 m/s. With spatially separated Raman beams to coherently manipulate the atomic wave packets, Mach–Zehnder interference fringes are observed at an interference distance of 2L = 19 mm. The apparatus operates within a small enclosed area of 0.07 mm{sup 2} at a bandwidth of 190 Hz with a deduced sensitivity of 7.8×10{sup −5} rad/s/√(Hz) for rotations. Using a low-velocity continuous atomic source in an atom interferometer enables high sampling rates and bandwidths without sacrificing sensitivity and compactness, which are important for applications in real dynamic environments.
Analysis of a free oscillation atom interferometer
Kafle, Rudra P.; Zozulya, Alex A.; Anderson, Dana Z.
2011-09-15
We analyze a Bose-Einstein condensate (BEC)-based free oscillation atom Michelson interferometer in a weakly confining harmonic magnetic trap. A BEC at the center of the trap is split into two harmonics by a laser standing wave. The harmonics move in opposite directions with equal speeds and turn back under the influence of the trapping potential at their classical turning points. The harmonics are allowed to pass through each other and a recombination pulse is applied when they overlap at the end of a cycle after they return for the second time. We derive an expression for the contrast of the interferometric fringes and obtain the fundamental limit of performance of the interferometer in the parameter space.
Space interferometer mission (SIM) instrument design concepts.
NASA Astrophysics Data System (ADS)
Duncan, A. L.
SIM is a 12 meter baseline interferometer to be built as part of the NASA Origins program, designed to fly in space and provide high precision astrometry measurements of astronomical objects. SIM will provide angular measurements three orders of magnitude more precise than current space or ground based sensors, allowing the indirect detection of Earth-like planets around neighboring stars. The SIM mission will also include the ability to synthesize images by varying the interferometer baseline lengths and will demonstrate a nulling beam combiner as a technology pathfinder for future missions. A team at Lockheed Martin Missiles and Space (LMMS) in Sunnyvale, CA has been chosen by JPL to enter a partnership to design and build the SIM instrument. This paper describes the overall LMMS SIM instrument concept and its unique features, including the full aperture laser metrology approach for high precision metrology.
Error estimates for Gaussian quadratures of analytic functions
NASA Astrophysics Data System (ADS)
Milovanovic, Gradimir V.; Spalevic, Miodrag M.; Pranic, Miroslav S.
2009-12-01
For analytic functions the remainder term of Gaussian quadrature formula and its Kronrod extension can be represented as a contour integral with a complex kernel. We study these kernels on elliptic contours with foci at the points ±1 and the sum of semi-axes [varrho]>1 for the Chebyshev weight functions of the first, second and third kind, and derive representation of their difference. Using this representation and following Kronrod's method of obtaining a practical error estimate in numerical integration, we derive new error estimates for Gaussian quadratures.
The development of accurate and efficient methods of numerical quadrature
NASA Technical Reports Server (NTRS)
Feagin, T.
1973-01-01
Some new methods for performing numerical quadrature of an integrable function over a finite interval are described. Each method provides a sequence of approximations of increasing order to the value of the integral. Each approximation makes use of all previously computed values of the integrand. The points at which new values of the integrand are computed are selected in such a way that the order of the approximation is maximized. The methods are compared with the quadrature methods of Clenshaw and Curtis, Gauss, Patterson, and Romberg using several examples.
An Algorithm to Evaluate Imbalances of Quadrature Mixers
NASA Astrophysics Data System (ADS)
Asami, Koji; Arai, Michiaki
It is essential, as bandwidths of wireless communications get wider, to evaluate the imbalances among quadrature mixer ports, in terms of carrier phase offset, IQ gain imbalance, and IQ skew. Because it is time consuming to separate skew, gain imbalance and carrier phase offset evaluation during test is often performed using a composite value, without separation of the imbalance factors. This paper describes an algorithm for enabling separation among quadrature mixer gain imbalance, carrier phase offset, and skew. Since the test time is reduced by the proposed method, it can be applied during high volume production testing.
Multipulsed dynamic moire interferometer
Deason, Vance A.
1991-01-01
An improved dynamic moire interferometer comprised of a lasing medium providing a plurality of beams of coherent light, a multiple q-switch producing multiple trains of 100,000 or more pulses per second, a combining means collimating multiple trains of pulses into substantially a single train and directing beams to specimen gratings affixed to a test material, and a controller, triggering and sequencing the emission of the pulses with the occurrence and recording of a dynamic loading event.
NASA Astrophysics Data System (ADS)
Colavita, M. M.; Wizinowich, P. L.; Akeson, R. L.; Ragland, S.; Woillez, J. M.; Millan-Gabet, R.; Serabyn, E.; Abajian, M.; Acton, D. S.; Appleby, E.; Beletic, J. W.; Beichman, C. A.; Bell, J.; Berkey, B. C.; Berlin, J.; Boden, A. F.; Booth, A. J.; Boutell, R.; Chaffee, F. H.; Chan, D.; Chin, J.; Chock, J.; Cohen, R.; Cooper, A.; Crawford, S. L.; Creech-Eakman, M. J.; Dahl, W.; Eychaner, G.; Fanson, J. L.; Felizardo, C.; Garcia-Gathright, J. I.; Gathright, J. T.; Hardy, G.; Henderson, H.; Herstein, J. S.; Hess, M.; Hovland, E. E.; Hrynevych, M. A.; Johansson, E.; Johnson, R. L.; Kelley, J.; Kendrick, R.; Koresko, C. D.; Kurpis, P.; Le Mignant, D.; Lewis, H. A.; Ligon, E. R.; Lupton, W.; McBride, D.; Medeiros, D. W.; Mennesson, B. P.; Moore, J. D.; Morrison, D.; Nance, C.; Neyman, C.; Niessner, A.; Paine, C. G.; Palmer, D. L.; Panteleeva, T.; Papin, M.; Parvin, B.; Reder, L.; Rudeen, A.; Saloga, T.; Sargent, A.; Shao, M.; Smith, B.; Smythe, R. F.; Stomski, P.; Summers, K. R.; Swain, M. R.; Swanson, P.; Thompson, R.; Tsubota, K.; Tumminello, A.; Tyau, C.; van Belle, G. T.; Vasisht, G.; Vause, J.; Vescelus, F.; Walker, J.; Wallace, J. K.; Wehmeier, U.; Wetherell, E.
2013-10-01
The Keck Interferometer (KI) combined the two 10 m W. M. Keck Observatory telescopes on Mauna Kea, Hawaii, as a long-baseline near- and mid-infrared interferometer. Funded by NASA, it operated from 2001 until 2012. KI used adaptive optics on the two Keck telescopes to correct the individual wavefronts, as well as active fringe tracking in all modes for path-length control, including the implementation of cophasing to provide long coherent integration times. KI implemented high sensitivity fringe-visibility measurements at H (1.6 μm), K (2.2 μm), and L (3.8 μm) bands, and nulling measurements at N band (10 μm), which were used to address a broad range of science topics. Supporting these capabilities was an extensive interferometer infrastructure and unique instrumentation, including some additional functionality added as part of the NSF-funded ASTRA program. This paper provides an overview of the instrument architecture and some of the key design and implementation decisions, as well as a description of all of the key elements and their configuration at the end of the project. The objective is to provide a view of KI as an integrated system, and to provide adequate technical detail to assess the implementation. Included is a discussion of the operational aspects of the system, as well as of the achieved system performance. Finally, details on V2 calibration in the presence of detector nonlinearities as applied in the data pipeline are provided.
Interferometer density measurements of a high-velocity plasmoid
NASA Astrophysics Data System (ADS)
Case, A.; Messer, S.; Bomgardner, R.; Witherspoon, F. D.
2010-05-01
The plasmoid produced by a half-scale contoured gap coaxial plasma accelerator using ablative polyethylene capillary plasma injectors is measured using a quadrature heterodyne HeNe interferometer. The plasmoid is found to have a sharp rise in density at the leading edge, with a gradual falloff after the peak density. For this early test series, an average bulk density of 5×1014 cm-3 is observed, with densities up to 8×1014 cm-3 seen on some shots. Although plasmoid mass is only about 58 μg due to the low current and injected mass used in these tests, good shot-to-shot repeatability is attained making analysis relatively straightforward, thus providing a solid foundation for interpreting future experimental results.
Atom-Light Hybrid Interferometer.
Chen, Bing; Qiu, Cheng; Chen, Shuying; Guo, Jinxian; Chen, L Q; Ou, Z Y; Zhang, Weiping
2015-07-24
A new type of hybrid atom-light interferometer is demonstrated with atomic Raman amplification processes replacing the beam splitting elements in a traditional interferometer. This nonconventional interferometer involves correlated optical and atomic waves in the two arms. The correlation between atoms and light developed with the Raman process makes this interferometer different from conventional interferometers with linear beam splitters. It is observed that the high-contrast interference fringes are sensitive to the optical phase via a path change as well as the atomic phase via a magnetic field change. This new atom-light correlated hybrid interferometer is a sensitive probe of the atomic internal state and should find wide applications in precision measurement and quantum control with atoms and photons. PMID:26252684
Trapezoidal rule quadrature algorithms for MIMD distributed memory computers
Lyness, J.N.; Plowman, S.E.
1994-08-01
An approach to multi-dimensional quadrature, designed to exploit parallel architectures, is described. This involves transforming the integral in such a way that an accurate result is given by the trapezoidal rule; and by evaluating the resulting sum in a manner which may be efficiently implemented on parallel architectures. This approach is to be implemented in the Liverpool NAG transputer library.
From Lobatto Quadrature to the Euler Constant "e"
ERIC Educational Resources Information Center
Khattri, Sanjay Kumar
2010-01-01
Based on the Lobatto quadrature, we develop several new closed form approximations to the mathematical constant "e." For validating effectiveness of our approximations, a comparison of our results to the existing approximations is also presented. Another objective of our work is to inspire students to formulate other better approximations by using…
Wave-Based Inversion & Imaging for the Optical Quadrature Microscope
Lehman, S K
2005-10-27
The Center for Subsurface Sensing & Imaging System's (CenSSIS) Optical Quadrature Microscope (OQM) is a narrow band visible light microscope capable of measuring both amplitude and phase of a scattered field. We develop a diffraction tomography, that is, wave-based, scattered field inversion and imaging algorithm, for reconstructing the refractive index of the scattering object.
Gaussian rational quadrature formulas for ill-scaled integrands
NASA Astrophysics Data System (ADS)
Illán González, J. R.
2009-12-01
A flexible treatment of Gaussian quadrature formulas based on rational functions is given to evaluate the integral , when f is meromorphic in a neighborhood V of the interval I and W(x) is an ill-scaled weight function. Some numerical tests illustrate the power of this approach in comparison with Gautschi's method.
Archimedes Quadrature of the Parabola: A Mechanical View
ERIC Educational Resources Information Center
Oster, Thomas J.
2006-01-01
In his famous quadrature of the parabola, Archimedes found the area of the region bounded by a parabola and a chord. His method was to fill the region with infinitely many triangles each of whose area he could calculate. In his solution, he stated, without proof, three preliminary propositions about parabolas that were known in his time, but are…
Applying Quadrature Rules with Multiple Nodes to Solving Integral Equations
Hashemiparast, S. M.; Avazpour, L.
2008-09-01
There are many procedures for the numerical solution of Fredholm integral equations. The main idea in these procedures is accuracy of the solution. In this paper, we use Gaussian quadrature with multiple nodes to improve the solution of these integral equations. The application of this method is illustrated via some examples, the related tables are given at the end.
A Robust Ramsey Interferometer for Atomic Timekeeping in Dynamic Environments
NASA Astrophysics Data System (ADS)
Kotru, Krish; Brown, Justin; Butts, David; Choy, Jennifer; Galfond, Marissa; Johnson, David M.; Kinast, Joseph; Timmons, Brian; Stoner, Richard
2014-05-01
We present a laser-based approach to atomic timekeeping, in which atomic phase information is extracted using modified Raman pulses in a Ramsey sequence. We overcome systematic effects associated with differential AC Stark shifts by employing atom optics derived from Raman adiabatic rapid passage (ARP). ARP drives coherent transfer between two hyperfine ground states by sweeping the frequency difference of two optical fields and maintaining a large single-photon detuning. Compared to resonant, pulsed Raman transitions, ARP atom optics afford a >150x reduction in sensitivity to differential AC Stark shifts in a Ramsey interferometer. We also demonstrate that ARP preserves fringe contrast in Ramsey interferometers for cloud displacements reaching the 1/e2 intensity radius of the laser beam. ARP can thus be expected to improve the robustness of clock interferometers operating in dynamic environments. Copyright ©2014 by The Charles Stark Draper Laboratory, Inc. All rights reserved.
Shilyagin, P A; Gelikonov, G V; Gelikonov, V M; Moiseev, A A; Terpelov, D A
2014-07-31
We have thoroughly investigated the method of simultaneous reception of spectral components with the achromatised quadrature phase shift between two portions of a reference wave, designed for the effective suppression of the 'mirror' artefact in the resulting image obtained by means of spectral domain optical coherence tomography (SD OCT). We have developed and experimentally tested a phase-shifting element consisting of a beam divider, which splits the reference optical beam into the two beams, and of delay lines being individual for each beam, which create a mutual phase difference of π/2 in the double pass of the reference beam. The phase shift achromatism over a wide spectral range is achieved by using in the delay lines the individual elements with different dispersion characteristics. The ranges of admissible adjustment parameters of the achromatised delay line are estimated for exact and inexact conformity of the geometric characteristics of its components to those calculated. A possibility of simultaneous recording of the close-to-quadrature spectral components with a single linear photodetector element is experimentally confirmed. The suppression of the artefact mirror peak in the OCT-signal by an additional 9 dB relative to the level of its suppression is experimentally achieved when the air delay line is used. Two-dimensional images of the surface positioned at an angle to the axis of the probe beam are obtained with the correction of the 'mirror' artefact while maintaining the dynamic range of the image. (laser biophotonics)
Lambda/2 fringe-spacing interferometer.
de Chatellus, H Guillet; Pique, J-P
2009-03-15
The precision of interferometry is directly linked to the fringe spacing of the recorded interferogram. Whereas all interferometric devices show a fringe spacing equal to a wavelength of the laser light we present a novel scheme of a two-beam interferometer exhibiting a fringe spacing reduced by a factor of 2; the direct detection of the beat signal is replaced with the monitoring of the fluorescence of a twofold degenerate atomic system resonant with the laser. The lambda/2 fringe spacing in the fluorescence signal is demonstrated with a hot sodium vapor excited by a broadband laser tuned to the D1 line. In the saturation regime, the dark fringes are expected to be extremely narrow, leading to the possibility of nanoscale displacement measurements or atom localization. PMID:19282922
White light interferometer: applications in research and industry
NASA Astrophysics Data System (ADS)
Bandyopadhyay, Sujit
2015-06-01
Applications of interferometer are countless both in the research and commercial world. Laser sources offer precise measurements of relative path difference between two interfering beams. An exciting example is LIGO (laser Interferometer for Gravitational Observatory), which is aiming to resolve length change as small as 10-19 m over a 4 km length for detection of gravitational waves. However, laser is a disadvantage for microscopic imaging and surface topography applications usually required in semiconductor industry. A different approach for microscopy is to use white light in place of laser. White light due to its limited temporal coherence offers a multitude of benefits for imaging applications. An immediate benefit from white light is the sharp localisation of interference fringe that makes the 3D topography construction or OCT (Optical Coherence Topography) realisable using a Scanning White Light Interferometer (SWLI) imager. In Mirau Mode, SWLI performs high resolution imaging; whereas in Michelson mode Fourier Transform Spectroscopy (FTS) is realised. SWLI can easily be modified into PUPS (Pupil Plane SWLI) for Ellipsometry. Superimposing Michelson Interferometer known as VISAR (Velocity Interferometer System for Any reflector) can form interference fringes even in presence of wide angle light scattered from a moving illuminated object. This paper describes work undertaken at Nanometrics (UK) on simulation of SWLI fringes including high Numerical Aperture (NA) applications, thin film characterisation, OCT generation and Zemax modelling of compact dispersion-free vibration-immune Fourier-Transformed spectrometer. VISAR as a modified Mach-Zehnder Interferometer is also discussed based on the work at Rutherford-Appleton laboratory (UK).
The Design and Operation of Ultra-Sensitive and Tunable Radio-Frequency Interferometers
Cui, Yan; Wang, Pingshan
2015-01-01
Dielectric spectroscopy (DS) is an important technique for scientific and technological investigations in various areas. DS sensitivity and operating frequency ranges are critical for many applications, including lab-on-chip development where sample volumes are small with a wide range of dynamic processes to probe. In this work, we present the design and operation considerations of radio-frequency (RF) interferometers that are based on power-dividers (PDs) and quadrature-hybrids (QHs). Such interferometers are proposed to address the sensitivity and frequency tuning challenges of current DS techniques. Verified algorithms together with mathematical models are presented to quantify material properties from scattering parameters for three common transmission line sensing structures, i.e., coplanar waveguides (CPWs), conductor-backed CPWs, and microstrip lines. A high-sensitivity and stable QH-based interferometer is demonstrated by measuring glucose–water solution at a concentration level that is ten times lower than some recent RF sensors while our sample volume is ~1 nL. Composition analysis of ternary mixture solutions are also demonstrated with a PD-based interferometer. Further work is needed to address issues like system automation, model improvement at high frequencies, and interferometer scaling. PMID:26549891
Long-baseline optical fiber interferometer instruments and science
NASA Astrophysics Data System (ADS)
Kotani, Takayuki; Nishikawa, Jun; Sato, Koichi; Yoshizawa, Masanori; Ohishi, Naoko; Fukushima, Toshio; Torii, Yasuo; Matsuda, Ko; Kubo, Koichi; Iwashita, Hikaru; Suzuki, Shunsaku
2003-02-01
Developments of fiber linked optical interferometer are reported. This interferometer is a part of MIRA-I.2 interferometer (Mitaka InfraRed and optical Array). MIRA-I.2 is an optical interferometer with a 30 meters long baseline. It consists of two 30cm siderostats, tip-tilt mirrors, vacuum pipes delay lines and detectors. We plan to use two 60 meters long polarization-maintaining fibers for arms of the interferometer, instead of vacuum pipes. The developments include dispersion and polarization compensation of fiber and fiber injection module. In laboratory experiments, dispersion compensation succeeded. The fringe visibility was 0.93 for wide-band, where the central wavelength of light was 700nm, and bandwidth was 200nm, while 0.95 with a He-Ne laser. We used BK7 glass wedge for dispersion compensation. About fiber injection module, basic optical design has completed. The results of our fiber interferometer could contribute to OHANA (Optical Hawaiian Array for Nanoradian Astronomy) project. We present new science targets, white dwarves and T Tauri stars, and an 800 m delayline concept in CFHT for the project.
First gravity measurements using the mobile atom interferometer GAIN
NASA Astrophysics Data System (ADS)
Hauth, M.; Freier, C.; Schkolnik, V.; Senger, A.; Schmidt, M.; Peters, A.
2013-10-01
We present the compact Gravimetric Atom Interferometer (GAIN), based on laser-cooled 87Rb atoms, and discuss its first measurements of the local gravitational acceleration. In this context, we also describe an active vibration isolation system and a tip-tilt stage, which allow for a suppression of vibrational noise and systematic effects like the Coriolis force due to Earth's rotation.
MIKES’ primary phase stepping gauge block interferometer
NASA Astrophysics Data System (ADS)
Byman, V.; Lassila, A.
2015-08-01
MIKES’ modernized phase stepping interferometer for gauge block calibration (PSIGB) will be described. The instrument is based on the well-regarded NPL-TESA gauge block interferometer from 1994. The decision to upgrade the instrument resulted from several components, such as the PC and charge-coupled device (CCD) camera, having reached the end of their lifetime. In this paper modernized components, measurement method and analysis will be explained. The lasers are coupled to the instrument using single mode fiber. The instrument uses phase stepping generated by an added optical window on a controllable rotatory table in the reference arm with a recently developed nine-position phase stepping algorithm. Unwrapping is done with a robust path following algorithm. Procedures for adjusting the interferometer are explained. Determination and elimination of wavefront error, coherent noise and analysis of their influence on the results is described. Flatness and variation in length are also important parameters of gauge blocks to be characterized, and the corresponding analysis method is clarified. Uncertainty analysis for the central length, flatness and variation in length is also described. The results are compared against those of the old hardware and software. The standard uncertainty for central length measurement is u = [(9.5 nm)2 + (121 × 10-9 L)2]½, where L is measured length.
Integrated Optical Heterodyne Interferometer in Lithium Niobate
NASA Astrophysics Data System (ADS)
Rubiyanto, A.; Herrmann, H.; Ricken, R.; Tian, F.; Sohler, W.
A high performance integrated acousto-optical heterodyne interferometer has been developed for vibration measurement. All components including an acousto-optical TE-TM mode converters, two electro-optical TE-TM converters, two polarization splitters and two phase shifters are integrated on a X-cut Lithium Niobate substrate. The fully packaged optical integrated circuit (optical-IC) coupling with three fibers optics pigtails gave a signal-to-noise ratio of 69 dB with at 3 kHz bandwidth by using a commercial DFB laser diode as a light source with 1561 nm emission wavelength and a PIN-FET balanced receiver.
High-accuracy position-sensing with fiber-coupled white-light interferometers
NASA Astrophysics Data System (ADS)
Bosselmann, Th.; Ulrich, R.
A fiber-optic system for high-accuracy position sensing has been constructed using the principle of white-light interferometry. In the system, white light from a W-lamp is fed via multimode optical fibers consecutively through two conventional Michelson interferometers to a detector. The element whose position is to be sensed displaces one mirror of the transmitting interferometer in the sensor head; the receiving interferometer, located in a remote control room, is scanned periodically, and the appearance of white-light fringes is used to mark the coincidence of the optical path differences. To measure the transmitted positions, a HeNe laser beam is fed additionally through the receiving interferometer and the fringes are counted. The system provides the usual high accuracy of laser interferometers, yet it does not 'forget' its zero point when interrupted. The use of multimode fibers provides enough optical power to permit operation multiplex of several simple transmitters with a single, more sophisticated receiver.
NASA Astrophysics Data System (ADS)
Smith, Elwyn; Wada, Naoya; Chujo, Wataru; Sampson, David D.
2002-06-01
High-power ultra-broadband sources such as a supercontinuum are very attractive in optical coherence tomography (OCT) and optical coherence-domain reflectometry (OCDR) due to their very high resolution potential. However, the large and extensive coherence-function sidelobes typical of such sources preclude their use in conventional OCDR and OCT systems. In addition, device or sample dispersion over such broad bandwidths may also significantly limit the achievable performance. Here we describe a novel experiment using a supercontinuum source with a static Michelson interferometer to perform OCDR at 1.55micrometers . Quadrature spectral detection is used to maximize the scanning range and to allow direct compensation for both the undesirable spectral shape of the source and for the dispersion in the system. Such a non-scanning-interferometer approach is an interesting possible alternative for very broadband, ultra-high resolution OCT systems. We demonstrate that an otherwise obscured small reflection next to a large reflection can be revealed by appropriately weighting the data to reshape the supercontinuum spectrum and compensate for dispersion. Significant reduction of the supercontinuum coherence function sidelobes is achieved, and a resolution in air of 7micrometers (FWHM) is obtained, or less than 5micrometers in media of refractive index 1.45.
Extraction of quadrature phase information from multiple pulse NMR signals
NASA Technical Reports Server (NTRS)
Rhim, W.-K.; Burum, D. P.; Vaughan, R. W.
1976-01-01
A multiple pulse sequence (8-pulse sequence) used for high-resolution solid state NMR is analyzed with regard to the information available from each of the four wide sampling windows. It is demonstrated that full quadrature phase information can be obtained using only a single phase detector and that, for the commonly encountered situation where the spectral width is much less than the folding frequency, the signals from the various windows can be combined easily using standard complex Fourier transform software. An improvement in the signal-to-noise ratio equal to the square root of 3 is obtained over either standard single or quadrature phase detection schemes. Procedures for correcting spectral distortions are presented.
On a quadrature formula of Gori and Micchelli
NASA Astrophysics Data System (ADS)
Yang, Shijun
2005-04-01
Sparked by Bojanov (J. Comput. Appl. Math. 70 (1996) 349), we provide an alternate approach to quadrature formulas based on the zeros of the Chebyshev polynomial of the first kind for any weight function w introduced and studied in Gori and Micchelli (Math. Comp. 65 (1996) 1567), thereby improving on their observations. Upon expansion of the divided differences, we obtain explicit expressions for the corresponding Cotes coefficients in Gauss-Turan quadrature formulas for and I(fTn;w) for a Gori-Micchelli weight function. It is also interesting to mention what has been neglected for about 30 years by the literature is that, as a consequence of expansion of the divided differences in the special case when , the solution of the famous Turan's Problem 26 raised in 1980 was in fact implied by a result of Micchelli and Rivlin (IBM J. Res. Develop. 16 (1972) 372) in 1972. Some concluding comments are made in the final section.
Accurate Computation of Gaussian Quadrature for Tension Powers
NASA Astrophysics Data System (ADS)
Singer, Saša
2007-09-01
We consider Gaussian quadrature formulæ which exactly integrate a system of tension powers 1,x,x2,…,xn-3, sinh(px), cosh(px), on a given interval [a,b], where n⩾4 is an even integer and p>0 is a given tension parameter. In some applications it is essential that p can be changed dynamically, and we need an efficient "on-demand" algorithm that calculates the nodes and weights of Gaussian quadrature formulas for many different values of p, which are not known in advance. It is an interesting numerical challenge to achieve the required full machine precision accuracy in such an algorithm, for all possible values of p. By exploiting various analytic and numerical techniques, we show that this can be done efficiently for all reasonably low values of n that are of any practical importance.
Quadrature formulae for classes of functions of low smoothness
Nursultanov, E D; Tleukhanova, N T
2003-10-31
For Sobolev and Korobov spaces of functions of several variables a quadrature formula with explicitly defined coefficients and nodes is constructed. This formula is precise for trigonometric polynomials with harmonics from the corresponding step hyperbolic cross. The error of the quadrature formula in the classes W{sup {alpha}}{sub p}[0,1]{sup n}, E{sup {alpha}}[0,1]{sup n} is o((ln M){sup {beta}}/M{sup {alpha}}), where M is the number of nodes and {beta} is a parameter depending on the class. The problem of the approximate calculation of multiple integrals for functions in W{sup {alpha}}{sub p}[0,1]{sup n} is considered in the case when this class does not lie in the space of continuous functions, that is, for {alpha}{<=}1/p.
Efficient quadrature multichannel processor algorithms for MCD applications
NASA Astrophysics Data System (ADS)
Corden, I. R.; Carrasco, R. A.
1992-06-01
The forthcoming third generation of satellites incorporating multichannel demodulator (MCD) processors, and the needs apparent within aviation systems, induce the requirement for efficient band processing algorithms with specific regard to the quadrature processing arrangement. This paper presents a coherent z-domain formulation of the pertinent digital transmultiplexer algorithms for the on-board processing (OBP) scenario, with a view to establishig a set of desirable algorithmic properties suitable for the preferred complex oriented quadrature processing algorithms. Stemming from the principles set forth, an ensemble of new algorithms based upon mixes of Hilbert transforming and real transform algorithms is presented, wherein the established concepts relating to the telephone network transmultiplexer algorithms are able to be exploited in certain cases. Further, the computational load of one of the methods is lower than that of a known prominent OBP related technique. The computational necessities of the various algorithms are laid down for comparative purposes in addition to the mathematical descriptions.
Dual differential interferometer for measurements of broadband surface acoustic waves
NASA Technical Reports Server (NTRS)
Turner, T. M.; Claus, R. O.
1981-01-01
A simple duel interferometer which uses two pairs of orthogonally polarized optical beams to measure both the amplitude and direction of propagation of broadband ultrasonic surface waves is described. Each pair of focused laser probe beams is used in a separate wideband differential interferometer to independently detect the component of surface wave motion along one direction on the surface. By combining the two output signals corresponding to both components, the two dimensional surface profile and its variation as a function of time is determined.
Rational choices for the wavelengths of a two color interferometer
Jobes, F.C.
1995-07-01
If in a two color interferometer for plasma density measurements, the two wavelengths are chosen to have a ratio that is a rational number, and if the signals from each of the wavelengths are multiplied in frequency by the appropriate integer of the rational number and then heterodyned together, the resultant signal will have all effects of component motion nulled out. A phase measurement of this signal will have only plasma density information in it. With CO{sub 2} lasers, it is possible to find suitable wavelength pairs which are close enough to rational numbers to produce an improvement of about 100 in density resolution, compared to standard two color interferometers.
Optimization of the HCN interferometer on J-TEXT tokamak
NASA Astrophysics Data System (ADS)
Shi, P.; Gao, L.; Xiong, C. Y.; Liu, Y.; Chen, J.; Zhuang, G.
2015-12-01
Recently, the HCN interferometer on J-TEXT has been optimized in many aspects. Firstly, the output power of laser source is more stable after using a new designed movable mirror frame and upgrading the oil thermostatic system. Secondly, the electromagnetic interferences have been eliminated by designing a shielding system. Additionally, the signal-to-noise ratio of intermediate frequency (IF) signal has been increased more than five times by improving the detector circuit. The density resolution has been increased from 1×1018 m-3 to 2×1017 m-3 and the sawtooth oscillation has also been measured by the HCN interferometer on J- TEXT after optimizations.
A new multichannel interferometer system on HL-2A
Zhou, Y.; Deng, Z. C.; Liu, Z. T.; Yi, J.; Tang, Y. W.; Gao, B. Y.; Tian, C. L.; Li, Y. G.; Ding, X. T.
2007-11-15
A new multichannel HCN interferometer has been developed on HL-2A tokamak, which is characterized by two techniques: (1) the wave-guide HCN laser with cavity length of 6 m to increase the optical resource power and (2) high response room temperature waveguide Schottky diode detectors to obtain good beat signal. The space resolution is 7 cm by the use of focusing metal mirrors mounted on the vacuum chamber and a compensated optical system. In the 2006 experiment campaign, this new interferometer has been applied for plasma density profile and density sawtooth measurement.
A SIMPLE HETERODYNE TEMPORAL SPECKLE-PATTERN INTERFEROMETER
Wong, W. O.; Gao, Z.; Lu, J.
2010-05-28
A common light path design of heterodyne speckle pattern interferometer based on temporal speckle pattern interferometry is proposed for non-contact, full-field and real-time continuous displacement measurement. Double frequency laser is produced by rotating a half wave plate. An experiment was carried out to measure the dynamic displacement of a cantilever plate for testing the proposed common path heterodyne speckle pattern interferometer. The accuracy of displacement measurement was checked by measuring the motion at the mid-point of the plate with a point displacement sensor.
The high accuracy data processing system of laser interferometry signals based on MSP430
NASA Astrophysics Data System (ADS)
Qi, Yong-yue; Lin, Yu-chi; Zhao, Mei-rong
2009-07-01
Generally speaking there are two orthogonal signals used in single-frequency laser interferometer for differentiating direction and electronic subdivision. However there usually exist three errors with the interferential signals: zero offsets error, unequal amplitude error and quadrature phase shift error. These three errors have a serious impact on subdivision precision. Based on Heydemann error compensation algorithm, it is proposed to achieve compensation of the three errors. Due to complicated operation of the Heydemann mode, a improved arithmetic is advanced to decrease the calculating time effectively in accordance with the special characteristic that only one item of data will be changed in each fitting algorithm operation. Then a real-time and dynamic compensatory circuit is designed. Taking microchip MSP430 as the core of hardware system, two input signals with the three errors are turned into digital quantity by the AD7862. After data processing in line with improved arithmetic, two ideal signals without errors are output by the AD7225. At the same time two original signals are turned into relevant square wave and imported to the differentiating direction circuit. The impulse exported from the distinguishing direction circuit is counted by the timer of the microchip. According to the number of the pulse and the soft subdivision the final result is showed by LED. The arithmetic and the circuit are adopted to test the capability of a laser interferometer with 8 times optical path difference and the measuring accuracy of 12-14nm is achieved.
Best quadrature formula on Sobolev class with Chebyshev weight
NASA Astrophysics Data System (ADS)
Xie, Congcong
2008-05-01
Using best interpolation function based on a given function information, we present a best quadrature rule of function on Sobolev class KWr[-1,1] with Chebyshev weight. The given function information means that the values of a function f[set membership, variant]KWr[-1,1] and its derivatives up to r-1 order at a set of nodes x are given. Error bounds are obtained, and the method is illustrated by some examples.
Some new applications of truncated Gauss-Laguerre quadrature formulas
NASA Astrophysics Data System (ADS)
Mastroianni, G.; Monegato, G.
2008-12-01
We show how truncated Gauss-Laguerre quadrature formulas can be used to produce accurate approximations and high rates of convergence, also when they are applied to integrand functions having only an algebraic type decay to zero at infinity. The approach presented in the paper is proposed for the computation of integrals and for the construction of Nyström type interpolants for some second kind integral equations.
Solar Wind Characteristics from SOHO-Sun-Ulysses Quadrature Observations
NASA Technical Reports Server (NTRS)
Poletto, Giannina; Suess, Steve T.; Six, N. Frank (Technical Monitor)
2002-01-01
Over the past few years, we have been running SOHO (Solar and Heliospheric Observatory)-Sun-Ulysses quadrature campaigns, aimed at comparing the plasma properties at coronal altitudes with plasma properties at interplanetary distances. Coronal plasma has been observed by SOHO experiments: mainly, we used LASCO (Large Angle and Spectrometric Coronagraph Experiment) data to understand the overall coronal configuration at the time of quadratures and analyzed SUMER (Solar Ultraviolet Measurements of Emitted Radiation), CDS (Coronal Diagnostic Spectrometer) and UVCS (Ultraviolet Coronagraph Spectrometer) data to derive its physical characteristics. At interplanetary distances, SWICS (Solar Wind Ion Composition Spectrometer) and SWOOPS (Solar Wind Observation over the Poles of the Sun) aboard Ulysses provided us with interplanetary plasma data. Here we report on results from some of the campaigns. We notice that, depending on the geometry of the quadrature, i.e. on whether the radial to Ulysses traverses the corona at high or low latitudes, we are able to study different kinds of solar wind. In particular, a comparison between low-latitude and high-latitude wind, allowed us to provide evidence for differences in the acceleration of polar, fast plasma and equatorial, slow plasma: the latter occurring at higher levels and through a more extended region than fast wind. These properties are shared by both the proton and heavy ions outflows. Quadrature observations may provide useful information also on coronal vs. in situ elemental composition. To this end, we analyzed spectra taken in the corona, at altitudes ranging between approx. 1.02 and 2.2 solar radii, and derived the abundances of a number of ions, including oxygen and iron. Values of the O/Fe ratio, at coronal levels, have been compared with measurements of this ratio made by SWICS at interplanetary distances. Our results are compared with previous findings and predictions from modeling efforts.
Statistical Quadrature Evolution for Continuous-Variable Quantum Key Distribution
NASA Astrophysics Data System (ADS)
Gyongyosi, Laszlo; Imre, Sandor
2016-05-01
We propose a statistical quadrature evolution (SQE) method for multicarrier continuous-variable quantum key distribution (CVQKD). A multicarrier CVQKD protocol utilizes Gaussian subcarrier quantum continuous variables (CV) for information transmission. The SQE framework provides a minimal error estimate of the quadratures of the CV quantum states from the discrete, measured noisy subcarrier variables. We define a method for the statistical modeling and processing of noisy Gaussian subcarrier quadratures. We introduce the terms statistical secret key rate and statistical private classical information, which quantities are derived purely by the statistical functions of our method. We prove the secret key rate formulas for a multiple access multicarrier CVQKD. The framework can be established in an arbitrary CVQKD protocol and measurement setting, and are implementable by standard low-complexity statistical functions, which is particularly convenient for an experimental CVQKD scenario. This work was partially supported by the GOP-1.1.1-11-2012-0092 project sponsored by the EU and European Structural Fund, by the Hungarian Scientific Research Fund - OTKA K-112125, and by the COST Action MP1006.
Discrete ordinates with new quadrature sets and modified source conditions
Ganguly, K.; Allen, E.J., Victory, H.D. Jr. )
1989-01-01
A major shortcoming of the discrete ordinates method with the Gauss-Legendre quadrature set is that when the number of secondaries per primary c and the order of approximation N are not too large, all the (N + 1)v (the flux being of the form exp({minus}x/v)) lie in ({minus}1,1). It is known, however, that the largest v{sub j} corresponding to the asymptotic flux is greater than unity. The Legendre polynomial used for obtaining the quadrature set is orthogonal with respect to weight unity in the range ({minus}1,1). However, the Case and Zweifel eigenfunctions derived from the exact solution of one-speed transport theory are orthogonal with respect to a complicated weight function w({mu}) and {mu} in the half-range and full-range cases, respectively. In this paper, the authors have used a set of orthogonal polynomials with respect to w ({mu}) to develop quadrature sets to be used in the discrete ordinates calculation.
Fast evaluation of quadrature formulae on the sphere
NASA Astrophysics Data System (ADS)
Keiner, Jens; Potts, Daniel
2008-03-01
Recently, a fast approximate algorithm for the evaluation of expansions in terms of standard mathrm{L}^2left(mathbb{S}^2right) -orthonormal spherical harmonics at arbitrary nodes on the sphere mathbb{S}^2 has been proposed in [S. Kunis and D. Potts. Fast spherical Fourier algorithms. JE Comput. Appl. Math., 161:75-98, 2003]. The aim of this paper is to develop a new fast algorithm for the adjoint problem which can be used to compute expansion coefficients from sampled data by means of quadrature rules. We give a formulation in matrix-vector notation and an explicit factorisation of the spherical Fourier matrix based on the former algorithm. Starting from this, we obtain the corresponding factorisation of the adjoint spherical Fourier matrix and are able to describe the associated algorithm for the adjoint transformation which can be employed to evaluate quadrature rules for arbitrary weights and nodes on the sphere. We provide results of numerical tests showing the stability of the obtained algorithm using as examples classical Gauss-Legendre and Clenshaw-Curtis quadrature rules as well as the HEALPix pixelation scheme and an equidistribution.
Testing the Empirical Shock Arrival Model Using Quadrature Observations
NASA Technical Reports Server (NTRS)
Gopalswamy, N.; Makela, P.; Xie, H.; Yashiro, S.
2013-01-01
The empirical shock arrival (ESA) model was developed based on quadrature data from Helios (in situ) and P-78 (remote sensing) to predict the Sun-Earth travel time of coronal mass ejections (CMEs). The ESA model requires earthward CME speed as input, which is not directly measurable from coronagraphs along the Sun-Earth line. The Solar Terrestrial Relations Observatory (STEREO) and the Solar and Heliospheric Observatory (SOHO) were in quadrature during 20102012, so the speeds of Earth-directed CMEs were observed with minimal projection effects. We identified a set of 20 full halo CMEs in the field of view of SOHO that were also observed in quadrature by STEREO. We used the earthward speed from STEREO measurements as input to the ESA model and compared the resulting travel times with the observed ones from L1 monitors. We find that the model predicts the CME travel time within about 7.3 h, which is similar to the predictions by the ENLIL model. We also find that CME-CME and CME-coronal hole interaction can lead to large deviations from model predictions.
An Application of the Quadrature-Free Discontinuous Galerkin Method
NASA Technical Reports Server (NTRS)
Lockard, David P.; Atkins, Harold L.
2000-01-01
The process of generating a block-structured mesh with the smoothness required for high-accuracy schemes is still a time-consuming process often measured in weeks or months. Unstructured grids about complex geometries are more easily generated, and for this reason, methods using unstructured grids have gained favor for aerodynamic analyses. The discontinuous Galerkin (DG) method is a compact finite-element projection method that provides a practical framework for the development of a high-order method using unstructured grids. Higher-order accuracy is obtained by representing the solution as a high-degree polynomial whose time evolution is governed by a local Galerkin projection. The traditional implementation of the discontinuous Galerkin uses quadrature for the evaluation of the integral projections and is prohibitively expensive. Atkins and Shu introduced the quadrature-free formulation in which the integrals are evaluated a-priori and exactly for a similarity element. The approach has been demonstrated to possess the accuracy required for acoustics even in cases where the grid is not smooth. Other issues such as boundary conditions and the treatment of non-linear fluxes have also been studied in earlier work This paper describes the application of the quadrature-free discontinuous Galerkin method to a two-dimensional shear layer problem. First, a brief description of the method is given. Next, the problem is described and the solution is presented. Finally, the resources required to perform the calculations are given.
The Fizeau Interferometer Testbed
NASA Technical Reports Server (NTRS)
Zhang, Xiaolei; Carpenter, Kenneth G.; Lyon, Richard G,; Huet, Hubert; Marzouk, Joe; Solyar, Gregory
2003-01-01
The Fizeau Interferometer Testbed (FIT) is a collaborative effort between NASA's Goddard Space Flight Center, the Naval Research Laboratory, Sigma Space Corporation, and the University of Maryland. The testbed will be used to explore the principles of and the requirements for the full, as well as the pathfinder, Stellar Imager mission concept. It has a long term goal of demonstrating closed-loop control of a sparse array of numerous articulated mirrors to keep optical beams in phase and optimize interferometric synthesis imaging. In this paper we present the optical and data acquisition system design of the testbed, and discuss the wavefront sensing and control algorithms to be used. Currently we have completed the initial design and hardware procurement for the FIT. The assembly and testing of the Testbed will be underway at Goddard's Instrument Development Lab in the coming months.
Radio Seeing Monitor Interferometer
NASA Astrophysics Data System (ADS)
Hiriart, David; Valdez, Jorge; Zaca, Placido; Medina, José L.
2002-10-01
A two-element interferometer for monitoring atmospheric phase fluctuations (radio seeing) is presented; this uses the unmodulated beacon signal at 11.715 GHz from a geostationary satellite. The system measures phase differences on the signal received by two small antennas separated by 50 m. The system incorporates the best features from previous designs: a heterodyne phase-lock receiver and an IQ demodulator system. Phase fluctuations measured at this frequency may be extrapolated to millimetric and submillimetric wavelengths since the atmosphere is not dispersive at these frequencies. The instrument has been tested at the Observatory San Pedro Martir (Mexico) at 2800 m above sea level. The final destination of the instrument is Cerro la Negra (Mexico), where the Large Millimeter Telescope is under construction, at an altitude of 4600 m.
The Keck Interferometer Nuller
NASA Technical Reports Server (NTRS)
Serabyn, E.; Mennesson, B.; Colavita, M. M.; Koresko, C.; Kuchner, M. J.
2012-01-01
The Keck Interferometer Nuller (KIN), the first operational separated-aperture infrared nulling interferometer, was designed to null the mid-infrared emission from nearby stars so as to ease the measurement of faint circumstellar emission. This paper describes the basis of the KIN's four-beam, two-stage measurement approach and compares it 10 the simpler case of a two-beam nuller. In the four-beam KIN system, the starlight is first nulled in a pair of nullers operating on parallel 85 m Keck-Keck baselines, after which "cross-combination" on 4 m baselines across the Keck apertures is used to modulate and detect residual coherent off-axis emission. Comparison to the constructive itellar fringe provides calibration. The response to an extended source is similar in the two cases, except that the four-beam response includes a term due to the visibility of the source on the cross-combiner baseline-a small effect for relatively compact sources. The characteristics of the dominant null depth errors are also compared for the two cases. In the two-beam nuller, instrumental imperfections and asymmetries lead to a series of quadratic, positivedefinite null leakage terms. For the four-beam nuller, the leakage is instead a series of correlation cross-tenns combining corresponding errors in each of the two nullers, which contribute offsets only to the extent that these errors are correlated on the timescale of the measurement. This four-beam architecture has allowed a significant (approx. order of magnitude) improvement in mid-infrared long-baseline fringe-visibility accuracies.
MIT's interferometer CST testbed
NASA Technical Reports Server (NTRS)
Hyde, Tupper; Kim, ED; Anderson, Eric; Blackwood, Gary; Lublin, Leonard
1990-01-01
The MIT Space Engineering Research Center (SERC) has developed a controlled structures technology (CST) testbed based on one design for a space-based optical interferometer. The role of the testbed is to provide a versatile platform for experimental investigation and discovery of CST approaches. In particular, it will serve as the focus for experimental verification of CSI methodologies and control strategies at SERC. The testbed program has an emphasis on experimental CST--incorporating a broad suite of actuators and sensors, active struts, system identification, passive damping, active mirror mounts, and precision component characterization. The SERC testbed represents a one-tenth scaled version of an optical interferometer concept based on an inherently rigid tetrahedral configuration with collecting apertures on one face. The testbed consists of six 3.5 meter long truss legs joined at four vertices and is suspended with attachment points at three vertices. Each aluminum leg has a 0.2 m by 0.2 m by 0.25 m triangular cross-section. The structure has a first flexible mode at 31 Hz and has over 50 global modes below 200 Hz. The stiff tetrahedral design differs from similar testbeds (such as the JPL Phase B) in that the structural topology is closed. The tetrahedral design minimizes structural deflections at the vertices (site of optical components for maximum baseline) resulting in reduced stroke requirements for isolation and pointing of optics. Typical total light path length stability goals are on the order of lambda/20, with a wavelength of light, lambda, of roughly 500 nanometers. It is expected that active structural control will be necessary to achieve this goal in the presence of disturbances.
A thermal beam calcium matter-wave interferometer
NASA Astrophysics Data System (ADS)
Birrell, Jeremiah; Christensen, Dan; Erickson, Christopher; Paul, Justin; Tang, Rebecca; Durfee, Dallin
2006-10-01
We report on progress toward a calcium-beam atom interferometer. The design uses a novel alignment scheme using precision prisms which will cause first-order Doppler shifts to cancel out to high accuracy. The device will utilize a thermal beam of atoms for simplicity and high signals. The atom waves will be split and recombined using a single-photon transition at a wavelength of 657 nm. We are currently working to improve the linewidth of the 657 nm laser and constructing a 423 nm blue laser to transversely cool the atoms and to detect the output of the interferometer. We are also characterizing a thermal Ca beam using laser absorption and working on precise control of the temperature and flux of the beam.
Advanced quadratures and periodic boundary conditions in parallel 3D S{sub n} transport
Manalo, K.; Yi, C.; Huang, M.; Sjoden, G.
2013-07-01
Significant updates in numerical quadratures have warranted investigation with 3D Sn discrete ordinates transport. We show new applications of quadrature departing from level symmetric (S{sub 2}o). investigating 3 recently developed quadratures: Even-Odd (EO), Linear-Discontinuous Finite Element - Surface Area (LDFE-SA), and the non-symmetric Icosahedral Quadrature (IC). We discuss implementation changes to 3D Sn codes (applied to Hybrid MOC-Sn TITAN and 3D parallel PENTRAN) that can be performed to accommodate Icosahedral Quadrature, as this quadrature is not 90-degree rotation invariant. In particular, as demonstrated using PENTRAN, the properties of Icosahedral Quadrature are suitable for trivial application using periodic BCs versus that of reflective BCs. In addition to implementing periodic BCs for 3D Sn PENTRAN, we implemented a technique termed 'angular re-sweep' which properly conditions periodic BCs for outer eigenvalue iterative loop convergence. As demonstrated by two simple transport problems (3-group fixed source and 3-group reflected/periodic eigenvalue pin cell), we remark that all of the quadratures we investigated are generally superior to level symmetric quadrature, with Icosahedral Quadrature performing the most efficiently for problems tested. (authors)
New system of miniaturization online testing of spherical interferometer
NASA Astrophysics Data System (ADS)
Lin, Jian; Han, Zhenhua; Zhuo, Jinzhai; Wang, Min
2011-11-01
With the aim to measure the radius of curvature deviation and the surface form deviation of optical elements online in the workshop, based on laser testing technique, two solutions of miniaturization without precision guider sphericity interferometer systems are described, and the principle of systems are analyzed. Compared with the traditional sphericity interferometer, the systems designed are particularly suitable for testing mirror noncontact online in workshop, advantaged in compact structure, no precision guider, high efficiency and low costs. The interferometer systems use the different radius of curvature of existing spherical test plates in the production workshop to produce reference wavefront. The systems can not only get the astigmatism and local deviation of lens, but also can reflect the radius of curvature deviation, which provides rapid and nondestructive noncontact online testing.
White light velocity interferometer
Erskine, David J.
1997-01-01
The invention is a technique that allows the use of broadband and incoherent illumination. Although denoted white light velocimetry, this principle can be applied to any wave phenomenon. For the first time, powerful, compact or inexpensive sources can be used for remote target velocimetry. These include flash and arc lamps, light from detonations, pulsed lasers, chirped frequency lasers, and lasers operating simultaneously in several wavelengths. The technique is demonstrated with white light from an incandescent source to measure a target moving at 16 m/s.
White light velocity interferometer
Erskine, D.J.
1997-06-24
The invention is a technique that allows the use of broadband and incoherent illumination. Although denoted white light velocimetry, this principle can be applied to any wave phenomenon. For the first time, powerful, compact or inexpensive sources can be used for remote target velocimetry. These include flash and arc lamps, light from detonations, pulsed lasers, chirped frequency lasers, and lasers operating simultaneously in several wavelengths. The technique is demonstrated with white light from an incandescent source to measure a target moving at 16 m/s. 41 figs.
White light velocity interferometer
Erskine, David J.
1999-01-01
The invention is a technique that allows the use of broadband and incoherent illumination. Although denoted white light velocimetry, this principle can be applied to any wave phenomenon. For the first time, powerful, compact or inexpensive sources can be used for remote target velocimetry. These include flash and arc lamps, light from detonations, pulsed lasers, chirped frequency lasers, and lasers operating simultaneously in several wavelengths. The technique is demonstrated with white light from an incandescent source to measure a target moving at 16 m/s.
White light velocity interferometer
Erskine, D.J.
1999-06-08
The invention is a technique that allows the use of broadband and incoherent illumination. Although denoted white light velocimetry, this principle can be applied to any wave phenomenon. For the first time, powerful, compact or inexpensive sources can be used for remote target velocimetry. These include flash and arc lamps, light from detonations, pulsed lasers, chirped frequency lasers, and lasers operating simultaneously in several wavelengths. The technique is demonstrated with white light from an incandescent source to measure a target moving at 16 m/s. 41 figs.
Dual interferometer system for measuring index of refraction
NASA Astrophysics Data System (ADS)
Goodwin, Eric Peter
The optical power of a lens is determined by the surface curvature and the refractive index, n. Knowledge of the index is required for accurate lens design models and for examining material variations from sample to sample. The refractive index of glass can be accurately measured using a prism spectrometer, but measuring the index of soft contact lens materials presents many challenges. These materials are non-rigid, thin, and must remain hydrated in a saline solution during testing. Clearly an alternative to a prism spectrometer must be used to accurately measure index. A Dual Interferometer System has been designed, built and characterized as a novel method for measuring the refractive index of transparent optical materials, including soft contact lens materials. The first interferometer is a Low Coherence Interferometer in a Twyman-Green configuration with a scanning reference mirror. The contact lens material sample is placed in a measurement cuvette, where it remains hydrated. By measuring the locations of the multiple optical interfaces, the physical thickness t of the material is measured. A new algorithm has been developed for processing the low coherence signals obtained from the reflection at each optical interface. The second interferometer is a Mach-Zehnder interferometer with a tunable HeNe laser light source. This interferometer measures the optical path length (OPL) of the test sample in the cuvette in transmission as a function of five wavelengths in the visible spectrum. This is done using phase-shifting interferometry. Multiple thickness regions are used to solve 2pi phase ambiguities in the OPL. The outputs of the two interferometers are combined to determine the refractive index as a function of wavelength: n(lambda) = OPL(lambda)/t. Since both t and OPL are measured using a detector array, n is measured at hundreds of thousands of data points. A measurement accuracy of 0.0001 in refractive index is achieved with this new instrument, which is
Investigating the frequency-dependent amplification of a tapered amplifier in atom interferometers.
Zhan, Su; Duan, Xiao-Chun; Zhou, Min-Kang; Yao, Hui-Bin; Xu, Wen-Jie; Hu, Zhong-Kun
2015-01-01
We present the investigation on the frequency-dependent amplification (FDA) of a tapered amplifier (TA) and the corresponding influence on Raman-type atom interferometers. In our interferometer, the output of two phase-locked diode lasers is injected into a TA to generate Raman beams. The frequency of one laser is chirped during the interfering process, which induces a variance of the Raman lasers power as a result of the FDA of the TA. The corresponding power ratio variation of the Raman lasers is measured by beat note method, which shows a linear dependence with a slope of -0.087(4)/GHz when the laser frequency changes over 2 GHz at 780 nm. The corresponding error related to AC Stark effect due to this frequency-dependent variation is estimated for our atom interferometer. The investigation presented here may provide hints for other experiments involving TAs. PMID:25531600
Interferometer-Controlled Optical Tweezers Constructed for Nanotechnology and Biotechnology
NASA Technical Reports Server (NTRS)
Decker, Arthur J.
2002-01-01
A new method to control microparticles was developed in-house at the NASA Glenn Research Center in support of the nanotechnology project under NASA's Aerospace Propulsion and Power Base Research Program. A prototype interferometer-controlled optical tweezers was constructed to manipulate scanning probe microscope (SPM) tips. A laser beam passed through a Mach-Zehnder interferometer, and a microscope objective then produced an optical trap from the coaxial beams. The trap levitated and generated the coarse motion of a 10-mm polystyrene sphere used to simulate a SPM tip. The interference between the beams provided fine control of the forces and moments on the sphere. The interferometer included a piezoelectric-scanned mirror to modulate the interference pattern. The 10-mm sphere was observed to oscillate about 1 mm as the mirror and fringe pattern oscillated. The prototype tweezers proved the feasibility of constructing a more sophisticated interferometer tweezers to hold and manipulate SPM tips. The SPM tips are intended to interrogate and manipulate nanostructures. A more powerful laser will be used to generate multiple traps to hold nanostructures and SPM tips. The vibrating mirror in the interferometer will be replaced with a spatial light modulator. The modulator will allow the optical phase distribution in one leg of the interferometer to be programmed independently at 640 by 480 points for detailed control of the forces and moments. The interference patterns will be monitored to measure the motion of the SPM tips. Neuralnetwork technology will provide fast analysis of the interference patterns for diagnostic purposes and for local or remote feedback control of the tips. This effort also requires theoretical and modeling support in the form of scattering calculations for twin coherent beams from nonspherical particles.
Mielke, S L; Ryan, R E; Hilgeman, T; Lesyna, L; Madonna, R G; Van Nostrand, W C
1997-11-01
A simple technique based on a Fizeau interferometer to measure the absolute phase shift on reflection for a Fabry-Perot interferometer dielectric stack mirror is described. Excellent agreement between the measured and predicted phase shift on reflection was found. Also described are the salient features of low-order Fabry-Perot interferometers and the demonstration of a near ideal low-order (1-10) Fabry-Perot interferometer through minimizing the phase dispersion on reflection of the dielectric stack. This near ideal performance of a low-order Fabry-Perot interferometer should enable several applications such as compact spectral imagers for solid and gas detection. The large free spectral range of such systems combined with an active control system will also allow simple interactive tuning of wavelength agile laser sources such as CO(2) lasers, external cavity diode lasers, and optical parametric oscillators. PMID:18264347
Polarizing optical interferometer having a dual use optical element
Kotidis, Petros A.; Woodroffe, Jaime A.; Rostler, Peter S.
1995-01-01
A system for non-destructively measuring an object and controlling industrial processes in response to the measurement is disclosed in which an impulse laser generates a plurality of sound waves over timed increments in an object. A polarizing interferometer is used to measure surface movement of the object caused by the sound waves and sensed by phase shifts in the signal beam. A photon multiplier senses the phase shift and develops an electrical signal. A signal conditioning arrangement modifies the electrical signals to generate an average signal correlated to the sound waves which in turn is correlated to a physical or metallurgical property of the object, such as temperature, which property may then be used to control the process. External, random vibrations of the workpiece are utilized to develop discernible signals which can be sensed in the interferometer by only one photon multiplier. In addition the interferometer includes an arrangement for optimizing its sensitivity so that movement attributed to various waves can be detected in opaque objects. The interferometer also includes a mechanism for sensing objects with rough surfaces which produce speckle light patterns. Finally the interferometer per se, with the addition of a second photon multiplier is capable of accurately recording beam length distance differences with only one reading.
Polarizing optical interferometer having a dual use optical element
Kotidis, P.A.; Woodroffe, J.A.; Rostler, P.S.
1995-04-04
A system for nondestructively measuring an object and controlling industrial processes in response to the measurement is disclosed in which an impulse laser generates a plurality of sound waves over timed increments in an object. A polarizing interferometer is used to measure surface movement of the object caused by the sound waves and sensed by phase shifts in the signal beam. A photon multiplier senses the phase shift and develops an electrical signal. A signal conditioning arrangement modifies the electrical signals to generate an average signal correlated to the sound waves which in turn is correlated to a physical or metallurgical property of the object, such as temperature, which property may then be used to control the process. External, random vibrations of the workpiece are utilized to develop discernible signals which can be sensed in the interferometer by only one photon multiplier. In addition the interferometer includes an arrangement for optimizing its sensitivity so that movement attributed to various waves can be detected in opaque objects. The interferometer also includes a mechanism for sensing objects with rough surfaces which produce speckle light patterns. Finally the interferometer per se, with the addition of a second photon multiplier is capable of accurately recording beam length distance differences with only one reading. 38 figures.
Balloon Exoplanet Nulling Interferometer (BENI)
NASA Technical Reports Server (NTRS)
Lyon, Richard G.; Clampin, Mark; Woodruff, Robert A.; Vasudevan, Gopal; Ford, Holland; Petro, Larry; Herman, Jay; Rinehart, Stephen; Carpenter, Kenneth; Marzouk, Joe
2009-01-01
We evaluate the feasibility of using a balloon-borne nulling interferometer to detect and characterize exosolar planets and debris disks. The existing instrument consists of a 3-telescope Fizeau imaging interferometer with 3 fast steering mirrors and 3 delay lines operating at 800 Hz for closed-loop control of wavefront errors and fine pointing. A compact visible nulling interferometer is under development which when coupled to the imaging interferometer would in-principle allow deep suppression of starlight. We have conducted atmospheric simulations of the environment above 100,000 feet and believe balloons are a feasible path forward towards detection and characterization of a limited set of exoplanets and their debris disks. Herein we will discuss the BENI instrument, the balloon environment and the feasibility of such as mission.
Goodwin, William L.
1977-01-01
An electronic circuit is provided which may be used to preset a digital display unit of a Zeeman-effect layer interferometer system which derives distance measurements by comparing a reference signal to a Doppler signal generated at the output of the interferometer laser head. The circuit presets dimensional offsets in the interferometer digital display by electronically inducing a variation in either the Doppler signal or the reference signal, depending upon the direction of the offset, to achieve the desired display preset.
Fiber Sagnac interferometer temperature sensor
Starodumov, A.N.; Zenteno, L.A.; Monzon, D.; De La Rosa, E.
1997-01-01
A modified Sagnac interferometer-based fiber temperature sensor is proposed. Polarization independent operation and high temperature sensitivity of this class of sensors make them cost effective instruments for temperature measurements. A comparison of the proposed sensor with Bragg grating and long-period grating fiber sensors is derived. A temperature-induced spectral displacement of 0.99 nm/K is demonstrated for an internal stress birefringent fiber-based Sagnac interferometer. {copyright} {ital 1997 American Institute of Physics.}
Quadrature rules with multiple nodes for evaluating integrals with strong singularities
NASA Astrophysics Data System (ADS)
Milovanovic, Gradimir V.; Spalevic, Miodrag M.
2006-05-01
We present a method based on the Chakalov-Popoviciu quadrature formula of Lobatto type, a rather general case of quadrature with multiple nodes, for approximating integrals defined by Cauchy principal values or by Hadamard finite parts. As a starting point we use the results obtained by L. Gori and E. Santi (cf. On the evaluation of Hilbert transforms by means of a particular class of Turan quadrature rules, Numer. Algorithms 10 (1995), 27-39; Quadrature rules based on s-orthogonal polynomials for evaluating integrals with strong singularities, Oberwolfach Proceedings: Applications and Computation of Orthogonal Polynomials, ISNM 131, Birkhauser, Basel, 1999, pp. 109-119). We generalize their results by using some of our numerical procedures for stable calculation of the quadrature formula with multiple nodes of Gaussian type and proposed methods for estimating the remainder term in such type of quadrature formulae. Numerical examples, illustrations and comparisons are also shown.
ERIC Educational Resources Information Center
Schewe, Phillip F.
1981-01-01
Examines the nature of laser light. Topics include: (1) production and characteristics of laser light; (2) nine types of lasers; (3) five laser techniques including holography; (4) laser spectroscopy; and (5) laser fusion and other applications. (SK)
Michelson Interferometer (MINT)
NASA Technical Reports Server (NTRS)
Lacis, Andrew; Carlson, Barbara
1993-01-01
MINT is a Michelson interferometer designed to measure the thermal emission from the earth at high spectral resolution (2/cm) over a broad spectral range (250-1700/cm, 6-40 mu m) with contiguous 3-pixel wide (12 mrad, 8 km field of view) along-track sampling. MINT is particularly well suited for monitoring cloud properties (cloud cover, effective temperature, optical thickness, ice/water phase, and effective particle size) both day and night, as well as tropospheric water vapor, ozone, and temperature. The key instrument characteristics that make MINT ideally suited for decadal monitoring purposes are: high wavelength to wavelength precision across the full IR spectrum with high spectral resolution; space-proven long-term durability and calibration stability; and small size, low cost, low risk instrument incorporating the latest detector and electronics technology. MINT also incorporates simplicity in design and operation by utilizing passively cooled DTGS detectors and nadir viewing geometry (with target motion compensation). MINT measurement objectives, instrument characteristics, and key advantages are summarized in this paper.
Atmospheric Emitted Radiance Interferometer
Gero, Jonathan; Ermold, Brian; Gaustad, Krista; Koontz, Annette; Hackel, Denny; Garcia, Raymond
2005-01-01
The atmospheric emitted radiance interferometer (AERI) is a ground-based instrument that measures the downwelling infrared radiance from the Earths atmosphere. The observations have broad spectral content and sufficient spectral resolution to discriminate among gaseous emitters (e.g., carbon dioxide and water vapor) and suspended matter (e.g., aerosols, water droplets, and ice crystals). These upward-looking surface observations can be used to obtain vertical profiles of tropospheric temperature and water vapor, as well as measurements of trace gases (e.g., ozone, carbon monoxide, and methane) and downwelling infrared spectral signatures of clouds and aerosols. The AERI is a passive remote sounding instrument, employing a Fourier transform spectrometer operating in the spectral range 3.319.2 μm (5203020 cm-1) at an unapodized resolution of 0.5 cm-1 (max optical path difference of 1 cm). The extended-range AERI (ER-AERI) deployed in dry climates, like in Alaska, have a spectral range of 3.325.0 μm (4003020 cm-1) that allow measurements in the far-infrared region. Typically, the AERI averages views of the sky over a 16-second interval and operates continuously.
Michelson Interferometer (MINT)
NASA Astrophysics Data System (ADS)
Lacis, Andrew; Carlson, Barbara
1993-09-01
MINT is a Michelson interferometer designed to measure the thermal emission from the earth at high spectral resolution (2/cm) over a broad spectral range (250-1700/cm, 6-40 mu m) with contiguous 3-pixel wide (12 mrad, 8 km field of view) along-track sampling. MINT is particularly well suited for monitoring cloud properties (cloud cover, effective temperature, optical thickness, ice/water phase, and effective particle size) both day and night, as well as tropospheric water vapor, ozone, and temperature. The key instrument characteristics that make MINT ideally suited for decadal monitoring purposes are: high wavelength to wavelength precision across the full IR spectrum with high spectral resolution; space-proven long-term durability and calibration stability; and small size, low cost, low risk instrument incorporating the latest detector and electronics technology. MINT also incorporates simplicity in design and operation by utilizing passively cooled DTGS detectors and nadir viewing geometry (with target motion compensation). MINT measurement objectives, instrument characteristics, and key advantages are summarized in this paper.
102({h_bar}/2{pi})k Large Area Atom Interferometers
Chiow, Sheng-wey; Kovachy, Tim; Chien, Hui-Chun; Kasevich, Mark A.
2011-09-23
We demonstrate atom interferometers utilizing a novel beam splitter based on sequential multiphoton Bragg diffractions. With this sequential Bragg large momentum transfer (SB-LMT) beam splitter, we achieve high contrast atom interferometers with momentum splittings of up to 102 photon recoil momenta (102({h_bar}/2{pi})k). To our knowledge, this is the highest momentum splitting achieved in any atom interferometer, advancing the state-of-the-art by an order of magnitude. We also demonstrate strong noise correlation between two simultaneous SB-LMT interferometers, which alleviates the need for ultralow noise lasers and ultrastable inertial environments in some future applications. Our method is intrinsically scalable and can be used to dramatically increase the sensitivity of atom interferometers in a wide range of applications, including inertial sensing, measuring the fine structure constant, and detecting gravitational waves.
A robust sinusoidal signal processing method for interferometers
NASA Astrophysics Data System (ADS)
Wu, Xiang-long; Zhang, Hui; Tseng, Yang-Yu; Fan, Kuang-Chao
2013-10-01
Laser interferometers are widely used as a reference for length measurement. Reliable bidirectional optical fringe counting is normally obtained by using two orthogonally sinusoidal signals derived from the two outputs of an interferometer with path difference. These signals are subject to be disturbed by the geometrical errors of the moving target that causes the separation and shift of two interfering light spots on the detector. It results in typical Heydemann errors, including DC drift, amplitude variation and out-of-orthogonality of two sinusoidal signals that will seriously reduce the accuracy of fringe counting. This paper presents a robust sinusoidal signal processing method to correct the distorted waveforms by hardware. A corresponding circuit board has been designed. A linear stage equipped with a laser displacement interferometer and a height gauge equipped with a linear grating interferometer are used as the test beds. Experimental results show that, even with a seriously disturbed input waveform, the output Lissajous circle can always be stabilized after signal correction. This robust method increases the stability and reliability of the sinusoidal signals for data acquisition device to deal with pulse count and phase subdivision.
Multiple beam Fizeau interferometer with filtered frequency comb illumination
NASA Astrophysics Data System (ADS)
Schwider, J.
2009-08-01
Real wedge interferometers of the Fizeau-type do not allow for fringes in case of a spectral broadband source - or in short: for white light fringes. Here, the use of a suitable frequency comb source will help to overcome this limitation on the one hand and on the other will offer the capability for enhanced phase sensitivity in high precision measurements of surface deviations. Frequency combs can be produced either by using a pulse train from a fs-laser or by passive filtering of the light emitted by a broadband source as a superlum-diode or a fs-laser. The frequency comb produced by a common fs-laser is extremely fine, i.e., the frequency difference of consecutive peaks is very small or the distance of consecutive pulses of the pulse train might be of the order of 1 m. Therefore, the pulse train produced by passive filtering of a broadband source is better adapted to the needs of surface testing interferometers. White light fringes are either applied for the profiling of discontinuous surfaces and/or can serve as an indication for the correct choice of multiplication factors in superposition interferometry. During the last decennium it became more and more clear that spatially incoherent sources provide better measuring accuracy in surface measurements due to the reduced influence of dust diffraction patterns. The advantage of laser illumination can nevertheless be maintained if the laser light is made spatially incoherent through moving scatterers in the light path. Here, we will discuss the application of spatially incoherent broadband light frequency filtered through a Fabry-Perot filter. The main applications are in the following fields: (1) surface profiling applications using two-beam Fizeau interferometers, (2) selection of single cavities out of a series of interlaced cavities, and (3) sensitivity enhancement for multi-beam interferometers for planeness or sphericity measurements. Some of the discussed possibilities will be experimentally demonstrated.
Chen, Hua-Pin
2014-01-01
The electronically tunable quadrature oscillator using a single multiple-output current controlled current differencing transconductance amplifier (MO-CCCDTA) and grounded passive components is presented. The proposed configuration uses a single MO-CCCDTA, two grounded capacitors and one grounded resistor. Two high-output impedance quadrature current signals and two quadrature voltage signals with 90° phase difference. The oscillation condition and oscillation frequency of the proposed quadrature oscillator are independently controllable. The use of only grounded passive components makes the proposed circuit ideal for integrated circuit implementation. PMID:25121124
Parametric generation of quadrature squeezing of mirrors in cavity optomechanics
Liao, Jie-Qiao; Law, C. K.
2011-03-15
We propose a method to generate quadrature-squeezed states of a moving mirror in a Fabry-Perot cavity. This is achieved by exploiting the fact that when the cavity is driven by an external field with a large detuning, the moving mirror behaves as a parametric oscillator. We show that parametric resonance can be reached approximately by modulating the driving field amplitude at a frequency matching the frequency shift of the mirror. The parametric resonance leads to an efficient generation of squeezing, which is limited by the thermal noise of the environment.
Experimental demonstration of microring quadrature phase-shift keying modulators.
Dong, Po; Xie, Chongjin; Chen, Long; Fontaine, Nicolas K; Chen, Young-kai
2012-04-01
Advanced optical modulation formats are a key technology to increase the capacity of optical communication networks. Mach-Zehnder modulators are typically used to generate various modulation formats. Here, we report the first experimental demonstration of quadrature phase-shift keying (QPSK) modulation using compact microring modulators. Generation of 20 Gb/s QPSK signals is demonstrated with 30 μm radius silicon ring modulators with drive voltages of ~6 V. These compact QPSK modulators may be used in miniature optical transponders for high-capacity optical data links. PMID:22466187
Noise-cancelling quadrature magnetic position, speed and direction sensor
Preston, Mark A.; King, Robert D.
1996-01-01
An array of three magnetic sensors in a single package is employed with a single bias magnet for sensing shaft position, speed and direction of a motor in a high magnetic noise environment. Two of the three magnetic sensors are situated in an anti-phase relationship (i.e., 180.degree. out-of-phase) with respect to the relationship between the other of the two sensors and magnetically salient target, and the third magnetic sensor is situated between the anti-phase sensors. The result is quadrature sensing with noise immunity for accurate relative position, speed and direction measurements.
Determining Fault Orientation with Sagnac Interferometers
NASA Astrophysics Data System (ADS)
Gruenwald, Konstantin; Dunn, Robert
2014-03-01
Typically, earthquake fault ruptures emit seismic waves in directions dependent on the fault's orientation. Specifically, as the fault slips to release strain, compressional P-waves propagate parallel and perpendicular to the fault plane, and transverse S-waves propagate at 45 degree angles to the fault-a result of the double-couple model of fault slippage. Sagnac Interferometers (ring-lasers) have been used to study wave components of several natural phenomena. We used the initial responses of a ring-laser from transverse S-waves to determine the orientation of the nearby Guy/Greenbrier fault, the source of an earthquake swarm in 2010-11 purportedly caused by hydraulic fracturing. This orientation was compared to the structure of the fault extracted by nearby seismogram responses. Our goal was to determine if ring-lasers could reinforce or add to the models of fault orientation constructed from seismographs. The results indicate that the ring-laser's responses can aid in constructing fault orientation in a manner similar to traditional seismographs. Funded by the Arkansas Space Grant Consortium and the National Science Foundation.
Phase-lock fiber optic interferometer
NASA Astrophysics Data System (ADS)
Bush, I. J.
1984-12-01
A fiber optic acoustic sensing system for tracking a phase shift linearly over a wide range thereby allowing accurate tracking in the presence of temperature induced phase fluctuation is described. In one embodiment, light from a laser is split and coupled into both legs of a fiber interferometer. One leg is phase modulated by the acoustic signal while the other leg is phase modulated by a first and second piezoelectric cylindrical modulators. The second modulator is driven at omega sub m to effectively shift the acoustic information up in frequency into the sidebands of the carrier omega sub m. The light signals in the two legs are combined, detected, cross-correlated with the carrier omega sub m to produce an error signal, and then low pass filtered. This filtered error signal is fed back to control the first modulator. The first modulator keeps the interferometer locked in phase by effectively cancelling out the phase produced by temperature and acoustic pressure fluctuations. To effect this cancellation, the first modulator must inversely duplicate the phase shift thereby producing the desired output signal.
Phase-Shifting Liquid Crystal Interferometers for Microgravity Fluid Physics
NASA Astrophysics Data System (ADS)
Griffin, DeVon W.; Marshall, Keneth L.
2002-11-01
The initial focus of this project was to eliminate both of these problems in the Liquid Crystal Point-Diffraction Interferometer (LCPDI). Progress toward that goal will be described, along with the demonstration of a phase shifting Liquid Crystal Shearing Interferometer (LCSI) that was developed as part of this work. The latest LCPDI, other than a lens to focus the light from a test section onto a diffracting microsphere within the interferometer and a collimated laser for illumination, the pink region contained within the glass plates on the rod-mounted platform is the complete interferometer. The total width is approximately 1.5 inches with 0.25 inches on each side for bonding the electrical leads. It is 1 inch high and there are only four diffracting microspheres within the interferometer. As a result, it is very easy to align, achieving the first goal. The liquid crystal electro-optical response time is a function of layer thickness, with thinner devices switching faster due to a reduction in long-range viscoelastic forces between the LC molecules. The LCPDI has a liquid crystal layer thickness of 10 microns, which is controlled by plastic or glass microspheres embedded in epoxy 'pads' at the corners of the device. The diffracting spheres are composed of polystyrene/divinyl benzene polymer with an initial diameter of 15 microns. The spheres deform slightly when the interferometer is assembled to conform to the spacing produced by the microsphere-filled epoxy spacer pads. While the speed of this interferometer has not yet been tested, previous LCPDIs fabricated at the Laboratory for Laser Energetics switched at a rate of approximately 3.3 Hz, a factor of 10 slower than desired. We anticipate better performance when the speed of these interferometers is tested since they are approximately three times thinner. Phase shifting in these devices is a function of the AC voltage level applied to the liquid crystal. As the voltage increases, the dye in the liquid crystal
Phase-Shifting Liquid Crystal Interferometers for Microgravity Fluid Physics
NASA Technical Reports Server (NTRS)
Griffin, DeVon W.; Marshall, Keneth L.
2002-01-01
The initial focus of this project was to eliminate both of these problems in the Liquid Crystal Point-Diffraction Interferometer (LCPDI). Progress toward that goal will be described, along with the demonstration of a phase shifting Liquid Crystal Shearing Interferometer (LCSI) that was developed as part of this work. The latest LCPDI, other than a lens to focus the light from a test section onto a diffracting microsphere within the interferometer and a collimated laser for illumination, the pink region contained within the glass plates on the rod-mounted platform is the complete interferometer. The total width is approximately 1.5 inches with 0.25 inches on each side for bonding the electrical leads. It is 1 inch high and there are only four diffracting microspheres within the interferometer. As a result, it is very easy to align, achieving the first goal. The liquid crystal electro-optical response time is a function of layer thickness, with thinner devices switching faster due to a reduction in long-range viscoelastic forces between the LC molecules. The LCPDI has a liquid crystal layer thickness of 10 microns, which is controlled by plastic or glass microspheres embedded in epoxy 'pads' at the corners of the device. The diffracting spheres are composed of polystyrene/divinyl benzene polymer with an initial diameter of 15 microns. The spheres deform slightly when the interferometer is assembled to conform to the spacing produced by the microsphere-filled epoxy spacer pads. While the speed of this interferometer has not yet been tested, previous LCPDIs fabricated at the Laboratory for Laser Energetics switched at a rate of approximately 3.3 Hz, a factor of 10 slower than desired. We anticipate better performance when the speed of these interferometers is tested since they are approximately three times thinner. Phase shifting in these devices is a function of the AC voltage level applied to the liquid crystal. As the voltage increases, the dye in the liquid crystal
Spurious acceleration noise in spaceborne gravitational wave interferometers
NASA Astrophysics Data System (ADS)
Purdue, Patricia; Larson, Shane L.
2007-12-01
An important source of noise in the Laser Interferometer Space Antenna (LISA) is residual acceleration on the proof masses at the heart of the interferometer system. Two proof masses are carried by each sciencecraft in the LISA constellation, oriented along each of two laser links that are maintained between the distant partners in the constellation. Any change in the local mass distribution will create spurious forces on the individual proof masses which will have to be understood as part of the data analysis reduction. This paper considers the general case of accelerations on the individual proof masses in three dimensions for perturbing masses passing by a LISA sciencecraft with arbitrary velocity vectors and impact parameters. Encounters of this kind are impulsive, occurring over short time scales and appearing in the data record as bursts. The formalism is then applied in a few sample cases, including a meteor fly-by and a thruster maneuver.
High-resolution adaptive holographic interferometer for biomedical applications
NASA Astrophysics Data System (ADS)
Dovgalenko, George; Dagdanova, Ayuna
2007-07-01
We realized new adaptive holographic sensor and interferometer, which allows to visualize high-resolution 3D images of diffuse reflected objects in Continue Hologram Registration Regime- CHRR. The coupled laser wave nonlinear theory was applied for optimization of hologram recording in crystals symmetry 23 and optimized experimental set up. Experimentally demonstrated dynamical holographic image sensors on doped 23 symmetry photosensitive crystals, with resolution 7900-lines/mm at 632 nm and 11641 lines/mm at 440 nm for 15 mW CW HeNe and He-Cd lasers. The results are presented for holographic visualization of Cryogenic and Ultrasonic near field images of Surgical Medical Instrument. Application of CHRR interferometer for hologram registration of moving biological object in "vivo" is illustrated.
Quantum noise in differential-type gravitational-wave interferometer and signal recycling
NASA Astrophysics Data System (ADS)
Nishizawa, A.; Kawamura, S.; Sakagami, Masa-aki
2008-07-01
In the sensitivity of laser interferometer gravitational-wave detectors, there exists the standard quantum limit (SQL), derived from Heisenberg's uncertainty relation. The SQL can be overcome using the quantum correlation between shot noise and radiation-pressure noise. One of the methods to overcome SQL, signal recycling, is considered so far only in a recombined-type interferometer such as Advanced-LIGO, LCGT, and GEO600. In this paper, we investigated quantum noise and signal recycling in a differential-type interferometer. We also applied it to a real detector and compared the sensivity with a recombined type.
Vibration diagnosis and remediation design for an x-ray optics stitching interferometer system.
Preissner, C.; Assoufid, L.; Shu, D.; Experimental Facilities Division
2004-01-01
The Advanced Photon Source (APS) x-ray optics Metrology Laboratory currently operates a small-aperture Wyko laser interferometer in a stitching configuration. While the stitching configuration allows for easier surface characterization of long x-ray substrates and mirrors, the addition of mechanical components for optic element translation can compromise the ultimate measurement performance of the interferometer. A program of experimental vibration measurements, quantifying the laboratory vibration environment and identifying interferometer support-system behavior, has been conducted. Insight gained from the ambient vibration assessment and modal analysis has guided the development of a remediation technique. Discussion of the problem diagnosis and possible solutions are presented in this paper.
NASA Astrophysics Data System (ADS)
Kimble, H. J.; Levin, Yuri; Matsko, Andrey B.; Thorne, Kip S.; Vyatchanin, Sergey P.
2002-01-01
The LIGO-II gravitational-wave interferometers (ca. 2006-2008) are designed to have sensitivities near the standard quantum limit (SQL) in the vicinity of 100 Hz. This paper describes and analyzes possible designs for subsequent LIGO-III interferometers that can beat the SQL. These designs are identical to a conventional broad band interferometer (without signal recycling), except for new input and/or output optics. Three designs are analyzed: (i) a squeezed-input interferometer (conceived by Unruh based on earlier work of Caves) in which squeezed vacuum with frequency-dependent (FD) squeeze angle is injected into the interferometer's dark port; (ii) a variational-output interferometer (conceived in a different form by Vyatchanin, Matsko and Zubova), in which homodyne detection with FD homodyne phase is performed on the output light; and (iii) a squeezed-variational interferometer with squeezed input and FD-homodyne output. It is shown that the FD squeezed-input light can be produced by sending ordinary squeezed light through two successive Fabry-Pérot filter cavities before injection into the interferometer, and FD-homodyne detection can be achieved by sending the output light through two filter cavities before ordinary homodyne detection. With anticipated technology (power squeeze factor e-2R=0.1 for input squeezed vacuum and net fractional loss of signal power in arm cavities and output optical train ɛ*=0.01) and using an input laser power Io in units of that required to reach the SQL (the planned LIGO-II power, ISQL), the three types of interferometer could beat the amplitude SQL at 100 Hz by the following amounts μ≡(Sh)/(SSQLh) and with the following corresponding increase V=1/μ3 in the volume of the universe that can be searched for a given noncosmological source: Squeezed input-μ~=(e-2R)~=0.3 and V~=1/0.33~=30 using Io/ISQL=1. Variational-output-μ~=ɛ1/4*~=0.3 and V~=30 but only if the optics can handle a ten times larger power: Io/ISQL~=1/(ɛ*)=10
Mach-Zehnder interferometer for movement monitoring
NASA Astrophysics Data System (ADS)
Vasinek, Vladimir; Cubik, Jakub; Kepak, Stanislav; Doricak, Jan; Latal, Jan; Koudelka, Petr
2012-06-01
Fiber optical interferometers belong to highly sensitive equipments that are able to measure slight changes like distortion of shape, temperature and electric field variation and etc. Their great advantage is that they are insensitive on ageing component, from which they are composed of. It is in virtue of herewith, that there are evaluated no changes in optical signal intensity but number interference fringes. To monitor the movement of persons, eventually to analyze the changes in state of motion we developed method based on analysis the dynamic changes in interferometric pattern. We have used Mach- Zehnder interferometer with conventional SM fibers excited with the DFB laser at wavelength of 1550 nm. It was terminated with optical receiver containing InGaAs PIN photodiode. Its output was brought into measuring card module that performs on FFT of the received interferometer signal. The signal rises with the composition of two waves passing through single interferometer arm. The optical fiber SMF 28e in one arm is referential; the second one is positioned on measuring slab at dimensions of 1x2m. A movement of persons around the slab was monitored, signal processed with FFT and frequency spectra were evaluated. They rose owing to dynamic changes of interferometric pattern. The results reflect that the individual subjects passing through slab embody characteristic frequency spectra, which are individual for particular persons. The scope of measuring frequencies proceeded from zero to 10 kHz. It was also displayed in experiments that the experimental subjects, who walked around the slab and at the same time they have had changed their state of motion (knee joint fixation), embodied characteristic changes in their frequency spectra. At experiments the stability of interferometric patterns was evaluated as from time aspects, so from the view of repeated identical experiments. Two kinds of balls (tennis and ping-pong) were used to plot the repeatability measurements and
Basnak, D V; Bikmukhametov, K A; Dmitrieva, N I; Dmitriev, Aleksandr K; Lugovoi, A A; Pokasov, P V; Chepurov, S V
2010-10-15
A method for measuring the carrier envelope offset (CEO) frequency of the femtosecond frequency comb with a bandwidth of less than one octave by using a Fabry-Perot interferometer is proposed and experimentally demonstrated. (laser components)
A new method for determining the plasma electron density using three-color interferometer
Arakawa, Hiroyuki; Kawano, Yasunori; Itami, Kiyoshi
2012-06-15
A new method for determining the plasma electron density using the fractional fringes on three-color interferometer is proposed. Integrated phase shift on each interferometer is derived without using the temporal history of the fractional fringes. The dependence on the fringe resolution and the electrical noise are simulated on the wavelengths of CO{sub 2} laser. Short-time integrations of the fractional fringes enhance the reliability of this method.
Automatic control system design of laser interferometer
NASA Astrophysics Data System (ADS)
Lu, Qingjie; Li, Chunjie; Sun, Hao; Ren, Shaohua; Han, Sen
2015-10-01
There are a lot of shortcomings with traditional optical adjustment in interferometry, such as low accuracy, time-consuming, labor-intensive, uncontrollability, and bad repetitiveness, so we treat the problem by using wireless remote control system. Comparing to the traditional method, the effect of vibration and air turbulence will be avoided. In addition the system has some peculiarities of low cost, high reliability and easy operation etc. Furthermore, the switching between two charge coupled devices (CCDs) can be easily achieved with this wireless remote control system, which is used to collect different images. The wireless transmission is achieved by using Radio Frequency (RF) module and programming the controller, pulse width modulation (PWM) of direct current (DC) motor, real-time switching of relay and high-accuracy displacement control of FAULHABER motor are available. The results of verification test show that the control system has good stability with less than 5% packet loss rate, high control accuracy and millisecond response speed.
A molecular interferometer using YbF
NASA Astrophysics Data System (ADS)
Hinds, E. A.; Redgrave, G. D.; Cahn, S. B.; Sauer, B. E.
1998-05-01
We have demonstrated a molecular interferometer for electron spin using the paramagnetic molecule YbF. In a molecular beam of YbF, a coherent superposition of hyperfine states |F=1, m_F=± 1> is produced, either via an rf transition or adiabatic Raman passage through suitably detuned laser beams. This state evolves in a weak magnetic field B, is recombined and probed with fluorescence detection. For a time of flight T the signal intensity is proportional to \\cos ^2 (μ_BBT/hbar). For our apparatus the fringe spacing is on the order of 100μ G. We will discuss the sensitivity in the context of an experiment to measure time reversal violation in YbF, as has been described in Physics World, April 1997.
A Robust Ramsey Interferometer for Atomic Timekeeping in Dynamic Environments
NASA Astrophysics Data System (ADS)
Kotru, Krish; Brown, Justin; Butts, David; Choy, Jennifer; Galfond, Marissa; Johnson, David M.; Kinast, Joseph; Timmons, Brian; Stoner, Richard
2014-05-01
We present a laser-based approach to atomic timekeeping, in which atomic phase information is extracted using modified Raman pulses in a Ramsey sequence. We overcome systematic effects associated with differential AC Stark shifts and variations in laser beam intensity by employing atom optics derived from Raman adiabatic rapid passage (ARP). This technique drives coherent transfer between two hyperfine ground states by sweeping the frequency difference of two optical fields and maintaining a large single-photon detuning. Compared to a Raman-pulse Ramsey interferometer, we show a >150x reduction in sensitivity to differential AC Stark shifts. We also demonstrate that ARP preserves fringe contrast in Ramsey interferometers for cloud displacements reaching the 1/e2 intensity radius of the laser beam. Deviations of the phase in response to changes in duration, rate, and range of the ARP frequency sweep are bounded to <7 mrad, implying a per-shot fractional frequency uncertainty of 1e-11 for an interrogation time of 10 ms. These characteristics are expected to improve the robustness of clock interferometers operating in dynamic environments. Copyright ©2014 by The Charles Stark Draper Laboratory, Inc. All rights reserved.
Sun, Changsen; Zhao, Yang; Tennant, Adam; Ansari, Farhad
2003-08-01
A Vernier principle is employed to improve the spatial resolution of a fiber-optic white-light interferometer to the accuracy of 0.2 microm. The Vernier principle is implemented by combination of interference fringes itself and a virtual fringe that is generated by means of software tracing the scanning mirror. Two rulers are read with respect to each other. This design is insensitive to intensity fluctuation of the interference fringe. The applications, submicrometer estimation for the quadrature-locking selection and the tolerance of the relative measurement, demonstrate its effectiveness. PMID:12916606
The Nature of the Nodes, Weights and Degree of Precision in Gaussian Quadrature Rules
ERIC Educational Resources Information Center
Prentice, J. S. C.
2011-01-01
We present a comprehensive proof of the theorem that relates the weights and nodes of a Gaussian quadrature rule to its degree of precision. This level of detail is often absent in modern texts on numerical analysis. We show that the degree of precision is maximal, and that the approximation error in Gaussian quadrature is minimal, in a…
Regenerative Fourier transformation for dual-quadrature regeneration of multilevel rectangular QAM.
Sorokina, Mariia; Sygletos, Stylianos; Ellis, Andrew; Turitsyn, Sergei
2015-07-01
We propose a new nonlinear optical loop mirror based configuration capable of regenerating regular rectangular quadrature amplitude modulated (QAM) signals. The scheme achieves suppression of noise distortion on both signal quadratures through the realization of two orthogonal regenerative Fourier transformations. Numerical simulations show the performance of the scheme for high constellation complexities (including 256-QAM formats). PMID:26125381
A Family of Exponential Fitting Direct Quadrature Methods for Volterra Integral Equations
NASA Astrophysics Data System (ADS)
Cardone, A.; Ferro, M.; Ixaru, L. Gr.; Paternoster, B.
2010-09-01
A new class of direct quadrature methods for the solution of Volterra Integral Equations with periodic solution is illustrated. Such methods are based on an exponential fitting gaussian quadrature formula, whose coefficients depend on the problem parameters, in order to better reproduce the behavior the analytical solution. The construction of the methods is described, together with the analysis of the order of accuracy.
General n-dimensional quadrature transform and its application to interferogram demodulation.
Servin, Manuel; Quiroga, Juan Antonio; Marroquin, Jose Luis
2003-05-01
Quadrature operators are useful for obtaining the modulating phase phi in interferometry and temporal signals in electrical communications. In carrier-frequency interferometry and electrical communications, one uses the Hilbert transform to obtain the quadrature of the signal. In these cases the Hilbert transform gives the desired quadrature because the modulating phase is monotonically increasing. We propose an n-dimensional quadrature operator that transforms cos(phi) into -sin(phi) regardless of the frequency spectrum of the signal. With the quadrature of the phase-modulated signal, one can easily calculate the value of phi over all the domain of interest. Our quadrature operator is composed of two n-dimensional vector fields: One is related to the gradient of the image normalized with respect to local frequency magnitude, and the other is related to the sign of the local frequency of the signal. The inner product of these two vector fields gives us the desired quadrature signal. This quadrature operator is derived in the image space by use of differential vector calculus and in the frequency domain by use of a n-dimensional generalization of the Hilbert transform. A robust numerical algorithm is given to find the modulating phase of two-dimensional single-image closed-fringe interferograms by use of the ideas put forward. PMID:12747439
The NIST Length Scale Interferometer
Beers, John S.; Penzes, William B.
1999-01-01
The National Institute of Standards and Technology (NIST) interferometer for measuring graduated length scales has been in use since 1965. It was developed in response to the redefinition of the meter in 1960 from the prototype platinum-iridium bar to the wavelength of light. The history of the interferometer is recalled, and its design and operation described. A continuous program of modernization by making physical modifications, measurement procedure changes and computational revisions is described, and the effects of these changes are evaluated. Results of a long-term measurement assurance program, the primary control on the measurement process, are presented, and improvements in measurement uncertainty are documented.
Interferometer mirror tilt correcting system
NASA Technical Reports Server (NTRS)
Schindler, R. A. (Inventor)
1977-01-01
An interferometer is described, having several means for automatically adjusting the angular tilt of a reflecting surface in one of two paths to maintain the exit beams from the two paths parallel to each other. Three detectors at the output of the interferometer were disposed on mutually perpendicular axes which define a plane normal to the nominal exit beam axis. One detector at the origin of the axes was used as a reference for separate phase difference comparison with the outputs of the other two detectors on the X and Y axes to develop servo error signals.
Optical interferometer in space
NASA Technical Reports Server (NTRS)
Bender, Peter L.; Faller, J. E.; Hall, J. L.; Hils, D.; Stebbins, R. T.; Vincent, Mark A.
1989-01-01
The present design concepts for a Laser Gravitational Wave Observatory in Space are described. Laser heterodyne distance measurements are made between test masses located in three spacecraft separated by roughly 10(exp 6) km. The major technology issues are: the reduction of spurious acceleration noise for the test masses to below 2 x 10(exp -15) cm/sq sec/Hz(0.5) from 10(exp -5) to 10(exp -3) Hz; and the measurement of changes in the difference of the antenna arm lengths to 5 x 10(exp -11) cm/Hz(0.5) from 10(exp -3) to 1 Hz with high reliability. The science objectives are: to measure discrete sinusoidal gravitational wave signals from individual sources with periods of 1 second to 1 day; to measure the stochastic background due to unresolved binaries; and to search for gravitational wave pulses with periods longer than 1 sec from possible exotic sources such as gravitational collapse of very massive objects.
Microwave interferometer controls cutting depth of plastics
NASA Technical Reports Server (NTRS)
Heisman, R. M.; Iceland, W. F.
1969-01-01
Microwave interferometer system controls the cutting of plastic materials to a prescribed depth. The interferometer is mounted on a carriage with a spindle and cutting tool. A cross slide, mounted on the carriage, allows the interferometer and cutter to move toward or away from the plastic workpiece.
Automatic Alignment of Displacement-Measuring Interferometer
NASA Technical Reports Server (NTRS)
Halverson, Peter; Regehr, Martin; Spero, Robert; Alvarez-Salazar, Oscar; Loya, Frank; Logan, Jennifer
2006-01-01
A control system strives to maintain the correct alignment of a laser beam in an interferometer dedicated to measuring the displacement or distance between two fiducial corner-cube reflectors. The correct alignment of the laser beam is parallel to the line between the corner points of the corner-cube reflectors: Any deviation from parallelism changes the length of the optical path between the reflectors, thereby introducing a displacement or distance measurement error. On the basis of the geometrical optics of corner-cube reflectors, the length of the optical path can be shown to be L = L(sub 0)cos theta, where L(sub 0) is the distance between the corner points and theta is the misalignment angle. Therefore, the measurement error is given by DeltaL = L(sub 0)(cos theta - 1). In the usual case in which the misalignment is small, this error can be approximated as DeltaL approximately equal to -L(sub 0)theta sup 2/2. The control system (see figure) is implemented partly in hardware and partly in software. The control system includes three piezoelectric actuators for rapid, fine adjustment of the direction of the laser beam. The voltages applied to the piezoelectric actuators include components designed to scan the beam in a circular pattern so that the beam traces out a narrow cone (60 microradians wide in the initial application) about the direction in which it is nominally aimed. This scan is performed at a frequency (2.5 Hz in the initial application) well below the resonance frequency of any vibration of the interferometer. The laser beam makes a round trip to both corner-cube reflectors and then interferes with the launched beam. The interference is detected on a photodiode. The length of the optical path is measured by a heterodyne technique: A 100- kHz frequency shift between the launched beam and a reference beam imposes, on the detected signal, an interferometric phase shift proportional to the length of the optical path. A phase meter comprising analog
Weighted discrete least-squares polynomial approximation using randomized quadratures
NASA Astrophysics Data System (ADS)
Zhou, Tao; Narayan, Akil; Xiu, Dongbin
2015-10-01
We discuss the problem of polynomial approximation of multivariate functions using discrete least squares collocation. The problem stems from uncertainty quantification (UQ), where the independent variables of the functions are random variables with specified probability measure. We propose to construct the least squares approximation on points randomly and uniformly sampled from tensor product Gaussian quadrature points. We analyze the stability properties of this method and prove that the method is asymptotically stable, provided that the number of points scales linearly (up to a logarithmic factor) with the cardinality of the polynomial space. Specific results in both bounded and unbounded domains are obtained, along with a convergence result for Chebyshev measure. Numerical examples are provided to verify the theoretical results.
Double-Referential Holography and Spatial Quadrature Amplitude Modulation
NASA Astrophysics Data System (ADS)
Zukeran, Keisuke; Okamoto, Atsushi; Takabayashi, Masanori; Shibukawa, Atsushi; Sato, Kunihiro; Tomita, Akihisa
2013-09-01
We proposed a double-referential holography (DRH) that allows phase-detection without external additional beams. In the DRH, phantom beams, prepared in the same optical path as signal beams and preliminary multiplexed in a recording medium along with the signal, are used to produce interference fringes on an imager for converting a phase into an intensity distribution. The DRH enables stable and high-accuracy phase detection independent of the fluctuations and vibrations of the optical system owing to medium shift and temperature variation. Besides, the collinear arrangement of the signal and phantom beams leads to the compactness of the optical data storage system. We conducted an experiment using binary phase modulation signals for verifying the DRH operation. In addition, 38-level spatial quadrature amplitude modulation signals were successfully reproduced with the DRH by numerical simulation. Furthermore, we verified that the distributed phase-shifting method moderates the dynamic range consumption for the exposure of phantom beams.
Terahertz single-shot quadrature phase-shifting interferometry.
Földesy, Péter
2012-10-01
A single-shot quadrature phase-shifting interferometry architecture is presented that is applicable to antenna coupled detector technologies. The method is based on orthogonally polarized object and reference beams and on linear and circular polarization sensitive antennas in space-division multiplexing. The technique can be adapted to two-, three-, and four-step and Gabor holography recordings. It is also demonstrated that the space-division multiplexing does not necessarily cause sparse sampling. A sub-THz detector array is presented containing multiple on-chip antennas and FET plasma wave detectors implemented in a 90 nm complementary metal-oxide semiconductor technology. As an example, two-step phase-shifting reconstruction results are given at 360 GHz. PMID:23027273
Quadrature squeezed photons from a two-level system.
Schulte, Carsten H H; Hansom, Jack; Jones, Alex E; Matthiesen, Clemens; Le Gall, Claire; Atatüre, Mete
2015-09-10
Resonance fluorescence arises from the interaction of an optical field with a two-level system, and has played a fundamental role in the development of quantum optics and its applications. Despite its conceptual simplicity, it entails a wide range of intriguing phenomena, such as the Mollow-triplet emission spectrum, photon antibunching and coherent photon emission. One fundamental aspect of resonance fluorescence--squeezing in the form of reduced quantum fluctuations in the single photon stream from an atom in free space--was predicted more than 30 years ago. However, the requirement to operate in the weak excitation regime, together with the combination of modest oscillator strength of atoms and low collection efficiencies, has continued to necessitate stringent experimental conditions for the observation of squeezing with atoms. Attempts to circumvent these issues had to sacrifice antibunching, owing to either stimulated forward scattering from atomic ensembles or multi-photon transitions inside optical cavities. Here, we use an artificial atom with a large optical dipole enabling 100-fold improvement of the photon detection rate over the natural atom counterpart and reach the necessary conditions for the observation of quadrature squeezing in single resonance-fluorescence photons. By implementing phase-dependent homodyne intensity-correlation detection, we demonstrate that the electric field quadrature variance of resonance fluorescence is three per cent below the fundamental limit set by vacuum fluctuations, while the photon statistics remain antibunched. The presence of squeezing and antibunching simultaneously is a fully non-classical outcome of the wave-particle duality of photons. PMID:26322581
Residual Distribution Schemes for Conservation Laws Via Adaptive Quadrature
NASA Technical Reports Server (NTRS)
Barth, Timothy; Abgrall, Remi; Biegel, Bryan (Technical Monitor)
2000-01-01
This paper considers a family of nonconservative numerical discretizations for conservation laws which retains the correct weak solution behavior in the limit of mesh refinement whenever sufficient order numerical quadrature is used. Our analysis of 2-D discretizations in nonconservative form follows the 1-D analysis of Hou and Le Floch. For a specific family of nonconservative discretizations, it is shown under mild assumptions that the error arising from non-conservation is strictly smaller than the discretization error in the scheme. In the limit of mesh refinement under the same assumptions, solutions are shown to satisfy an entropy inequality. Using results from this analysis, a variant of the "N" (Narrow) residual distribution scheme of van der Weide and Deconinck is developed for first-order systems of conservation laws. The modified form of the N-scheme supplants the usual exact single-state mean-value linearization of flux divergence, typically used for the Euler equations of gasdynamics, by an equivalent integral form on simplex interiors. This integral form is then numerically approximated using an adaptive quadrature procedure. This renders the scheme nonconservative in the sense described earlier so that correct weak solutions are still obtained in the limit of mesh refinement. Consequently, we then show that the modified form of the N-scheme can be easily applied to general (non-simplicial) element shapes and general systems of first-order conservation laws equipped with an entropy inequality where exact mean-value linearization of the flux divergence is not readily obtained, e.g. magnetohydrodynamics, the Euler equations with certain forms of chemistry, etc. Numerical examples of subsonic, transonic and supersonic flows containing discontinuities together with multi-level mesh refinement are provided to verify the analysis.
Holographic Twyman-Green interferometer
NASA Technical Reports Server (NTRS)
Chen, C. W.; Breckinridge, J. B.
1982-01-01
A dichromated gelatin off-axis Fresnel zone plate was designed, fabricated, and used in a new type of interferometer for optical metrology. This single hologram optical element combines the functions of a beam splitter, beam diverger, and aberrated null lens. Data presented show the successful application for an interferometric test of an f/6, 200-mm diam parabolic mirror.
Stitching interferometer for large optics: recent developments of a system
NASA Astrophysics Data System (ADS)
Bray, Michael
1999-07-01
We have developed a stitching interferometer for use with large optics, such as those used in Laser MegaJoule and NIF. The idea behind this technique is that, to keep resolution at its highest value, a 'small' aperture phase-shifting interferometer must be used. But this means that measurement of the complete component has to be performed by stitching the individual sub-apertures together. We have overcome most of the obstacles involved in stitching interferometry, and have been using one such system at our optical and laser production facility in Orleans for some years. We have previously reported transmission measurements performed on a Phebus laser slab. We present here measurements performed on Laser MegaJoule transport mirrors measured in a horizontal position, and on glass inhomogeneity without oil plates. This latter requires multiple takes on each sub-aperture, and additional computing, but the whole process has been automated. In addition, we contribute to the pretty picture department by showing graphical results perhaps never previously seen. This work was performed under contract from CEA-LV, as part of the Laser Megajoule development.
Jain, N.
1999-03-01
Phase-shifting interferometry has many advantages, and the phase shifting nature of the Liquid Crystal Point Diffraction Interferometer (LCPDI) promises to provide significant improvement over other current OMEGA wavefront sensors. However, while phase-shifting capabilities improve its accuracy as an interferometer, phase-shifting itself introduces errors. Phase-shifting algorithms are designed to eliminate certain types of phase-shift errors, and it is important to chose an algorithm that is best suited for use with the LCPDI. Using polarization microscopy, the authors have observed a correlation between LC alignment around the microsphere and fringe behavior. After designing a procedure to compare phase-shifting algorithms, they were able to predict the accuracy of two particular algorithms through computer modeling of device-specific phase shift-errors.
Design of a dual species atom interferometer for space
NASA Astrophysics Data System (ADS)
Schuldt, Thilo; Schubert, Christian; Krutzik, Markus; Bote, Lluis Gesa; Gaaloul, Naceur; Hartwig, Jonas; Ahlers, Holger; Herr, Waldemar; Posso-Trujillo, Katerine; Rudolph, Jan; Seidel, Stephan; Wendrich, Thijs; Ertmer, Wolfgang; Herrmann, Sven; Kubelka-Lange, André; Milke, Alexander; Rievers, Benny; Rocco, Emanuele; Hinton, Andrew; Bongs, Kai; Oswald, Markus; Franz, Matthias; Hauth, Matthias; Peters, Achim; Bawamia, Ahmad; Wicht, Andreas; Battelier, Baptiste; Bertoldi, Andrea; Bouyer, Philippe; Landragin, Arnaud; Massonnet, Didier; Lévèque, Thomas; Wenzlawski, Andre; Hellmig, Ortwin; Windpassinger, Patrick; Sengstock, Klaus; von Klitzing, Wolf; Chaloner, Chris; Summers, David; Ireland, Philip; Mateos, Ignacio; Sopuerta, Carlos F.; Sorrentino, Fiodor; Tino, Guglielmo M.; Williams, Michael; Trenkel, Christian; Gerardi, Domenico; Chwalla, Michael; Burkhardt, Johannes; Johann, Ulrich; Heske, Astrid; Wille, Eric; Gehler, Martin; Cacciapuoti, Luigi; Gürlebeck, Norman; Braxmaier, Claus; Rasel, Ernst
2015-06-01
Atom interferometers have a multitude of proposed applications in space including precise measurements of the Earth's gravitational field, in navigation & ranging, and in fundamental physics such as tests of the weak equivalence principle (WEP) and gravitational wave detection. While atom interferometers are realized routinely in ground-based laboratories, current efforts aim at the development of a space compatible design optimized with respect to dimensions, weight, power consumption, mechanical robustness and radiation hardness. In this paper, we present a design of a high-sensitivity differential dual species 85Rb/87Rb atom interferometer for space, including physics package, laser system, electronics and software. The physics package comprises the atom source consisting of dispensers and a 2D magneto-optical trap (MOT), the science chamber with a 3D-MOT, a magnetic trap based on an atom chip and an optical dipole trap (ODT) used for Bose-Einstein condensate (BEC) creation and interferometry, the detection unit, the vacuum system for 10-11 mbar ultra-high vacuum generation, and the high-suppression factor magnetic shielding as well as the thermal control system. The laser system is based on a hybrid approach using fiber-based telecom components and high-power laser diode technology and includes all laser sources for 2D-MOT, 3D-MOT, ODT, interferometry and detection. Manipulation and switching of the laser beams is carried out on an optical bench using Zerodur bonding technology. The instrument consists of 9 units with an overall mass of 221 kg, an average power consumption of 608 W (814 W peak), and a volume of 470 liters which would well fit on a satellite to be launched with a Soyuz rocket, as system studies have shown.
Digital services using quadrature amplitude modulation (QAM) over CATV analog DWDM system
NASA Astrophysics Data System (ADS)
Yeh, JengRong; Selker, Mark D.; Trail, J.; Piehler, David; Levi, Israel
2000-04-01
Dense Wavelength Division Multiplexing (DWDM) has recently gained great popularity as it provides a cost effective way to increase the transmission capacity of the existing fiber cable plant. For a long time, Dense WDM was exclusively used for baseband digital applications, predominantly in terrestrial long haul networks and in some cases in metropolitan and enterprise networks. Recently, the performance of DWDM components and frequency-stabilized lasers has substantially improved while the costs have down significantly. This makes a variety of new optical network architectures economically viable. The first commercial 8- wavelength DWDM system designed for Hybrid Fiber Coax networks was reported in 1998. This type of DWDM system utilizes Sub-Carrier Multiplexing (SCM) of Quadrature Amplitude Modulated (QAM) signals to transport IP data digital video broadcast and Video on Demand on ITU grid lightwave carriers. The ability of DWDM to provide scalable transmission capacity in the optical layer with SCM granularity is now considered by many to be the most promising technology for future transport and distribution of broadband multimedia services.
Non-destructive testing using two-component/two-wave mixing interferometer
Wartelle, A.; Pouet, B.; Breugnot, S.
2011-06-23
A new laser-based ultrasonic (LBU) receiver was recently introduced. The novel architecture is based on two-wave mixing in photorefractive materials and allows simultaneous measurement of in-plane and out-of-plane displacements (two- component). By taking advantage of recent developments in electronic processing and our knowledge on multi-channel interferometer, we achieved a compact optical system using only a single collecting aperture and a single laser probe beam. We will characterize the system performances and present experimental results demonstrating the capability for this compact two-component interferometer.
V-groove all-fiber core-cladding intermodal interferometer for high-temperature sensing.
Yin, Zhen; Geng, Youfu; Li, Xuejin; Tan, Xiaoling; Gao, Rong
2015-01-10
Novel V-groove all-fiber core-cladding intermodal interferometers fabricated by CO2 laser irradiation on a standard single-mode fiber are described. The high-order cladding modes are excited due to the special V-groove structure. The interferometers are classified as Mach-Zehnder and Michelson type based on the way they are structured. Benefiting from the large difference of thermal coefficients of the core and high-order cladding modes, both types receive high temperature sensitivity by monitoring the wavelength shift of the interference spectrum, and their responses to temperature are similar. Compared with the Mach-Zehnder interferometer, the Michelson interferometer is more compact and more flexible in application. PMID:25967632
NASA Astrophysics Data System (ADS)
Holler, Mirko; Raabe, Jörg
2015-05-01
The nonaxial interferometric position measurement of rotating objects can be performed by imaging the laser beam of the interferometer to a rotating mirror which can be a sphere or a cylinder. This, however, requires such rotating mirrors to be centered on the axis of rotation as a wobble would result in loss of the interference signal. We present a tracking-type interferometer that performs such measurement in a general case where the rotating mirror may wobble on the axis of rotation, or even where the axis of rotation may be translating in space. Aside from tracking, meaning to measure and follow the position of the rotating mirror, the interferometric measurement errors induced by the tracking motion of the interferometer itself are optically compensated, preserving nanometric measurement accuracy. As an example, we show the application of this interferometer in a scanning x-ray tomography instrument.
Multidimensional Hermite-Gaussian quadrature formulae and their application to nonlinear estimation
NASA Technical Reports Server (NTRS)
Mcreynolds, S. R.
1975-01-01
A simplified technique is proposed for calculating multidimensional Hermite-Gaussian quadratures that involves taking the square root of a matrix by the Cholesky algorithm rather than computation of the eigenvectors of the matrix. Ways of reducing the dimension, number, and order of the quadratures are set forth. If the function f(x) under the integral sign is not well approximated by a low-order algebraic expression, the order of the quadrature may be reduced by factoring f(x) into an expression that is nearly algebraic and one that is Gaussian.
NASA Technical Reports Server (NTRS)
Desmarais, R. N.
1975-01-01
Computer programs for computing Gaussian quadrature abscissas and weights are described. For the classical case the programs use Laguerre iteration to compute abscissas as zeros of orthogonal polynomials. The polynomials are evaluated from known recursion coefficients. The nonclassical case is handled similarly except that the recursion coefficients are computed by numerical integration. A sample problem, with input and output, is presented to illustrate the use of the programs. It computes the quadrature abscissas and weights associated with the weight function over the interval (0,1) for quadrature orders from 16 to 96 in increments of 8.
NASA Astrophysics Data System (ADS)
Knizhnerman, Leonid
2010-01-01
Stability of passing from Gaussian quadrature data to the Lanczos recurrence coefficients is considered. Special attention is paid to estimates explicitly expressed in terms of quadrature data and not having weights in denominators. It has been shown that the recent approach, exploiting integral representation of Hankel determinants, implies quantitative improvement of D. Laurie's constructive estimate. It has also been demonstrated that a particular implementation on the Hankel determinant approach gives an estimate being unimprovable up to a coefficient; the corresponding example involves quadrature data with a small but not too small weight. It follows that polynomial increase of a general case upper bound in terms of the dimension is unavoidable.
NASA Astrophysics Data System (ADS)
Bauman, Brian J.; Xiao, Hong
2010-08-01
Forbes introduced the usage of Gaussian quadratures in optical design for circular pupils and fields, and for a specific visible wavelength band. In this paper, Gaussian quadrature methods of selecting rays in ray-tracing are derived for noncircular pupil shapes, such as obscured and vignetted apertures. In addition, these methods are generalized for square fields, and for integrating performance over arbitrary wavelength bands. Integration over wavelength is aided by the use of a novel chromatic coordinate. These quadratures achieve low calculations with fewer rays (by orders of magnitude) than uniform sampling schemes.
A Fiber Interferometer for the Magnetized Shock Experiment
Yoo, Christian
2012-08-30
The Magnetized Shock Experiment (MSX) at Los Alamos National Laboratory requires remote diagnostics of plasma density. Laser interferometry can be used to determine the line-integrated density of the plasma. A multi-chord heterodyne fiber optic Mach-Zehnder interferometer is being assembled and integrated into the experiment. The advantage of the fiber coupling is that many different view chords can be easily obtained by simply moving transmit and receive fiber couplers. Several such fiber sets will be implemented to provide a time history of line-averaged density for several chords at once. The multiple chord data can then be Abel inverted to provide radially resolved spatial profiles of density. We describe the design and execution of this multiple fiber interferometer.
Matter-wave soliton interferometer based on a nonlinear splitter
NASA Astrophysics Data System (ADS)
Sakaguchi, Hidetsugu; Malomed, Boris A.
2016-02-01
We elaborate a model of the interferometer which, unlike previously studied ones, uses a local (δ-functional) nonlinear repulsive potential, embedded into a harmonic-oscillator trapping potential, as the splitter for the incident soliton. An estimate demonstrates that this setting may be implemented by means of the localized Feshbach resonance controlled by a focused laser beam. The same system may be realized as a nonlinear waveguide in optics. Subsequent analysis produces an exact solution for scattering of a plane wave in the linear medium on the δ -functional nonlinear repulsive potential, and an approximate solution for splitting of the incident soliton when the ambient medium is nonlinear. The most essential result, obtained by means of systematic simulations, is that the use of the nonlinear splitter provides the sensitivity of the soliton-based interferometer to the target, inserted into one of its arms, which is much higher than the sensitivity provided by the usual linear splitter.
Astrophysical Adaptation of Points, the Precision Optical Interferometer in Space
NASA Technical Reports Server (NTRS)
Reasenberg, Robert D.; Babcock, Robert W.; Murison, Marc A.; Noecker, M. Charles; Phillips, James D.; Schumaker, Bonny L.; Ulvestad, James S.; McKinley, William; Zielinski, Robert J.; Lillie, Charles F.
1996-01-01
POINTS (Precision Optical INTerferometer in Space) would perform microarcsecond optical astrometric measurements from space, yielding submicroarcsecond astrometric results from the mission. It comprises a pair of independent Michelson stellar interferometers and a laser metrology system that measures both the critical starlight paths and the angle between the baselines. The instrument has two baselines of 2 m, each with two subapertures of 35 cm; by articulating the angle between the baselines, it observes targets separated by 87 to 93 deg. POINTS does global astrometry, i.e., it measures widely separated targets, which yields closure calibration, numerous bright reference stars, and absolute parallax. Simplicity, stability, and the mitigation of systematic error are the central design themes. The instrument has only three moving-part mechanisms, and only one of these must move with sub-milliradian precision; the other two can tolerate a precision of several tenths of a degree. Optical surfaces preceding the beamsplitter or its fold flat are interferometrically critical; on each side of the interferometer, there are only three such. Thus, light loss and wavefront distortion are minimized. POINTS represents a minimalistic design developed ab initio for space. Since it is intended for astrometry, and therefore does not require the u-v-plane coverage of an imaging, instrument, each interferometer need have only two subapertures. The design relies on articulation of the angle between the interferometers and body pointing to select targets; the observations are restricted to the 'instrument plane.' That plane, which is fixed in the pointed instrument, is defined by the sensitive direction for the two interferometers. Thus, there is no need for siderostats and moving delay lines, which would have added many precision mechanisms with rolling and sliding parts that would be required to function throughout the mission. Further, there is no need for a third interferometer
Fourier transform interferometer alignment method.
Goldberg, Kenneth A; Naulleau, Patrick; Bokor, Jeffrey
2002-08-01
A rapid and convenient method has been developed to facilitate the alignment of the image-plane components of point-diffraction interferometers, including the phase-shifting point-diffraction interferometer. In real time, the Fourier transform of the detected image is used to calculate a pseudoimage of the electric field in the image plane of the test optic where thecritical alignment o f variousoptical components is performed. Reconstruction of the pseudoimage is similar to off-axis, Fourier transform holography. Intermediate steps in the alignment procedure are described. Fine alignment is aided by the introduction and optimization of a global-contrast parameter that is easily calculated from the Fourier transform. Additional applications include the alignment of image-plane apertures in general optical systems, the rapid identification of patterned image-plane alignment marks, and the probing of important image-plane field properties. PMID:12153074
Lensless reflective point diffraction interferometer.
Zhu, Wenhua; Chen, Lei; Zheng, Donghui; Yang, Ying; Han, Zhigang; Li, Jinpeng
2016-07-01
A lensless reflective point diffraction interferometer (LRPDI) is proposed for dynamic wavefront measurement. The point diffraction interferometer is integrated on a small substrate with properly designed thin film, which is used for generating the interferogram with high carrier frequency at a CCD target. By lensless imaging, the complex amplitude at the CCD target can be propagated to the conjugated plane of the exit pupil of an incident wavefront, which not only avoids the edge diffraction in the interferogram, but also eliminates systematic error. The accuracy of LRPDI is demonstrated by simulation and experiment, and a precision better than 1/150 wavelength is achieved. The new design with lensless imaging processing is suitable for dynamic wavefront measurement. PMID:27409204
Stellar Interferometer Technology Experiment (SITE)
NASA Technical Reports Server (NTRS)
Crawley, Edward F.; Miller, David; Laskin, Robert; Shao, Michael
1995-01-01
The MIT Space Engineering Research Center and the Jet Propulsion Laboratory stand ready to advance science sensor technology for discrete-aperture astronomical instruments such as space-based optical interferometers. The objective of the Stellar Interferometer Technology Experiment (SITE) is to demonstrate system-level functionality of a space-based stellar interferometer through the use of enabling and enhancing Controlled-Structures Technologies (CST). SITE mounts to the Mission Peculiar Experiment Support System inside the Shuttle payload bay. Starlight, entering through two apertures, is steered to a combining plate where it is interferred. Interference requires 27 nanometer pathlength (phasing) and 0.29 archsecond wavefront-tilt (pointing) control. The resulting 15 milli-archsecond angular resolution exceeds that of current earth-orbiting telescopes while maintaining low cost by exploiting active optics and structural control technologies. With these technologies, unforeseen and time-varying disturbances can be rejected while relaxing reliance on ground alignment and calibration. SITE will reduce the risk and cost of advanced optical space systems by validating critical technologies in their operational environment. Moreover, these technologies are directly applicable to commercially driven applications such as precision matching, optical scanning, and vibration and noise control systems for the aerospace, medical, and automotive sectors. The SITE team consists of experienced university, government, and industry researchers, scientists, and engineers with extensive expertise in optical interferometry, nano-precision opto-mechanical control and spaceflight experimentation. The experience exists and the technology is mature. SITE will validate these technologies on a functioning interferometer science sensor in order to confirm definitely their readiness to be baselined for future science missions.
Polarized-interferometer feasibility study
NASA Technical Reports Server (NTRS)
Raab, F. H.
1983-01-01
The feasibility of using a polarized-interferometer system as a rendezvous and docking sensor for two cooperating spacecraft was studied. The polarized interferometer is a radio frequency system for long range, real time determination of relative position and attitude. Range is determined by round trip signal timing. Direction is determined by radio interferometry. Relative roll is determined from signal polarization. Each spacecraft is equipped with a transponder and an antenna array. The antenna arrays consist of four crossed dipoles that can transmit or receive either circularly or linearly polarized signals. The active spacecraft is equipped with a sophisticated transponder and makes all measurements. The transponder on the passive spacecraft is a relatively simple repeater. An initialization algorithm is developed to estimate position and attitude without any a priori information. A tracking algorithm based upon minimum variance linear estimators is also developed. Techniques to simplify the transponder on the passive spacecraft are investigated and a suitable configuration is determined. A multiple carrier CW signal format is selected. The dependence of range accuracy and ambiguity resolution error probability are derived and used to design a candidate system. The validity of the design and the feasibility of the polarized interferometer concept are verified by simulation.
Polarized-interferometer feasibility study
NASA Astrophysics Data System (ADS)
Raab, F. H.
1983-07-01
The feasibility of using a polarized-interferometer system as a rendezvous and docking sensor for two cooperating spacecraft was studied. The polarized interferometer is a radio frequency system for long range, real time determination of relative position and attitude. Range is determined by round trip signal timing. Direction is determined by radio interferometry. Relative roll is determined from signal polarization. Each spacecraft is equipped with a transponder and an antenna array. The antenna arrays consist of four crossed dipoles that can transmit or receive either circularly or linearly polarized signals. The active spacecraft is equipped with a sophisticated transponder and makes all measurements. The transponder on the passive spacecraft is a relatively simple repeater. An initialization algorithm is developed to estimate position and attitude without any a priori information. A tracking algorithm based upon minimum variance linear estimators is also developed. Techniques to simplify the transponder on the passive spacecraft are investigated and a suitable configuration is determined. A multiple carrier CW signal format is selected. The dependence of range accuracy and ambiguity resolution error probability are derived and used to design a candidate system. The validity of the design and the feasibility of the polarized interferometer concept are verified by simulation.
Reissner-Mindlin Legendre Spectral Finite Elements with Mixed Reduced Quadrature
Brito, K. D.; Sprague, M. A.
2012-10-01
Legendre spectral finite elements (LSFEs) are examined through numerical experiments for static and dynamic Reissner-Mindlin plate bending and a mixed-quadrature scheme is proposed. LSFEs are high-order Lagrangian-interpolant finite elements with nodes located at the Gauss-Lobatto-Legendre quadrature points. Solutions on unstructured meshes are examined in terms of accuracy as a function of the number of model nodes and total operations. While nodal-quadrature LSFEs have been shown elsewhere to be free of shear locking on structured grids, locking is demonstrated here on unstructured grids. LSFEs with mixed quadrature are, however, locking free and are significantly more accurate than low-order finite-elements for a given model size or total computation time.
NASA Astrophysics Data System (ADS)
Evans, W. A. B.; Torre, A.
2012-11-01
The paper focusses on the advantages of using high-order Gauss-Legendre quadratures for the precise evaluation of integrals with both smooth and rapidly changing integrands. Aspects of their precision are analysed in the light of Gauss' error formula. Some "test examples" are considered and evaluated in multiple precision to ≈ 200 significant decimal digits with David Bailey's multiprecision package to eliminate truncation/rounding errors. The increase of precision on doubling the number of subintervals is analysed, the relevant quadrature attribute being the precision increment. In order to exemplify the advantages that high-order quadrature afford, the technique is then used to evaluate several plots of the Rayleigh-Sommerfeld diffraction integral for axi-symmetric source fields defined on a planar aperture. A comparison of the high-order quadrature method against various FFT-based methods is finally given.
Information entropy of Gegenbauer polynomials and Gaussian quadrature
NASA Astrophysics Data System (ADS)
Sánchez-Ruiz, Jorge
2003-05-01
In a recent paper (Buyarov V S, López-Artés P, Martínez-Finkelshtein A and Van Assche W 2000 J. Phys. A: Math. Gen. 33 6549-60), an efficient method was provided for evaluating in closed form the information entropy of the Gegenbauer polynomials C(lambda)n(x) in the case when lambda = l in Bbb N. For given values of n and l, this method requires the computation by means of recurrence relations of two auxiliary polynomials, P(x) and H(x), of degrees 2l - 2 and 2l - 4, respectively. Here it is shown that P(x) is related to the coefficients of the Gaussian quadrature formula for the Gegenbauer weights wl(x) = (1 - x2)l-1/2, and this fact is used to obtain the explicit expression of P(x). From this result, an explicit formula is also given for the polynomial S(x) = limnrightarrowinfty P(1 - x/(2n2)), which is relevant to the study of the asymptotic (n rightarrow infty with l fixed) behaviour of the entropy.
Electronically Tunable Differential Integrator: Linear Voltage Controlled Quadrature Oscillator
Nandi, Rabindranath; Pattanayak, Sandhya; Das, Sagarika
2015-01-01
A new electronically tunable differential integrator (ETDI) and its extension to voltage controlled quadrature oscillator (VCQO) design with linear tuning law are proposed; the active building block is a composite current feedback amplifier with recent multiplication mode current conveyor (MMCC) element. Recently utilization of two different kinds of active devices to form a composite building block is being considered since it yields a superior functional element suitable for improved quality circuit design. The integrator time constant (τ) and the oscillation frequency (ωo) are tunable by the control voltage (V) of the MMCC block. Analysis indicates negligible phase error (θe) for the integrator and low active ωo-sensitivity relative to the device parasitic capacitances. Satisfactory experimental verifications on electronic tunability of some wave shaping applications by the integrator and a double-integrator feedback loop (DIFL) based sinusoid oscillator with linear fo variation range of 60 KHz~1.8 MHz at low THD of 2.1% are verified by both simulation and hardware tests. PMID:27347537
Radiation transport modeling using extended quadrature method of moments
NASA Astrophysics Data System (ADS)
Vikas, V.; Hauck, C. D.; Wang, Z. J.; Fox, R. O.
2013-08-01
The radiative transfer equation describes the propagation of radiation through a material medium. While it provides a highly accurate description of the radiation field, the large phase space on which the equation is defined makes it numerically challenging. As a consequence, significant effort has gone into the development of accurate approximation methods. Recently, an extended quadrature method of moments (EQMOM) has been developed to solve univariate population balance equations, which also have a large phase space and thus face similar computational challenges. The distinct advantage of the EQMOM approach over other moment methods is that it generates moment equations that are consistent with a positive phase space density and has a moment inversion algorithm that is fast and efficient. The goal of the current paper is to present the EQMOM method in the context of radiation transport, to discuss advantages and disadvantages, and to demonstrate its performance on a set of standard one-dimensional benchmark problems that encompass optically thin, thick, and transition regimes. Special attention is given in the implementation to the issue of realizability—that is, consistency with a positive phase space density. Numerical results in one dimension are promising and lay the foundation for extending the same framework to multiple dimensions.
Electronically Tunable Differential Integrator: Linear Voltage Controlled Quadrature Oscillator.
Nandi, Rabindranath; Pattanayak, Sandhya; Venkateswaran, Palaniandavar; Das, Sagarika
2015-01-01
A new electronically tunable differential integrator (ETDI) and its extension to voltage controlled quadrature oscillator (VCQO) design with linear tuning law are proposed; the active building block is a composite current feedback amplifier with recent multiplication mode current conveyor (MMCC) element. Recently utilization of two different kinds of active devices to form a composite building block is being considered since it yields a superior functional element suitable for improved quality circuit design. The integrator time constant (τ) and the oscillation frequency (ω o ) are tunable by the control voltage (V) of the MMCC block. Analysis indicates negligible phase error (θ e ) for the integrator and low active ω o -sensitivity relative to the device parasitic capacitances. Satisfactory experimental verifications on electronic tunability of some wave shaping applications by the integrator and a double-integrator feedback loop (DIFL) based sinusoid oscillator with linear f o variation range of 60 KHz~1.8 MHz at low THD of 2.1% are verified by both simulation and hardware tests. PMID:27347537
A fast method of numerical quadrature for p-version finite element matrices
NASA Technical Reports Server (NTRS)
Hinnant, Howard E.
1993-01-01
A new technique of numerical quadrature especially suited for p-version finite element matrices is presented. This new technique separates the integrand into two parts, and numerically operates on each part separately. The objective of this scheme is to minimize the computational cost of integrating the entire element matrix as opposed to minimizing the cost of integrating a single function. The efficiency of the new technique is compared with Gaussian quadrature and found to take a small fraction of the computational effort.
Optimization and Experimentation of Dual-Mass MEMS Gyroscope Quadrature Error Correction Methods
Cao, Huiliang; Li, Hongsheng; Kou, Zhiwei; Shi, Yunbo; Tang, Jun; Ma, Zongmin; Shen, Chong; Liu, Jun
2016-01-01
This paper focuses on an optimal quadrature error correction method for the dual-mass MEMS gyroscope, in order to reduce the long term bias drift. It is known that the coupling stiffness and demodulation error are important elements causing bias drift. The coupling stiffness in dual-mass structures is analyzed. The experiment proves that the left and right masses’ quadrature errors are different, and the quadrature correction system should be arranged independently. The process leading to quadrature error is proposed, and the Charge Injecting Correction (CIC), Quadrature Force Correction (QFC) and Coupling Stiffness Correction (CSC) methods are introduced. The correction objects of these three methods are the quadrature error signal, force and the coupling stiffness, respectively. The three methods are investigated through control theory analysis, model simulation and circuit experiments, and the results support the theoretical analysis. The bias stability results based on CIC, QFC and CSC are 48 °/h, 9.9 °/h and 3.7 °/h, respectively, and this value is 38 °/h before quadrature error correction. The CSC method is proved to be the better method for quadrature correction, and it improves the Angle Random Walking (ARW) value, increasing it from 0.66 °/√h to 0.21 °/√h. The CSC system general test results show that it works well across the full temperature range, and the bias stabilities of the six groups’ output data are 3.8 °/h, 3.6 °/h, 3.4 °/h, 3.1 °/h, 3.0 °/h and 4.2 °/h, respectively, which proves the system has excellent repeatability. PMID:26751455
Nonlocal polarization interferometer for entanglement detection
Williams, Brian P.; Humble, Travis S.; Grice, Warren P.
2014-10-30
We report a nonlocal interferometer capable of detecting entanglement and identifying Bell states statistically. This is possible due to the interferometer's unique correlation dependence on the antidiagonal elements of the density matrix, which have distinct bounds for separable states and unique values for the four Bell states. The interferometer consists of two spatially separated balanced Mach-Zehnder or Sagnac interferometers that share a polarization-entangled source. Correlations between these interferometers exhibit nonlocal interference, while single-photon interference is suppressed. This interferometer also allows for a unique version of the Clauser-Horne-Shimony-Holt Bell test where the local reality is the photon polarization. In conclusion, we present the relevant theory and experimental results.
Nonlocal polarization interferometer for entanglement detection
NASA Astrophysics Data System (ADS)
Williams, Brian P.; Humble, Travis S.; Grice, Warren P.
2014-10-01
We report a nonlocal interferometer capable of detecting entanglement and identifying Bell states statistically. This is possible due to the interferometer's unique correlation dependence on the antidiagonal elements of the density matrix, which have distinct bounds for separable states and unique values for the four Bell states. The interferometer consists of two spatially separated balanced Mach-Zehnder or Sagnac interferometers that share a polarization-entangled source. Correlations between these interferometers exhibit nonlocal interference, while single-photon interference is suppressed. This interferometer also allows for a unique version of the Clauser-Horne-Shimony-Holt-Bell test where the local reality is the photon polarization. We present the relevant theory and experimental results.
Nonlocal polarization interferometer for entanglement detection
Williams, Brian P.; Humble, Travis S.; Grice, Warren P.
2014-10-30
We report a nonlocal interferometer capable of detecting entanglement and identifying Bell states statistically. This is possible due to the interferometer's unique correlation dependence on the antidiagonal elements of the density matrix, which have distinct bounds for separable states and unique values for the four Bell states. The interferometer consists of two spatially separated balanced Mach-Zehnder or Sagnac interferometers that share a polarization-entangled source. Correlations between these interferometers exhibit nonlocal interference, while single-photon interference is suppressed. This interferometer also allows for a unique version of the Clauser-Horne-Shimony-Holt Bell test where the local reality is the photon polarization. In conclusion, wemore » present the relevant theory and experimental results.« less
Directional dual-tree complex wavelet packet transforms for processing quadrature signals.
Serbes, Gorkem; Gulcur, Halil Ozcan; Aydin, Nizamettin
2016-03-01
Quadrature signals containing in-phase and quadrature-phase components are used in many signal processing applications in every field of science and engineering. Specifically, Doppler ultrasound systems used to evaluate cardiovascular disorders noninvasively also result in quadrature format signals. In order to obtain directional blood flow information, the quadrature outputs have to be preprocessed using methods such as asymmetrical and symmetrical phasing filter techniques. These resultant directional signals can be employed in order to detect asymptomatic embolic signals caused by small emboli, which are indicators of a possible future stroke, in the cerebral circulation. Various transform-based methods such as Fourier and wavelet were frequently used in processing embolic signals. However, most of the times, the Fourier and discrete wavelet transforms are not appropriate for the analysis of embolic signals due to their non-stationary time-frequency behavior. Alternatively, discrete wavelet packet transform can perform an adaptive decomposition of the time-frequency axis. In this study, directional discrete wavelet packet transforms, which have the ability to map directional information while processing quadrature signals and have less computational complexity than the existing wavelet packet-based methods, are introduced. The performances of proposed methods are examined in detail by using single-frequency, synthetic narrow-band, and embolic quadrature signals. PMID:25388779
Dual-domain point diffraction interferometer.
Naulleau, P P; Goldberg, K A
1999-06-01
The phase-shifting point diffraction interferometer has recently been developed and implemented at Lawrence Berkeley National Laboratory to meet the significant metrology challenge of characterizing extreme ultraviolet projection lithography systems. Here we present a refined version of this interferometer that overcomes the original design's susceptibility to noise attributed to scattered light. The theory of the new hybrid spatial- and temporal-domain (dual-domain) point diffraction interferometer is described in detail and experimental results are presented. PMID:18319953
A multichannel interferometer for electron density measurements in COMPASS
NASA Astrophysics Data System (ADS)
Edlington, Trevor; Wylde, Richard
1992-10-01
A compact seven channel interferometer has been designed and built to measure electron density profiles in the COMPASS (compact assembly) tokamak. Two far-infrared (FIR) laser cavities are optically pumped with a single continuous-wave CO2 laser, generating two similar beams at λ=433 μm with a small, tunable difference frequency (0.5-1.0 MHz). The COMPASS facility incorporates a complex set of poloidal field coils close to the vacuum vessel as well as a versatile set of close coupled ``helical'' resonant magnetic perturbation windings which severely restrict diagnostic access. As a result a novel approach to the optical circuit has been necessary. Wire grid polarizers are used to divide the laser power equally between channels and to overlay probing and local oscillator beams after the probe beams have made a double pass through the plasma. Gaussian beam-mode optics is used to minimize the size of the optical components.
Golovashkin, A. I.; Zherikhina, L. N. Tskhovrebov, A. M.; Izmailov, G. N.; Ozolin, V. V.
2010-08-15
When comparing the operation of a superfluid helium matter wave quantum interferometer (He SQUID) with that of an ordinary direct-current quantum interferometer (dc SQUID), we estimate their resolution limitation that correspond to quantum fluctuations. An alternative mode of operation of the interferometer as a unified macroquantum system is considered.
NASA Astrophysics Data System (ADS)
Ming, Hai; Sun, Xiaohong; Bai, Ming; Liang, Zhongcheng; Zhang, Guoping; Li, Hui; Wu, Yunxia; Xie, Jiangping; Shu, Jizu
1998-08-01
By using composite self-focus rod, LD-pumped frequency doubling Nd:YVO4 laser and 3D fine regulation stand, a new Fizeau-type interferometer is developed. It can apply to measuring the micro-vibration and deformation of micro- surface at the same time. The theoretical analysis and the experimental results for the interferometer of self-focus sensor are given.
Simple Ultra-Low-Cost Undergraduate Holography Using a Modified Michelson Interferometer.
ERIC Educational Resources Information Center
Rudmin, J. W.; And Others
1980-01-01
A technique is presented for producing holograms using equipment which is already in the possesion of the majority of college physics departments, which includes a slightly modified Michelson interferometer, a helium-neon laser, and a long focal-length lens. Production of high quality holograms has been achieved by inexperienced undergraduates…
Integrated structural and optical modeling of the orbiting stellar interferometer
NASA Astrophysics Data System (ADS)
Shaklan, Stuart B.; Yu, Jeffrey W.; Briggs, Hugh C.
1993-11-01
The Integrated Modeling of Optical Systems (IMOS) Integration Workbench at JPL has been used to model the effects of structural perturbations on the optics in the proposed Orbiting Stellar Interferometer (OSI). OSI consists of 3 pairs of interferometers and delay lines attached to a 7.5 meter truss. They are interferometrically monitored from a separate boom by a laser metrology system. The spatially distributed nature of the science instrument calls for a high level of integration between the optics and support structure. Because OSI is designed to achieve micro-arcsecond astrometry, many of its alignment, stability, and knowledge tolerances are in the submicron regime. The spacecraft will be subject to vibrations caused by reaction wheels and on-board equipment, as well as thermal strain due to solar and terrestrial heating. These perturbations affect optical parameters such as optical path differences and beam co-parallelism which are critical to instrument performance. IMOS provides an environment that allows one to design and perturb the structure, attach optics to structural or non-structural nodes, trace rays, and analyze the impact of mechanical perturbations on optical performance. This tool makes it simple to change the structure and immediately see performance enhancement/degradation. We have employed IMOS to analyze the effect of reaction wheel disturbances on the optical path difference in both the science and metrology interferometers.
Imaging interferometers for analysis of Thomson scattered spectraa)
NASA Astrophysics Data System (ADS)
Howard, J.; Hatae, T.
2008-10-01
Polarization interferometers have some potential efficiency advantages for imaging Thomson scattering spectral analysis. In this article we present a number of designs for high-efficiency imaging polarization interferometers for Thomson scattering spectral analysis. The use of high-efficiency crystal polarizing beamsplitters (both displacement and angle) results in low-loss complementary passbands (no edge losses), simple imaging systems, and wide field of view. The efficiency and relative merits of both multiple-filter and dispersive-type configurations are being assessed before installation on the JT-60U ruby-laser Thomson scattering system. Light is transferred from the viewing port via a linear array of optical fiber bundles which will be imaged through the interferometer onto the photocathode of an intensified charge coupled device camera. Because of the broadband nature of the Thomson light, the optical delays required to Fourier analyze the spectrum are quite small. This leads to compact multicolor or dispersive systems based on combinations of Wollaston and Savart splitters and traditional waveplates.
Integrated Optical Interferometers with Micromachined Diaphragms for Pressure Sensing
NASA Technical Reports Server (NTRS)
DeBrabander, Gregory N.; Boyd, Joseph T.
1996-01-01
Optical pressure sensors have been fabricated which use an integrated optical channel waveguide that is part of an interferometer to measure the pressure-induced strain in a micromachined silicon diaphragm. A silicon substrate is etched from the back of the wafer leaving a rectangular diaphragm. On the opposite side of the wafer, ring resonator and Mach-Zehnder interferometers are formed with optical channel waveguides made from a low pressure chemical vapor deposited film of silicon oxynitride. The interferometer's phase is altered by pressure-induced stress in a channel segment positioned over the long edge of the diaphragm. The phase change in the ring resonator is monitored using a link-insensitive swept frequency laser diode, while in the Mach-Zehnder it is determined using a broad band super luminescent diode with subsequent wavelength separation. The ring resonator was found to be highly temperature sensitive, while the Mach-Zehnder, which had a smaller optical path length difference, was proportionally less so. The quasi-TM mode was more sensitive to pressure, in accord with calculations. Waveguide and sensor theory, sensitivity calculations, a fabrication sequence, and experimental results are presented.
Spectropolarimetric analysis of differential interferometer
NASA Astrophysics Data System (ADS)
Gut, Kazimierz
2014-08-01
The paper presents the principle of the operation of a spectropolarimetric interferometer. In the planar waveguide orthogonal modes of type TE and TM can be excited for the entire visible light. During the propagation the difference of the phases between the modes is determined, which is the function of the length of the path of propagation, the difference of the effective refractive index (NTM-NTE) and the wavelength. At the output of this system the spectral distribution of intensity is recorded, the shape of which depends on the value of the refractive index of the cover of the waveguides.
Solar CIV Vacuum-Ultraviolet Fabry-Perot Interferometers
NASA Technical Reports Server (NTRS)
Gary, G. Allen; West, Edward A.; Rees, David; McKay, Jack A.; Zukic, Maumer; Herman, Peter
2006-01-01
Aims: A tunable, high spectral resolution, high effective finesse, vacuum ultraviolet (VUV) Fabry-Perot interferometer (PPI) is designed for obtaining narrow-passband images, magnetograms, and Dopplergrams of the transition region emission line of CIV (155 nm). Methods: The integral part of the CIV narrow passband filter package (with a 2-10 pm FWHM) consists of a multiple etalon system composed of a tunable interferometer that provides high-spectral resolution and a static low-spectral resolution interferometer that allows a large effective free spectral range. The prefilter for the interferometers is provided by a set of four mirrors with dielectric high-reflective coatings. A tunable interferometer, a VUV piezoelectric-control etalon, has undergone testing using the surrogate F2 eximer laser line at 157 nm for the CIV line. We present the results of the tests with a description of the overall concept for a complete narrow-band CIV spectral filter. The static interferometer of the filter is envisioned as being hudt using a set of fixed MgF2 plates. The four-mirror prefilter is designed to have dielectric multilayer n-stacks employing the design concept used in the Ultraviolet Imager of NASA's Polar Spacecraft. A dual etalon system allows the effective free spectral range to be commensurate with the prefilter profile. With an additional etalon, a triple etalon system would allow a spectrographic resolution of 2 pm. The basic strategy has been to combine the expertise of spaceflight etalon manufacturing with VUV coating technology to build a VUV FPI which combines the best attributes of imagers and spectrographs into a single compact instrument. Results. Spectro-polarimetry observations of the transition region CIV emission can be performed to increase the understanding of the magnetic forces, mass motion, evolution, and energy release within the solar atmosphere at the base of the corona where most of the magnetic field is approximately force-free. The 2D imaging
NASA Technical Reports Server (NTRS)
Taghavi-Larigani, Shervin (Inventor); Vanzyl, Jakob J. (Inventor); Yariv, Amnon (Inventor)
2006-01-01
Tunable semiconductor lasers are disclosed requiring minimized coupling regions. Multiple laser embodiments employ ring resonators or ring resonator pairs using only a single coupling region with the gain medium are detailed. Tuning can be performed by changing the phase of the coupling coefficient between the gain medium and a ring resonator of the laser. Another embodiment provides a tunable laser including two Mach-Zehnder interferometers in series and a reflector coupled to a gain medium.
In-fiber modal interferometer for high sensitivity gas detection
NASA Astrophysics Data System (ADS)
Yang, Fan; Jin, Wei
2015-09-01
We demonstrate a gas sensor based on mode interference in a hollow-core photonic bandgap fiber. Gas absorption of a pump laser beam induces phase modulation of a propagating probe beam, which is detected by use of an in-fiber modal interferometer. An estimated detection limit of ~2 ppm acetylene (~7x10-5 in terms of noise equivalent absorbance or NEA) is achieved with 30-cm-long HC-PBF operating at the near infrared wavelength. This NEA is ~22 times better than state-of-the-art HC-PBF gas sensors based on direct absorption spectroscopy.
Hand held phase-shifting diffraction moire interferometer
Deason, Vance A.; Ward, Michael B.
1994-01-01
An interferometer in which a coherent beam of light is generated within a remote case and transmitted to a hand held unit tethered to said remote case, said hand held unit having optical elements for directing a pair of mutually coherent collimated laser beams at a diffraction grating. Data from the secondary or diffracted beams are then transmitted to a separate video and data acquisition system for recording and analysis for load induced deformation or for identification purposes. Means are also provided for shifting the phase of one incident beam relative to the other incident beam and being controlled from within said remote case.
Hand held phase-shifting diffraction Moire interferometer
Deason, V.A.; Ward, M.B.
1994-09-20
An interferometer is described in which a coherent beam of light is generated within a remote case and transmitted to a hand held unit tethered to said remote case, said hand held unit having optical elements for directing a pair of mutually coherent collimated laser beams at a diffraction grating. Data from the secondary or diffracted beams are then transmitted to a separate video and data acquisition system for recording and analysis for load induced deformation or for identification purposes. Means are also provided for shifting the phase of one incident beam relative to the other incident beam and being controlled from within said remote case. 4 figs.
Modernization of Koesters interferometer and high accuracy calibration gauge blocks
NASA Astrophysics Data System (ADS)
França, R. S.; Silva, I. L. M.; Couceiro, I. B.; Torres, M. A. C.; Bessa, M. S.; Costa, P. A.; Oliveira, W., Jr.; Grieneisen, H. P. H.
2016-07-01
The Optical Metrology Division (Diopt) of Inmetro is responsible for maintaining the national reference of the length unit according to International System of Units (SI) definitions. The length unit is realized by interferometric techniques and is disseminated to the dimensional community through calibrations of gauge blocks. Calibration of large gauge blocks from 100 mm to 1000 mm has been performed by Diopt with a Koesters interferometer with reference to spectral lines of a krypton discharge lamp. Replacement of this lamp by frequency stabilized lasers, traceable now to the time and frequency scale, is described and the first results are reported.
Quantum noise in differential-type gravitational-wave interferometer and signal recycling
NASA Astrophysics Data System (ADS)
Nishizawa, Atsushi; Sakagami, Masa-Aki; Kawamura, Seiji
2007-08-01
There exists the standard quantum limit (SQL), derived from Heisenberg’s uncertainty relation, in the sensitivity of laser interferometer gravitational-wave (GW) detectors. However, in the context of a full quantum-mechanical approach, SQL can be overcome using the correlation of shot noise and radiation-pressure noise. So far, signal recycling, which is one of the methods to overcome SQL, is considered only in a recombined-type interferometer such as Advanced LIGO, LCGT, and GEO600. In this paper, we investigated quantum noise and the possibility of signal recycling in a differential-type interferometer. As a result, we found that signal recycling is possible and creates at most three dips in the sensitivity curve of the detector due to two coupled resonators. The additional third dip makes it possible to decrease quantum noise at low frequencies, keeping the moderate sensitivity at high frequencies. Then, taking advantage of the third dip and comparing the sensitivity of a differential-type interferometer with that of a next-generation Japanese GW interferometer, LCGT, we found that signal-to-noise ratio (SNR) of inspiral binary is improved by a factor of ≈1.43 for neutron star binary, ≈2.28 for 50M⊙ black hole binary, and ≈2.94 for 100M⊙ black hole binary. We also found that power recycling to increase laser power is possible in our signal-recycling configuration of a detector.
Takada, Kazumasa; Seino, Mitsuyoshi; Chiba, Akito; Okamoto, Katsunari
2013-04-20
We calculate the root mean square (rms) value of the spectral noise caused by optical path phase measurement errors in a spatial heterodyne spectrometer (SHS) featuring a complex Fourier transformation. In our calculation the deviated phases of each Mach-Zehnder interferometer in the in-phase and quadrature states are treated as statistically independent random variables. We show that the rms value is proportional to the rms error of the phase measurement and that the proportionality coefficient is given analytically. The relationship enables us to estimate the potential performance of the SHS such as the sidelobe suppression ratio for a given measurement error. PMID:23669661
Extended Gaussian quadratures for functions with an end-point singularity of logarithmic type
NASA Astrophysics Data System (ADS)
Pachucki, K.; Puchalski, M.; Yerokhin, V. A.
2014-11-01
The extended Gaussian quadrature rules are shown to be an efficient tool for numerical integration of wide class of functions with singularities of logarithmic type. The quadratures are exact for the functions pol1n-1(x)+lnx pol2n-1(x), where pol1n-1(x) and pol2n-1(x) are two arbitrary polynomials of degree n-1 and n is the order of the quadrature formula. We present an implementation of numerical algorithm that calculates the nodes and the weights of the quadrature formulas, provide a Fortran code for numerical integration, and test the performance of different kinds of Gaussian quadratures for functions with logarithmic singularities. Catalogue identifier: AETP_v1_0 Program summary URL:http://cpc.cs.qub.ac.uk/summaries/AETP_v1_0.html Program obtainable from: CPC Program Library, Queen’s University, Belfast, N. Ireland Licensing provisions: Standard CPC licence, http://cpc.cs.qub.ac.uk/licence/licence.html No. of lines in distributed program, including test data, etc.: 2535 No. of bytes in distributed program, including test data, etc.: 39 963 Distribution format: tar.gz Programming language: Mathematica, Fortran. Computer: PCs or higher performance computers. Operating system: Linux, Windows, MacOS. RAM: Kilobytes. Classification: 4.11. Nature of problem: Quadrature formulas for numerical integration, effective for a wide class of functions with end-point singularities of logarithmic type. Solution method: The method of solution is based on the algorithm developed in Ref. [1] with some modifications. Running time: Milliseconds to minutes. J. Ma, V. Rokhlin, S. Wandzura, Generalized Gaussian quadrature rules for systems of arbitrary functions, Soc. Indust. Appl. Math. J. Numer. Anal. 33 (3) (1996) 971-996.
A new method for determining the plasma electron density using optical frequency comb interferometer
Arakawa, Hiroyuki Tojo, Hiroshi; Sasao, Hajime; Kawano, Yasunori; Itami, Kiyoshi
2014-04-15
A new method of plasma electron density measurement using interferometric phases (fractional fringes) of an optical frequency comb interferometer is proposed. Using the characteristics of the optical frequency comb laser, high density measurement can be achieved without fringe counting errors. Simulations show that the short wavelength and wide wavelength range of the laser source and low noise in interferometric phases measurements are effective to reduce ambiguity of measured density.
The DELTA Synchrotron Light Interferometer
Berges, U.
2004-05-12
Synchrotron radiation sources like DELTA, the Dortmund Electron Accelerator, a third generation synchrotron light source, need an optical monitoring system to measure the beam size at different points of the ring with high resolution and accuracy. These measurements also allow an investigation of the emittance of the storage ring, an important working parameter for the efficiency of working beamlines with experiments using the synchrotron radiation. The resolution limits of the different types of optical synchrotron light monitors at DELTA are investigated. The minimum measurable beamsize with the normal synchrotron light monitor using visible light at DELTA is about 80 {mu}m. Due to this a synchrotron light interferometer was built up and tested at DELTA. The interferometer uses the same beamline in the visible range. The minimum measurable beamsize is with about 8 {mu}m one order of magnitude smaller. This resolution is sufficient for the expected small vertical beamsizes at DELTA. The electron beamsize and emittance were measured with both systems at different electron beam energies of the storage ring. The theoretical values of the present optics are smaller than the measured emittance. So possible reasons for beam movements are investigated.
Interferometer for Space Station Windows
NASA Technical Reports Server (NTRS)
Hall, Gregory
2003-01-01
Inspection of space station windows for micrometeorite damage would be a difficult task insitu using current inspection techniques. Commercially available optical profilometers and inspection systems are relatively large, about the size of a desktop computer tower, and require a stable platform to inspect the test object. Also, many devices currently available are designed for a laboratory or controlled environments requiring external computer control. This paper presents an approach using a highly developed optical interferometer to inspect the windows from inside the space station itself using a self- contained hand held device. The interferometer would be capable as a minimum of detecting damage as small as one ten thousands of an inch in diameter and depth while interrogating a relatively large area. The current developmental state of this device is still in the proof of concept stage. The background section of this paper will discuss the current state of the art of profilometers as well as the desired configuration of the self-contained, hand held device. Then, a discussion of the developments and findings that will allow the configuration change with suggested approaches appearing in the proof of concept section.
Dual-prism interferometer for collimation testing
Hii, King Ung; Kwek, Kuan Hiang
2009-01-10
An air-wedge lateral-shear interferometer using two prisms is presented. With a variable shear, the interferometer is suitable for testing collimation of a wide range of beam sizes down to a few millimeters in diameter. No antireflection coatings are necessary. Collimation for a light source with short coherent length is also demonstrated.
MBI: Millimetre-wave bolometric interferometer
NASA Astrophysics Data System (ADS)
Ali, S.; Rossinot, P.; Piccirillo, L.; Gear, W. K.; Mauskopf, P.; Ade, P.; Haynes, V.; Timbie, P.
2002-05-01
We present the design of the prototype of a millimeter-wave bolometric interferometer (MBI). This interferometer uses two arrays bolometers as detectors. The combination of high sensitivity bolometers and interferometric imaging appears to be well suited for precision measurements in observational cosmology. .
CIST....CORRTEX interferometer simulation test
Heinle, R.A.
1994-12-01
Testing was performed in order to validate and cross calibrate an RF interferometer and the crush threshold of cable. Nitromethane was exploded (inside of PVC pipe). The explosion was used to crush the interferometer sensor cables which had been placed inside and outside the pipe. Results are described.
Study Of Space-Based Optical Interferometer
NASA Technical Reports Server (NTRS)
Redding, David C.; Laskin, Robert A.; Breckenridge, William G.; Shao, Michael
1992-01-01
Report discusses calibration and operation of conceptual Focus Mission Interferometer (FMI), consisting of component instruments mounted at widely separated locations on large truss structure in orbit 1,400 km above Earth. Includes six telescopes in linear array. Outputs combined in pairlike fashion so FMI operates as three distinct two-telescope interferometers. Accurate enough for submilliarcsecond astrometry.
NASA Astrophysics Data System (ADS)
Čársky, Petr
2010-09-01
The UGU term was used as a model of the UGT term, and its evaluation by numerical quadrature was examined systematically with a training set of eight molecules. Minimum numbers of points have been determined for radial Gauss-Legendre and angular Lebedev quadratures that preserve the accuracy needed for practical applications. These quadratures are recommended for efficient calculation of electron scattering by polyatomic molecules.
A Concept of Multi-Mode High Spectral Resolution Lidar Using Mach-Zehnder Interferometer
NASA Astrophysics Data System (ADS)
Jin, Yoshitaka; Sugimoto, Nobuo; Nishizawa, Tomoaki; Ristori, Pablo; Otero, Lidia
2016-06-01
In this paper, we present the design of a High Spectral Resolution Lidar (HSRL) using a laser that oscillates in a multi-longitudinal mode. Rayleigh and Mie scattering components are separated using a Mach-Zehnder Interferometer (MZI) with the same free spectral range (FSR) as the transmitted laser. The transmitted laser light is measured as a reference signal with the same MZI. By scanning the MZI periodically with a scanning range equal to the mode spacing, we can identify the maximum Mie and the maximum Rayleigh signals using the reference signal. The cross talk due to the spectral width of each laser mode can also be estimated.
Orientational atom interferometers sensitive to gravitational waves
Lorek, Dennis; Laemmerzahl, Claus; Wicht, Andreas
2010-02-15
We present an atom interferometer that differs from common atom interferometers as it is not based on the spatial splitting of electronic wave functions, but on orienting atoms in space. As an example we present how an orientational atom interferometer based on highly charged hydrogen-like atoms is affected by gravitational waves. We show that a monochromatic gravitational wave will cause a frequency shift that scales with the binding energy of the system rather than with its physical dimension. For a gravitational wave amplitude of h=10{sup -23} the frequency shift is of the order of 110 {mu}Hz for an atom interferometer based on a 91-fold charged uranium ion. A frequency difference of this size can be resolved by current atom interferometers in 1 s.
Focused-laser interferometric position sensor
Friedman, Stephen J.; Barwick, Brett; Batelaan, Herman
2005-12-15
We describe a simple method to measure the position shifts of an object with a range of tens of micrometers using a focused-laser (FL) interferometric position sensor. In this article we examine the effects of mechanical vibration on FL and Michelson interferometers. We tested both interferometers using vibration amplitudes ranging from 0 to 20 {mu}m. Our FL interferometer has a resolution much better than the diffraction grating periodicities of 10 and 14 {mu}m used in our experiments. A FL interferometer provides improved mechanical stability at the expense of spatial resolution. Our experimental results show that Michelson interferometers cannot be used when the vibration amplitude is more than an optical wavelength. The main purpose of this article is to demonstrate that a focused-laser interferometric position sensor can be used to measure the position shifts of an object on a less sensitive, micrometer scale when the vibration amplitude is too large to use a Michelson interferometer.